Thermo Scientific (c) 2011

          Versa Star Pro    

Meter S/N               V122329
SW Rev                  11.50

Channel                 1
Module S/n              VA36273
Method                  100
Calibration Time        06-10-24,00:39

-- pH Calibration Report -- 

Point                   1
pH                      1.68
mV                      294.7
Temperature             22.8 C (ATC)
Calibration Type        Auto

Point                   2
pH                      4.01
mV                      152.8
Temperature             22.8 C (ATC)
Calibration Type        Auto

Point                   3
pH                      7.00
mV                      -16.5
Temperature             22.9 C (ATC)
Calibration Type        Auto

Point                   4
pH                      10.01
mV                      -188.6
Temperature             22.8 C (ATC)
Calibration Type        Manual

Point                   5
pH                      12.46
mV                      -335.8
Temperature             22.7 C (ATC)
Calibration Type        Manual

Slope 1                103.7 %
Slope 2                96.4 %
Slope 3                97.4 %
Slope 4                102.4 %
Eo    1                -29.2 mV
Eo    2                -16.5 mV
Eo    3                -16.5 mV
Eo    4                -7.6 mV
Average Slope           100.0 %
Cal #                   109

