(1497,'2024-11-08','07:15:32','90.00000','105.00000','pH Buffer','  NA','NA','E4INS24005847','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','1.68 pH',1,'84378','Shivanand. W','2024-11-08 07:44:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   301.3 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\nLog #                886\r\n\r\n\r\n'),
(1498,'2024-11-08','07:22:17','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005847','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','4.01pH',1,'84378','Shivanand. W','2024-11-08 07:45:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   162.5 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1499,'2024-11-08','07:25:47','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005847','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','7.00pH',1,'84378','Shivanand. W','2024-11-08 07:45:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   -12.5 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1500,'2024-11-08','07:29:17','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005847','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','10.01 pH',1,'84378','Shivanand. W','2024-11-08 07:46:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.96 pH\r\nmV                   -186.8 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1501,'2024-11-08','07:32:29','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005847','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','12.46 pH',1,'84378','Shivanand. W','2024-11-08 07:46:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.45 pH\r\nmV                   -333.4 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1502,'2024-11-08','08:46:58','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424120','E4FP24000734','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Pipertazoinj',1,'84378','Shivanand. W','2024-11-08 09:33:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.71 pH\r\nmV                   63.2 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1503,'2024-11-08','08:55:01','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424121','E4FP24000739','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Pipertazoinj',1,'84378','Shivanand. W','2024-11-08 09:34:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.61 pH\r\nmV                   69.1 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1504,'2024-11-08','08:59:11','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TT0024016','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','3',1,'95681','T. Sivanarayana','2024-11-08 09:20:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.29 pH\r\nmV                   22.2 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             08-11-24,00:29\r\n\r\n\r\n'),
(1505,'2024-11-08','09:07:07','90.00000','105.00000','Media Fill','pH','LY24007','E4AMV24000209','ASEPTIC MEDIA FILL','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Media fill pH',1,'84378','Shivanand. W','2024-11-08 09:35:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.33 pH\r\nmV                   -32.0 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1506,'2024-11-08','09:15:15','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TT0024016','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','3',1,'95681','T. Sivanarayana','2024-11-08 09:21:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.26 pH\r\nmV                   23.1 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             08-11-24,00:29\r\n\r\n\r\n'),
(1507,'2024-11-08','09:39:39','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005856','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-11-08 10:44:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.70 pH\r\nmV                   308.0 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1508,'2024-11-08','09:43:29','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005856','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-11-08 10:45:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.04 pH\r\nmV                   171.5 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1509,'2024-11-08','09:45:29','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005856','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-11-08 10:46:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -2.9 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1510,'2024-11-08','09:53:39','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005856','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-11-08 10:47:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.00 pH\r\nmV                   -175.2 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1511,'2024-11-08','09:57:46','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005856','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-11-08 10:47:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.47 pH\r\nmV                   -319.0 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1512,'2024-11-08','10:24:26','90.00000','105.00000','Piperacillin and Tazobactam For injection','UOD','PT0424120','E4FP24000734','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205075','G. Shyam Kumar',2,0,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 14:35:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   77.4 mV\r\nTemperature          31.6 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1513,'2024-11-08','10:34:21','90.00000','105.00000','Piperacillin and Tazobactam for Injection','IDENTIFICATION','2449202490','E4RM24005889','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','Mobile phase pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 14:26:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   77.4 mV\r\nTemperature          31.8 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1514,'2024-11-08','14:13:34','90.00000','105.00000','Nafcillin for Injection','pH','NF0124001A','E4STFP24000698','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Nafcillin pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 14:25:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.21 pH\r\nmV                   -25.0 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1515,'2024-11-08','14:16:16','90.00000','105.00000','Nafcillin for Injection','pH','NF0124001A','E4STFP24000699','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Nafcillin pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 14:25:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.13 pH\r\nmV                   -19.9 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1516,'2024-11-08','14:25:55','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TT0024016','E4IP24001328','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','pH (Bulk)',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 14:36:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.90 pH\r\nmV                   52.1 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1517,'2024-11-08','14:41:10','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24020','E4SO24000813','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SCDBIH',1,'85152','V. Chandra Shekara Achari','2024-11-08 21:31:23','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.44 pH\r\nmV                   -26.4 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1518,'2024-11-08','14:44:42','90.00000','105.00000','Macckankey broth','NA','RS009/24020','E4SO24000814','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','NA',1,'85152','V. Chandra Shekara Achari','2024-11-08 21:32:05','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.38 pH\r\nmV                   -23.0 mV\r\nTemperature          22.8 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1519,'2024-11-08','15:33:27','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24006019','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 16:23:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   64.5 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1520,'2024-11-08','15:36:25','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24006020','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 16:25:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.61 pH\r\nmV                   68.8 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\nLog #                900\r\n\r\n\r\n'),
(1521,'2024-11-08','15:41:53','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24006021','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 16:26:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   73.4 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1522,'2024-11-08','15:44:41','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24006022','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 16:26:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   73.1 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1523,'2024-11-08','15:47:16','90.00000','105.00000','Purified water','pH','PW/SP34','E4RM24006023','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 16:27:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.57 pH\r\nmV                   71.1 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1524,'2024-11-08','15:51:01','90.00000','105.00000','Purified water','pH','PW/SP36','E4RM24006024','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 16:28:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.20 pH\r\nmV                   93.1 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1525,'2024-11-08','15:52:55','90.00000','105.00000','Purified water','pH','PW/SP25','E4RM24006025','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 16:28:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.29 pH\r\nmV                   87.5 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1526,'2024-11-08','15:55:12','90.00000','105.00000','Water For Injection','pH','WFI/SP26','E4RM24006026','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 16:29:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.34 pH\r\nmV                   84.8 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1527,'2024-11-08','17:14:29','90.00000','105.00000','Piperacillin and Tazobactam for Injection','RS','PT0424074A','E4STFP24000670','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','PIPERTAZO',1,'86253','D. Jayapal Reddy','2024-11-09 