Thermo Scientific (c) 2011

          Versa Star Pro    

Meter S/N               V122322
SW Rev                  11.50

Channel                 1
Module S/n              VA36473
Method                  100
Calibration Time        03-10-24,10:33

-- pH Calibration Report -- 

Point                   1
pH                      1.68
mV                      326.0
Temperature             25.0 C (ATC)
Calibration Type        Manual

Point                   2
pH                      4.01
mV                      180.1
Temperature             25.3 C (ATC)
Calibration Type        Auto

Point                   3
pH                      7.00
mV                      4.7
Temperature             23.4 C (ATC)
Calibration Type        Auto

Point                   4
pH                      10.01
mV                      -167.1
Temperature             25.0 C (ATC)
Calibration Type        Auto

Point                   5
pH                      12.49
mV                      -314.4
Temperature             24.2 C (ATC)
Calibration Type        Auto

Slope 1                105.9 %
Slope 2                99.1 %
Slope 3                96.5 %
Slope 4                101.0 %
Eo    1                -7.4 mV
Eo    2                4.7 mV
Eo    3                4.7 mV
Eo    4                12.7 mV
Average Slope           100.6 %
Cal #                   107