13:34:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   3.82 pH\r\nmV                   177.7 mV\r\nTemperature          29.1 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1528,'2024-11-08','18:25:38','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24020','E4SO24000813','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','SCDBIH',1,'85152','V. Chandra Shekara Achari','2024-11-08 21:32:19','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.48 pH\r\nmV                   -28.9 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1529,'2024-11-08','18:29:37','90.00000','105.00000','Macckankey broth','NA','RS013/24009','E4SO24000814','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','NA',1,'85152','V. Chandra Shekara Achari','2024-11-08 21:32:33','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.34 pH\r\nmV                   -20.6 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1530,'2024-11-09','07:02:43','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005869','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','1.68 pH',1,'86253','D. Jayapal Reddy','2024-11-09 07:28:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   297.6 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1531,'2024-11-09','07:04:54','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005869','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','4.01pH',1,'86253','D. Jayapal Reddy','2024-11-09 07:29:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   159.2 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1532,'2024-11-09','07:07:36','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005869','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','7.00pH',1,'86253','D. Jayapal Reddy','2024-11-09 07:29:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -16.5 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1533,'2024-11-09','07:10:21','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005869','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','10.01 pH',1,'86253','D. Jayapal Reddy','2024-11-09 07:29:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.98 pH\r\nmV                   -189.5 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1534,'2024-11-09','07:12:52','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005869','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','12.46 pH',1,'86253','D. Jayapal Reddy','2024-11-09 07:29:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.47 pH\r\nmV                   -335.9 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1535,'2024-11-09','08:13:55','90.00000','105.00000','Piperacillin and Tazobactam for Injection','Assay pH','2449203350','E4RM24006009','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','Mobile phase pH',1,'86253','D. Jayapal Reddy','2024-11-09 12:27:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   74.0 mV\r\nTemperature          31.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1536,'2024-11-09','11:16:46','90.00000','105.00000','Ampicillin and Sulbactam','RS BY HPLC MP A','AS0324081','E4FP24000736','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','87697','M. Raghavendr Reddy',2,0,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-11-09 12:27:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.03 pH\r\nmV                   101.7 mV\r\nTemperature          28.9 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\nLog #                914\r\n\r\n\r\n'),
(1537,'2024-11-09','11:31:06','90.00000','105.00000','Ampicillin and Sulbactam','UOD','AS0324083','E4FP24000740','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205232','K. Narender Reddy',2,0,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-11-09 12:27:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.99 pH\r\nmV                   105.5 mV\r\nTemperature          32.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1538,'2024-11-09','11:40:25','90.00000','105.00000','Piperacillin and Tazobactam for Injection','RS','2449203350','E4RM24006009','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205189','P. Ravi Kumar',2,1,101,'pH','Mobile phase pH',1,'86253','D. Jayapal Reddy','2024-11-09 12:28:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.02 pH\r\nmV                   43.3 mV\r\nTemperature          28.0 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\nLog #                916\r\n\r\n\r\n'),
(1539,'2024-11-09','11:53:23','90.00000','105.00000','Ampicillin and Sulbactam for injection','Identification','2449101498','E4RM24005952','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,0,101,'pH','Ampi&Sulbactam',1,'86253','D. Jayapal Reddy','2024-11-09 12:28:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.02 pH\r\nmV                   103.5 mV\r\nTemperature          31.4 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1540,'2024-11-09','11:57:25','90.00000','105.00000','TAZOBACTAM','NA','TZ0024105','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','92936','Sameer',2,0,104,'pH','1',1,'95681','T. Sivanarayana','2024-11-09 14:00:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.69 pH\r\nmV                   117.6 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             09-11-24,10:25\r\nLog #                277\r\n\r\n\r\n'),
(1541,'2024-11-09','12:28:12','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324082','E4FP24000737','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-11-09 13:42:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.11 pH\r\nmV                   -139.3 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1542,'2024-11-09','12:32:40','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324083','E4FP24000740','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-11-09 13:43:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.07 pH\r\nmV                   -136.6 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1543,'2024-11-09','12:37:18','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324084','E4FP24000742','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-11-09 13:43:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.04 pH\r\nmV                   -134.7 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1544,'2024-11-09','12:45:01','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TT0424013','E4FP24000738','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-11-09 13:43:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.98 pH\r\nmV                   44.9 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1545,'2024-11-09','12:48:46','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TT0424014','E4FP24000741','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-11-09 13:44:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.97 pH\r\nmV                   45.5 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1546,'2024-11-09','13:23:03','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0222004B','E4STFP24000694','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'86253','D. Jayapal Reddy','2024-11-09 14:06:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.86 pH\r\nmV                   52.0 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1547,'2024-11-09','13:27:02','90.00000','105.00000','Nafcillin For Injection','pH','NF1024006A','E4STFP24000695','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','pH_Test',1,'86253','D. Jayapal Reddy','2024-11-09 14:06:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.37 pH\r\nmV                   -36.8 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1548,'2024-11-09','13:39:12','90.00000','105.00000','Flucloxacillin powder for solution for injection','pH','FL0122001A','E4STFP24000686','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','ph Flucloxacillin',1,'86253','D. Jayapal Reddy','2024-11-09 14:07:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.63 pH\r\nmV                   65.3 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1549,'2024-11-09','13:42:17','90.00000','105.00000','Flucloxacillin powder for solution for injection','pH','FL0124001B','E4STFP24000692','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','ph Flucloxacillin',1,'86253','D. Jayapal Reddy','2024-11-09 14:07:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.66 pH\r\nmV                   63.3 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1550,'2024-11-09','13:46:30','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24006048','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-09 14:25:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.61 pH\r\nmV                   66.2 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1551,'2024-11-09','13:48:50','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24006049','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-09 14:25:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.55 pH\r\nmV                   69.7 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1552,'2024-11-09','13:51:30','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24006050','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-09 14:25:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   72.8 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1553,'2024-11-09','13:54:04','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24006051','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-09 14:26:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   72.3 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1554,'2024-11-09','13:57:04','90.00000','105.00000','Water For Injection','pH','WFI/SP34','E4RM24006052','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-09 14:27:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.44 pH\r\nmV                   76.2 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1555,'2024-11-09','13:59:58','90.00000','105.00000','Purified water','pH','PW/SP26','E4RM24006053','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-09 14:28:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.57 pH\r\nmV                   68.8 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1556,'2024-11-09','14:02:46','90.00000','105.00000','Water For Injection','pH','WFI/SP27','E4RM24006054','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-09 14:28:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   73.0 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1557,'2024-11-09','15:29:56','90.00000','105.00000','Ampicillin and Sulbactam','UOD TEST','AS0324084','E4FP24000742','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,0,101,'pH','AS',1,'82735','M.D. Babar Qhadri','2024-11-11 14:51:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.01 pH\r\nmV                   104.9 mV\r\nTemperature          32.4 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1558,'2024-11-09','17:18:50','90.00000','105.00000','PIPERACILLIN','NA','TZ0024105','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','2',1,'88515','Prashanth B','2024-11-09 21:14:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.91 pH\r\nmV                   46.1 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             09-11-24,10:25\r\n\r\n\r\n'),
(1559,'2024-11-09','18:00:18','90.00000','105.00000','PIPERACILLIN','NA','TZ0024105','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','2',1,'88515','Prashanth B','2024-11-09 21:15:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.92 pH\r\nmV                   45.0 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             09-11-24,10:25\r\n\r\n\r\n'),
(1560,'2024-11-09','18:43:06','90.00000','105.00000','PIPERACILLIN','NA','TZ0024105','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','2',1,'88515','Prashanth B','2024-11-09 21:15:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.20 pH\r\nmV                   29.0 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             09-11-24,10:25\r\n\r\n\r\n'),
(1561,'2024-11-09','19:25:01','90.00000','105.00000','PIPERACILLIN','NA','TZ0024105','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','2',1,'88515','Prashanth B','2024-11-09 21:16:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.37 pH\r\nmV                   19.2 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             09-11-24,10:25\r\n\r\n\r\n'),
(1562,'2024-11-09','21:22:16','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024105','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','3',1,'88515','Prashanth B','2024-11-09 21:39:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.28 pH\r\nmV                   24.0 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             09-11-24,10:25\r\n\r\n\r\n'),
(1563,'2024-11-09','21:35:02','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TT0024105','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','3',1,'88515','Prashanth B','2024-11-09 21:40:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.31 pH\r\nmV                   22.7 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             09-11-24,10:25\r\n\r\n\r\n'),
(1564,'2024-11-10','07:51:07','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005891','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-11-10 08:29:34','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   301.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             10-11-24,07:35\r\n\r\n\r\n'),
(1565,'2024-11-10','07:55:19','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005891','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-11-10 08:29:46','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   163.6 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             10-11-24,07:35\r\n\r\n\r\n'),
(1566,'2024-11-10','08:00:30','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005891','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-11-10 08:29:56','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -12.5 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             10-11-24,07:35\r\n\r\n\r\n'),
(1567,'2024-11-10','08:10:15','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005891','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-11-10 08:30:06','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.04 pH\r\nmV                   -188.4 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             10-11-24,07:35\r\n\r\n\r\n'),
(1568,'2024-11-10','08:15:35','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005891','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-11-10 08:30:17','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.49 pH\r\nmV                   -332.4 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             10-11-24,07:35\r\n\r\n\r\n'),
(1569,'2024-11-10','12:12:55','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TZ0024105','E4IP24001334','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','pH (Bulk)',1,'82735','M.D. Babar Qhadri','2024-11-10 12:30:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.05 pH\r\nmV                   44.9 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             10-11-24,07:35\r\n\r\n\r\n'),
(1570,'2024-11-10','21:12:54','90.00000','105.00000','TAZOBACTAM','NA','TZ0024106','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','1',1,'88629','Raghuram Reddy','2024-11-10 21:33:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.40 pH\r\nmV                   132.8 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             10-11-24,10:18\r\n\r\n\r\n'),
(1571,'2024-11-11','00:37:07','90.00000','105.00000','PIPERACILLIN','NA','TZ0024106','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-11 04:39:05','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.90 pH\r\nmV                   45.8 mV\r\nTemperature          26.9 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-11-24,00:18\r\n\r\n\r\n'),
(1572,'2024-11-11','01:19:07','90.00000','105.00000','PIPERACILLIN','NA','TZ0024106','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-11 04:47:12','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.89 pH\r\nmV                   44.0 mV\r\nTemperature          23.1 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-11-24,00:18\r\n\r\n\r\n'),
(1573,'2024-11-11','01:21:03','90.00000','105.00000','PIPERACILLIN','NA','TZ0024106','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-11 04:47:31','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.91 pH\r\nmV                   44.2 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-11-24,00:18\r\n\r\n\r\n'),
(1574,'2024-11-11','02:04:15','90.00000','105.00000','PIPERACILLIN','NA','TZ0024106','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-11 04:47:52','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.22 pH\r\nmV                   26.5 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-11-24,00:18\r\n\r\n\r\n'),
(1575,'2024-11-11','02:47:43','90.00000','105.00000','PIPERACILLIN','NA','TZ0024106','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-11 04:48:15','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.42 pH\r\nmV                   14.8 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-11-24,00:18\r\n\r\n\r\n'),
(1576,'2024-11-11','05:07:52','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024106','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','3',1,'84472','G. Ashok','2024-11-11 06:12:11','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.22 pH\r\nmV                   26.2 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-11-24,00:18\r\n\r\n\r\n'),
(1577,'2024-11-11','05:27:31','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024106','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','3',1,'84472','G. Ashok','2024-11-11 06:12:28','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.23 pH\r\nmV                   25.8 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-11-24,00:18\r\n\r\n\r\n'),
(1578,'2024-11-11','06:53:00','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005900','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','1.68 pH',1,'86253','D. Jayapal Reddy','2024-11-11 07:28:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   302.3 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1579,'2024-11-11','06:57:23','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005900','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','4.01pH',1,'86253','D. Jayapal Reddy','2024-11-11 07:29:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   164.1 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1580,'2024-11-11','07:01:18','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005900','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','7.00pH',1,'86253','D. Jayapal Reddy','2024-11-11 07:30:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -11.8 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1581,'2024-11-11','07:08:00','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005900','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','10.01 pH',1,'86253','D. Jayapal Reddy','2024-11-11 07:30:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.06 pH\r\nmV                   -188.1 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1582,'2024-11-11','07:11:55','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005900','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','12.46 pH',1,'86253','D. Jayapal Reddy','2024-11-11 07:31:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.50 pH\r\nmV                   -331.4 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1583,'2024-11-11','10:18:10','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005905','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-11-11 12:43:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   309.1 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-11-24,10:14\r\n\r\n\r\n'),
(1584,'2024-11-11','10:20:14','90.00000','105.00000','Piperacillin and Tazobactam for Injection','Assay pH','2449203357','E4RM24006016','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','Mobile phase pH',1,'90018','Ch. Satyam','2024-11-11 12:21:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   79.6 mV\r\nTemperature          32.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1585,'2024-11-11','10:20:54','90.00000','105.00000','pH Buffer','NA','	NA','E4INS24005905','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-11-11 12:45:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   171.0 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-11-24,10:14\r\nLog #                398\r\n\r\n\r\n'),
(1586,'2024-11-11','10:45:41','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005905','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-11-11 12:47:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -6.3 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-11-24,10:14\r\n\r\n\r\n'),
(1587,'2024-11-11','10:48:09','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005905','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-11-11 12:49:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   9.99 pH\r\nmV                   -179.8 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-11-24,10:14\r\n\r\n\r\n'),
(1588,'2024-11-11','10:49:54','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005905','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-11-11 12:52:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.49 pH\r\nmV                   -327.0 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-11-24,10:14\r\n\r\n\r\n'),
(1589,'2024-11-11','11:53:41','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS','TT0424013','E4FP24000738','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205232','K. Narender Reddy',2,0,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-11 14:45:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.99 pH\r\nmV                   50.1 mV\r\nTemperature          27.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1590,'2024-11-11','12:25:17','90.00000','105.00000','Piperacillin and Tazobactam for Injection','TZ0024106','NA','E4IP24001339','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','pH (Bulk)',1,'90018','Ch. Satyam','2024-11-11 12:35:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.48 pH\r\nmV                   20.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1591,'2024-11-11','13:34:11','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24006078','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,2,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-11-11 14:42:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.62 pH\r\nmV                   70.8 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1592,'2024-11-11','13:37:11','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24006079','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,2,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-11-11 14:42:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.60 pH\r\nmV                   72.3 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1593,'2024-11-11','13:50:07','90.00000','105.00000','Tazobactam USP','Assay','A2403A020','E4AQR24000015','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205075','G. Shyam Kumar',2,0,101,'pH','Assay',1,'82735','M.D. Babar Qhadri','2024-11-11 14:47:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   80.9 mV\r\nTemperature          36.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1594,'2024-11-11','13:54:17','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24006080','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,2,101,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-11-11 14:42:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.64 pH\r\nmV                   69.9 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1595,'2024-11-11','13:57:13','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24006081','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,2,101,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-11-11 14:43:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   66.6 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1596,'2024-11-11','14:00:38','90.00000','105.00000','Purified water','pH','PW/SP03','E4RM24006082','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,2,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-11-11 14:43:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.68 pH\r\nmV                   67.1 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1597,'2024-11-11','14:03:26','90.00000','105.00000','Pure steam','pH','PSC/SP09','E4RM24006083','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,2,101,'pH','pH_Pure Steam',1,'82735','M.D. Babar Qhadri','2024-11-11 14:43:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   67.0 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1598,'2024-11-11','14:06:19','90.00000','105.00000','Purified water','pH','PW/SP21','E4RM24006084','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,2,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-11-11 14:43:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.63 pH\r\nmV                   70.2 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1599,'2024-11-11','14:09:22','90.00000','105.00000','Water For Injection','pH','WFI/SP22','E4RM24006085','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,2,101,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-11-11 14:43:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.66 pH\r\nmV                   68.5 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1600,'2024-11-11','14:13:31','90.00000','105.00000','Pure steam','pH','PSC/SP01','E4RM24006086','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,2,101,'pH','pH_Pure Steam',1,'82735','M.D. Babar Qhadri','2024-11-11 14:44:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.60 pH\r\nmV                   72.0 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1601,'2024-11-11','15:33:29','90.00000','105.00000','Piperacillin and Tazobactam For injection','UOD TEST','PT0424122','E4FP24000743','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-11 20:28:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   80.3 mV\r\nTemperature          32.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1602,'2024-11-11','15:52:13','90.00000','105.00000','Piperacillin and Tazobactam For injection','ASSAY','PT0424120','E4FP24000734','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-11 20:29:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   79.1 mV\r\nTemperature          32.3 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1603,'2024-11-11','16:41:34','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424125','E4PP24002024','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-11 20:29:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.00 pH\r\nmV                   48.7 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1604,'2024-11-11','16:48:04','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424125','E4PP24002025','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-11 20:30:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.97 pH\r\nmV                   50.9 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1605,'2024-11-11','16:52:51','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424125','E4PP24002026','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-11 20:30:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.94 pH\r\nmV                   52.3 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1606,'2024-11-11','16:57:16','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424125','E4PP24002043','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-11 20:31:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.99 pH\r\nmV                   49.3 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1607,'2024-11-11','16:58:56','90.00000','105.00000','Peptone','NA','RS007/24047','E4SO24000824','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-11-11 17:06:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.95 pH\r\nmV                   -5.3 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-11-24,10:14\r\nLog #                402\r\n\r\n\r\n'),
(1608,'2024-11-11','17:00:42','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424125','E4PP24002044','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-11 20:31:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.04 pH\r\nmV                   46.3 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\nLog #                965\r\n\r\n\r\n'),
(1609,'2024-11-11','17:14:39','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PR0424001A','E4STFP24000700','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-11-11 17:40:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.78 pH\r\nmV                   61.6 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1610,'2024-11-11','17:18:46','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TI0323005A','E4STFP24000701','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-11-11 17:40:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.64 pH\r\nmV                   69.9 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1611,'2024-11-11','17:21:58','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TI0323004A','E4STFP24000702','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-11-11 17:40:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   76.4 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1612,'2024-11-11','18:50:57','90.00000','105.00000','XLDA','NA','RS029/24008','E4SO24000826','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','XLDA',1,'205143','Ch. Prasad','2024-11-12 15:08:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.36 pH\r\nmV                   -24.7 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-11-24,10:14\r\n\r\n\r\n'),
(1613,'2024-11-11','21:35:39','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TT0424015','E4FP24000746','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-11 21:59:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.87 pH\r\nmV                   56.2 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1614,'2024-11-11','21:45:02','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424122','E4FP24000743','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-11 22:00:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.94 pH\r\nmV                   52.1 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\nLog #                970\r\n\r\n\r\n'),
(1615,'2024-11-11','21:47:48','90.00000','105.00000','Peptone','NA','RS007/24047','E4SO24000824','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-11-12 15:08:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.98 pH\r\nmV                   -2.7 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-11-24,10:14\r\nLog #                404\r\n\r\n\r\n'),
(1616,'2024-11-11','21:52:43','90.00000','105.00000','XLDA','NA','RS029/24008','E4SO24000826','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','XLDA',1,'205143','Ch. Prasad','2024-11-12 15:09:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.41 pH\r\nmV                   -27.4 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-11-24,10:14\r\n\r\n\r\n'),
(1617,'2024-11-12','07:12:03','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005916','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','1.68 pH',1,'86253','D. Jayapal Reddy','2024-11-12 07:35:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.65 pH\r\nmV                   302.6 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1618,'2024-11-12','07:15:46','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005916','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','4.01pH',1,'86253','D. Jayapal Reddy','2024-11-12 07:35:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   164.3 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1619,'2024-11-12','07:19:42','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005916','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','7.00pH',1,'86253','D. Jayapal Reddy','2024-11-12 07:36:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -12.5 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1620,'2024-11-12','07:23:00','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005916','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','10.01 pH',1,'86253','D. Jayapal Reddy','2024-11-12 07:36:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.03 pH\r\nmV                   -187.8 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1621,'2024-11-12','07:26:43','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005916','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','12.46 pH',1,'86253','D. Jayapal Reddy','2024-11-12 07:36:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.47 pH\r\nmV                   -330.4 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1622,'2024-11-12','08:57:42','90.00000','105.00000','Nafcillin for Injection','RS BY HPLC MP A','NF0224001A','E4STFP24000684','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','87697','M. Raghavendr Reddy',2,0,101,'pH','Nafcillin pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-12 11:26:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   3.50 pH\r\nmV                   198.7 mV\r\nTemperature          30.9 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\nLog #                976\r\n\r\n\r\n'),
(1623,'2024-11-12','09:25:22','90.00000','105.00000','TAZOBACTAM','NA','TZ0024107','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','1',1,'88629','Raghuram Reddy','2024-11-12 11:40:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.56 pH\r\nmV                   122.9 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             12-11-24,08:24\r\n\r\n\r\n'),
(1624,'2024-11-12','10:25:05','90.00000','105.00000','Nafcillin For Injection','ASSAY MP_PH','NF0224001A','E4STFP24000684','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','Assay Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-12 11:28:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.52 pH\r\nmV                   -41.7 mV\r\nTemperature          32.7 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1625,'2024-11-12','10:33:12','90.00000','105.00000','Nafcillin For Injection','ASSAY_DILUENT_PH','NF0224001A','E4STFP24000684','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','Assay Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-12 11:28:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -12.0 mV\r\nTemperature          31.5 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1626,'2024-11-12','11:16:37','90.00000','105.00000','Piperacillin and Tazobactam for Injection','RS','PT0424034A','E4STFP24000696','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205043','T.V. Subbaiah',2,0,101,'pH','PIPERTAZO',1,'205208','S. Siddhartha Kumar Reddy','2024-11-12 11:29:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.05 pH\r\nmV                   45.8 mV\r\nTemperature          28.4 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1627,'2024-11-12','12:19:39','90.00000','105.00000','Piperacillin And Tazobactam','pH','2449203350','E4RM24006009','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,2,101,'pH','PT',1,'205208','S. Siddhartha Kumar Reddy','2024-11-12 13:51:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.14 pH\r\nmV                   39.7 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1628,'2024-11-12','12:24:18','90.00000','105.00000','Piperacillin And Tazobactam','pH','2449203354','E4RM24006011','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','PT',1,'205208','S. Siddhartha Kumar Reddy','2024-11-12 13:51:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.93 pH\r\nmV                   52.4 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1629,'2024-11-12','12:27:52','90.00000','105.00000','Piperacillin And Tazobactam','pH','2449203355','E4RM24006013','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','PT',1,'205208','S. Siddhartha Kumar Reddy','2024-11-12 13:52:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.93 pH\r\nmV                   52.3 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1630,'2024-11-12','12:31:27','90.00000','105.00000','Piperacillin And Tazobactam','pH','2449203356','E4RM24006014','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','PT',1,'205208','S. Siddhartha Kumar Reddy','2024-11-12 13:52:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.99 pH\r\nmV                   49.0 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1631,'2024-11-12','13:25:42','90.00000','105.00000','PIPERACILLIN','NA','TZ0024107','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','92936','Sameer',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-11-12 14:22:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.86 pH\r\nmV                   46.8 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             12-11-24,08:24\r\n\r\n\r\n'),
(1632,'2024-11-12','13:55:20','90.00000','105.00000','Piperacillin And Tazobactam','pH','2449203357','E4RM24006016','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','PT',1,'205208','S. Siddhartha Kumar Reddy','2024-11-12 14:48:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.07 pH\r\nmV                   43.8 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\nLog #                984\r\n\r\n\r\n'),
(1633,'2024-11-12','14:00:28','90.00000','105.00000','Piperacillin And Tazobactam','pH','2449203007','E4RM24006027','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','PT',1,'205208','S. Siddhartha Kumar Reddy','2024-11-12 14:49:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.08 pH\r\nmV                   43.5 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1634,'2024-11-12','14:03:37','90.00000','105.00000','Piperacillin And Tazobactam','pH','2449203008','E4RM24006028','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','PT',1,'205208','S. Siddhartha Kumar Reddy','2024-11-12 14:50:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.24 pH\r\nmV                   33.7 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\nLog #                986\r\n\r\n\r\n'),
(1635,'2024-11-12','14:09:24','90.00000','105.00000','Piperacillin And Tazobactam','pH','2449203161','E4RM24006029','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','PT',1,'205208','S. Siddhartha Kumar Reddy','2024-11-12 14:51:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.12 pH\r\nmV                   41.1 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1636,'2024-11-12','14:09:28','90.00000','105.00000','PIPERACILLIN','NA','TZ0024107','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','92936','Sameer',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-11-12 14:23:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.88 pH\r\nmV                   45.8 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             12-11-24,08:24\r\nLog #                294\r\n\r\n\r\n'),
(1637,'2024-11-12','14:22:19','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005931','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-11-12 15:10:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.67 pH\r\nmV                   309.3 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             12-11-24,14:20\r\n\r\n\r\n'),
(1638,'2024-11-12','14:23:55','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005931','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-11-12 15:27:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.04 pH\r\nmV                   170.6 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             12-11-24,14:20\r\n\r\n\r\n'),
(1639,'2024-11-12','14:32:01','90.00000','105.00000','Tazobactam USP','Assay','A2403A020','E4AQ24000063','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205269','P. Srinivas varma',2,1,101,'pH','Assay',1,'205208','S. Siddhartha Kumar Reddy','2024-11-12 14:52:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   80.7 mV\r\nTemperature          36.9 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1640,'2024-11-12','14:40:49','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005931','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-11-12 15:27:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -5.6 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             12-11-24,14:20\r\n\r\n\r\n'),
(1641,'2024-11-12','14:39:36','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24006098','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-11-12 16:15:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.65 pH\r\nmV                   68.1 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\nLog #                989\r\n\r\n\r\n'),
(1642,'2024-11-12','14:43:43','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005931','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-11-12 15:27:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.02 pH\r\nmV                   -179.6 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             12-11-24,14:20\r\n\r\n\r\n'),
(1643,'2024-11-12','14:45:11','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005931','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-11-12 15:27:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.48 pH\r\nmV                   -325.5 mV\r\nTemperature          26.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             12-11-24,14:20\r\nLog #                410\r\n\r\n\r\n'),
(1644,'2024-11-12','14:45:22','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24006099','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-11-12 16:16:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.65 pH\r\nmV                   68.6 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\nLog #                990\r\n\r\n\r\n'),
(1645,'2024-11-12','14:52:02','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24006100','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-11-12 16:16:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   75.5 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1646,'2024-11-12','14:54:55','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24006101','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-11-12 16:17:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.49 pH\r\nmV                   77.6 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1647,'2024-11-12','14:58:17','90.00000','105.00000','PIPERACILLIN','NA','TZ0024107','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-11-12 18:12:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.19 pH\r\nmV                   27.5 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             12-11-24,08:24\r\n\r\n\r\n'),
(1648,'2024-11-12','15:01:47','90.00000','105.00000','Water For Injection','pH','WFI/SP13','E4RM24006102','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-11-12 16:17:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.62 pH\r\nmV                   70.2 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\nLog #                993\r\n\r\n\r\n'),
(1649,'2024-11-12','15:06:55','90.00000','105.00000','Pure steam','pH','PSC/SP10','E4RM24006103','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'84378','Shivanand. W','2024-11-12 16:18:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   75.9 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1650,'2024-11-12','15:11:30','90.00000','105.00000','Purified water','pH','PW/SP22','E4RM24006104','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-11-12 16:18:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.57 pH\r\nmV                   73.3 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1651,'2024-11-12','15:15:05','90.00000','105.00000','Purified water','pH','PW/SP49','E4RM24006105','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-11-12 16:19:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.58 pH\r\nmV                   72.4 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1652,'2024-11-12','15:18:40','90.00000','105.00000','Water For Injection','pH','WFI/SP23','E4RM24006106','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-11-12 16:19:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   76.1 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1653,'2024-11-12','15:21:25','90.00000','105.00000','Pure steam','pH','PSC/SP03','E4RM24006107','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'84378','Shivanand. W','2024-11-12 16:19:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   77.2 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1654,'2024-11-12','15:39:36','90.00000','105.00000','PIPERACILLIN','NA','TZ0024107','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-11-12 18:12:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.42 pH\r\nmV                   13.4 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             12-11-24,08:24\r\n\r\n\r\n'),
(1655,'2024-11-12','16:05:26','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424125 (LOT-A)','E4FP24000747','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'84378','Shivanand. W','2024-11-12 16:44:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.00 pH\r\nmV                   48.3 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1656,'2024-11-12','16:10:56','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424125(LOT-B)','E4FP24000748','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'84378','Shivanand. W','2024-11-12 16:45:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.95 pH\r\nmV                   50.9 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1657,'2024-11-12','16:36:56','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424123','E4FP24000749','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'84378','Shivanand. W','2024-11-12 16:44:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.25 pH\r\nmV                   33.4 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             12-11-24,07:08\r\n\r\n\r\n'),
(1658,'2024-11-12','17:06:56','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24021','E4SO24000830','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDBIH',1,'89384','M.V.V. Satish','2024-11-12 17:49:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.38 pH\r\nmV                   -25.5 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             12-11-24,14:20\r\n\r\n\r\n'),
(1659,'2024-11-12','18:06:21','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024107','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','3',1,'95681','T. Sivanarayana','2024-11-12 18:13:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.24 pH\r\nmV                   24.1 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             12-11-24,08:24\r\n\r\n\r\n'),
(1660,'2024-11-12','18:31:30','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024107','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','3',1,'95681','T. Sivanarayana','2024-11-12 19:04:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.22 pH\r\nmV                   25.4 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             12-11-24,08:24\r\n\r\n\r\n'),
(1661,'2024-11-12','21:05:55','90.00000','105.00000','TAZOBACTAM','NA','TR0024004','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-003','CW1-12031','1','NULL','88332','P.V.A.K. Raju',2,0,105,'pH','1',1,'91406','Sridhar','2024-11-12 21:13:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   4.45 pH\r\nmV                   149.9 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        96.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             12-11-24,09:46\r\n\r\n\r\n'),
(1662,'2024-11-12','21:26:56','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24021','E4SO24000830','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDBIH',1,'89384','M.V.V. Satish','2024-11-13 08:51:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.37 pH\r\nmV                   -25.0 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             12-11-24,14:20\r\n\r\n\r\n'),
(1663,'2024-11-12','22:09:07','90.00000','105.00000','PIPERACILLIN','NA','TR0024004','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-003','CW1-12031','1','NULL','88332','P.V.A.K. Raju',2,0,105,'pH','2',1,'91406','Sridhar','2024-11-12 22:18:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   6.48 pH\r\nmV                   34.7 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        96.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             12-11-24,09:46\r\n\r\n\r\n'),
(1664,'2024-11-13','01:17:42','90.00000','105.00000','Piperacillin and Tazobactam for Injection','NA','TR0024004','NA','BULK','PR-PHMTZ-003','CW1-12031','1','NULL','85705','P. Prakash Rao',2,0,105,'pH','pH (Bulk)',1,'84472','G. Ashok','2024-11-13 01:47:10','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   6.33 pH\r\nmV                   43.0 mV\r\nTemperature          26.5 C (ATC)\r\nAverage Slope        95.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             13-11-24,00:58\r\n\r\n\r\n'),
(1665,'2024-11-13','02:35:43','90.00000','105.00000','Piperacillin and Tazobactam for Injection','NA','TR0024004','NA','BULK','PR-PHMTZ-003','CW1-12031','1','NULL','85705','P. Prakash Rao',2,0,105,'pH','pH (Bulk)',1,'84472','G. Ashok','2024-11-13 05:56:24','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   6.35 pH\r\nmV                   41.5 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        95.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             13-11-24,00:58\r\n\r\n\r\n'),
(1666,'2024-11-13','07:06:19','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005938','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-11-13 07:22:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.68 pH\r\nmV                   302.7 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1667,'2024-11-13','07:08:23','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005938','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-11-13 07:22:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   164.7 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1668,'2024-11-13','07:10:19','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005938','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-11-13 07:22:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -11.2 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1669,'2024-11-13','07:13:27','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005938','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-11-13 07:22:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.99 pH\r\nmV                   -187.1 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1670,'2024-11-13','07:16:39','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005938','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-11-13 07:22:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.48 pH\r\nmV                   -334.4 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1671,'2024-11-13','08:09:08','90.00000','105.00000','Ampicillin and Sulbactam','UOD TEST Buffer pH','AS0324085','E4FP24000744','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','AS',1,'82735','M.D. Babar Qhadri','2024-11-13 12:01:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.99 pH\r\nmV                   110.2 mV\r\nTemperature          32.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1672,'2024-11-13','09:04:03','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005940','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','1.68 pH',1,'88842','R. Mallesh','2024-11-13 14:29:03','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.64 pH\r\nmV                   303.9 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             13-11-24,08:30\r\n\r\n\r\n'),
(1673,'2024-11-13','09:07:56','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005940','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','4.01pH',1,'88842','R. Mallesh','2024-11-13 14:30:29','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.00 pH\r\nmV                   166.6 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             13-11-24,08:30\r\n\r\n\r\n'),
(1674,'2024-11-13','09:10:16','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005940','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','7.00pH',1,'88842','R. Mallesh','2024-11-13 14:31:58','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.97 pH\r\nmV                   -5.8 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             13-11-24,08:30\r\n\r\n\r\n'),
(1675,'2024-11-13','09:15:52','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005940','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','10.01 pH',1,'88842','R. Mallesh','2024-11-13 14:33:59','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.02 pH\r\nmV                   -182.5 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             13-11-24,08:30\r\n\r\n\r\n'),
(1676,'2024-11-13','09:32:09','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005940','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','12.46 pH',1,'88842','R. Mallesh','2024-11-13 14:34:41','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.47 pH\r\nmV                   -325.7 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             13-11-24,08:30\r\nLog #                417\r\n\r\n\r\n'),
(1677,'2024-11-13','09:53:14','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TZ0024107','E4IP24001342','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','pH (Bulk)',1,'84378','Shivanand. W','2024-11-13 10:45:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.07 pH\r\nmV                   43.8 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1678,'2024-11-13','10:01:27','90.00000','105.00000','Piperacillin and Tazobactam For injection','UOD','PT0424123','E4FP24000749','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','90017','G. Srishailam',2,0,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-13 12:01:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   79.3 mV\r\nTemperature          31.1 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1679,'2024-11-13','10:09:20','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024004','E4PP24002057','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'84378','Shivanand. W','2024-11-13 10:44:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.59 pH\r\nmV                   71.9 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1680,'2024-11-13','10:13:05','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024004','E4PP24002054','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'84378','Shivanand. W','2024-11-13 10:47:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.44 pH\r\nmV                   139.8 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1681,'2024-11-13','10:17:29','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024004','E4PP24002056','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'84378','Shivanand. W','2024-11-13 10:44:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.18 pH\r\nmV                   37.3 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1682,'2024-11-13','10:32:06','90.00000','105.00000','Ampicillin and Sulbactam','UOD','AS0324086','E4FP24000745','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205075','G. Shyam Kumar',2,1,101,'pH','AS',1,'82735','M.D. Babar Qhadri','2024-11-13 12:02:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.01 pH\r\nmV                   109.1 mV\r\nTemperature          31.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1683,'2024-11-13','10:47:08','90.00000','105.00000','PIPERACILLIN MONOHYDRATE','RS MOBILE PHASE A','C66240901','E4RM24006055','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205189','P. Ravi Kumar',2,1,101,'pH','RS Mobile phase',1,'82735','M.D. Babar Qhadri','2024-11-13 12:05:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   77.5 mV\r\nTemperature          33.1 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1684,'2024-11-13','10:53:31','90.00000','105.00000','PIPERACILLIN MONOHYDRATE','RS MOBILE PHASE B','C66240901','E4RM24006055','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205189','P. Ravi Kumar',2,1,101,'pH','RS Mobile phase',1,'82735','M.D. Babar Qhadri','2024-11-13 12:05:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   79.3 mV\r\nTemperature          32.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1685,'2024-11-13','11:03:05','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','ASSAY','TR0024004','E4PP24002058','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,0,101,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-11-13 12:06:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   78.4 mV\r\nTemperature          31.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1686,'2024-11-13','11:25:06','90.00000','105.00000','Ampicillin and Sulbactam','UOD SST pH','AS0324085','E4FP24000744','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','AS',1,'82735','M.D. Babar Qhadri','2024-11-13 12:03:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.00 pH\r\nmV                   106.9 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1687,'2024-11-13','11:43:53','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TT0424016','E4FP24000750','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-13 12:03:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.75 pH\r\nmV                   62.6 mV\r\nTemperature          23.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1688,'2024-11-13','12:34:42','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024004','E4PP24002059','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-11-13 13:38:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.59 pH\r\nmV                   72.4 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1689,'2024-11-13','12:38:21','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024004','E4PP24002060','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-11-13 13:38:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.57 pH\r\nmV                   73.4 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1690,'2024-11-13','12:42:00','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024004','E4PP24002062','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-11-13 13:39:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.58 pH\r\nmV                   73.0 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1691,'2024-11-13','14:00:23','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324085','E4FP24000744','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-11-13 14:54:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.21 pH\r\nmV                   -141.6 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1692,'2024-11-13','14:04:57','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324086','E4FP24000745','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-11-13 14:54:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.19 pH\r\nmV                   -140.1 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1693,'2024-11-13','15:12:32','90.00000','105.00000','Glass Vials for pH','pH(Vial-1)','NA','E4EQ24000399','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-11-13 15:35:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.68 pH\r\nmV                   242.7 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1694,'2024-11-13','15:15:05','90.00000','105.00000','Glass Vials for pH','pH(Vial-2)','NA','E4EQ24000399','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-11-13 15:35:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.74 pH\r\nmV                   239.3 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1695,'2024-11-13','15:48:32','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0324001','E4PP24002073','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,3,101,'pH','PIPERTAZ',1,'86253','D. Jayapal Reddy','2024-11-13 16:20:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.49 pH\r\nmV                   78.3 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1696,'2024-11-13','15:52:15','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0324001','E4PP24002074','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,2,101,'pH','PIPERTAZ',1,'86253','D. Jayapal Reddy','2024-11-13 16:21:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.57 pH\r\nmV                   73.4 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1697,'2024-11-13','15:55:35','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0324001','E4PP24002075','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,2,101,'pH','PIPERTAZ',1,'86253','D. Jayapal Reddy','2024-11-13 16:21:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.59 pH\r\nmV                   72.1 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1698,'2024-11-13','16:21:13','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24006116','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-13 17:16:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.67 pH\r\nmV                   67.6 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1699,'2024-11-13','16:24:58','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24006117','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-13 17:16:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   66.2 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1700,'2024-11-13','16:29:26','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24006118','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-13 17:16:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.71 pH\r\nmV                   65.0 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1701,'2024-11-13','16:34:03','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24006119','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-13 17:17:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.67 pH\r\nmV                   67.4 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1702,'2024-11-13','16:37:31','90.00000','105.00000','Purified water','pH','PW/SP05','E4RM24006120','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-13 17:17:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.64 pH\r\nmV                   69.1 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1703,'2024-11-13','16:41:13','90.00000','105.00000','Pure steam','pH','PSC/SP20','E4RM24006121','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'86253','D. Jayapal Reddy','2024-11-13 17:17:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.66 pH\r\nmV                   68.0 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1704,'2024-11-13','16:44:24','90.00000','105.00000','Purified water','pH','PW/SP39','E4RM24006122','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-13 17:18:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.61 pH\r\nmV                   71.0 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1705,'2024-11-13','16:47:06','90.00000','105.00000','Pure steam','pH','PSC/SP17','E4RM24006123','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'86253','D. Jayapal Reddy','2024-11-13 17:18:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.60 pH\r\nmV                   71.6 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1706,'2024-11-13','16:50:31','90.00000','105.00000','Pure steam','pH','PSC/SP06','E4RM24006124','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'86253','D. Jayapal Reddy','2024-11-13 17:18:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.56 pH\r\nmV                   73.6 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1707,'2024-11-13','17:02:58','90.00000','105.00000','Pure steam','pH','PSC/SP09','E4RM24006166','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'86253','D. Jayapal Reddy','2024-11-13 17:18:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   66.1 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\nLog #                1038\r\n\r\n\r\n'),
(1708,'2024-11-13','17:07:39','90.00000','105.00000','0.9% SALINE','NA','RS008/24015','E4SO24000838','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SALIH',1,'205143','Ch. Prasad','2024-11-13 17:28:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.03 pH\r\nmV                   47.9 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             13-11-24,08:30\r\nLog #                418\r\n\r\n\r\n'),
(1709,'2024-11-13','17:11:48','90.00000','105.00000','Sabourarud dextrose agar','NA','RS021/24009','E4SO24000832','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SDAIH',1,'205143','Ch. Prasad','2024-11-13 17:25:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.77 pH\r\nmV                   63.6 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             13-11-24,08:30\r\n\r\n\r\n'),
(1710,'2024-11-13','17:17:18','90.00000','105.00000','Soyabean casein digest agar','NA','RS022/24009','E4SO24000831','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDAIH',1,'205143','Ch. Prasad','2024-11-13 17:25:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.42 pH\r\nmV                   -32.3 mV\r\nTemperature          26.7 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             13-11-24,08:30\r\n\r\n\r\n'),
(1711,'2024-11-13','18:34:43','90.00000','105.00000','Pure steam','pH','PSC/SP10','E4RM24006167','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'86253','D. Jayapal Reddy','2024-11-13 18:45:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.86 pH\r\nmV                   56.2 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             13-11-24,07:01\r\n\r\n\r\n'),
(1712,'2024-11-13','20:24:14','90.00000','105.00000','TAZOBACTAM','NA','TZ0024108','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','1',1,'88515','Prashanth B','2024-11-13 20:44:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.57 pH\r\nmV                   122.7 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             13-11-24,08:34\r\n\r\n\r\n'),
(1713,'2024-11-13','20:46:34','90.00000','105.00000','Soyabean casein digest agar','NA','RS022/24009','E4SO24000831','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDAIH',1,'205143','Ch. Prasad','2024-11-14 11:48:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.23 pH\r\nmV                   -20.7 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             13-11-24,08:30\r\n\r\n\r\n'),
(1714,'2024-11-13','20:53:51','90.00000','105.00000','Sabourarud dextrose agar','NA','RS021/24009','E4SO24000832','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SDAIH',1,'205143','Ch. Prasad','2024-11-14 11:49:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.76 pH\r\nmV                   63.5 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             13-11-24,08:30\r\n\r\n\r\n'),
(1715,'2024-11-13','20:55:46','90.00000','105.00000','0.9% SALINE','NA','RS008/24015','E4SO24000838','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SALIH',1,'205143','Ch. Prasad','2024-11-14 11:49:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.40 pH\r\nmV                   27.3 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             13-11-24,08:30\r\n\r\n\r\n'),
(1716,'2024-11-13','22:57:18','90.00000','105.00000','PIPERACILLIN','NA','TZ0024108','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-14 05:42:52','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.85 pH\r\nmV                   47.5 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             13-11-24,08:34\r\n\r\n\r\n'),
(1717,'2024-11-14','00:57:08','90.00000','105.00000','PIPERACILLIN','NA','TZ0024108','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-14 04:45:08','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.21 pH\r\nmV                   26.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        100.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             14-11-24,00:08\r\n\r\n\r\n'),
(1718,'2024-11-14','01:58:58','90.00000','105.00000','PIPERACILLIN','NA','TZ0024108','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-14 04:45:33','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.49 pH\r\nmV                   10.0 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        100.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             14-11-24,00:08\r\n\r\n\r\n'),
(1719,'2024-11-14','03:52:24','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024108','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','3',1,'84472','G. Ashok','2024-11-14 04:45:48','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.25 pH\r\nmV                   24.5 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        100.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             14-11-24,00:08\r\n\r\n\r\n'),
(1720,'2024-11-14','04:04:16','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024108','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','3',1,'84472','G. Ashok','2024-11-14 04:46:13','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.22 pH\r\nmV                   26.5 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        100.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             14-11-24,00:08\r\n\r\n\r\n'),
(1721,'2024-11-14','07:11:47','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005958','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','1.68 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-14 07:34:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.68 pH\r\nmV                   302.0 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1722,'2024-11-14','07:14:17','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005958','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','4.01pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-14 07:35:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   164.1 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1723,'2024-11-14','07:17:08','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005958','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','7.00pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-14 07:35:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -12.2 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1724,'2024-11-14','07:19:52','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005958','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','10.01 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-14 07:35:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.99 pH\r\nmV                   -187.5 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1725,'2024-11-14','07:22:45','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005958','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','12.46 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-14 07:36:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.50 pH\r\nmV                   -335.6 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1726,'2024-11-14','08:48:24','90.00000','105.00000','Piperacillin monohydrate','MOBILEPHASE-A','C66240901','E4RM24006055','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205266','K. Vara prasad',2,1,101,'pH','Assay Mobilephase pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-14 09:01:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   77.1 mV\r\nTemperature          29.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1727,'2024-11-14','08:55:22','90.00000','105.00000','Piperacillin monohydrate','Mobilephase-B','C66240901','E4RM24006055','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205266','K. Vara prasad',2,1,101,'pH','Assay Mobilephase pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-14 09:01:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   75.1 mV\r\nTemperature          23.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1728,'2024-11-14','10:02:20','90.00000','105.00000','Piperacillin and Tazobactam for injection','RS BY HPLC MP-A','TZ0424025A','E4CS24000528','Diluent compatibility','QC-PHMTZ-007','CX1-15733','1','NULL','205058','C. Srinivasulu',2,1,101,'pH','Piper Tazo',1,'205208','S. Siddhartha Kumar Reddy','2024-11-14 12:31:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   74.5 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1729,'2024-11-14','10:12:18','90.00000','105.00000','Piperacillin and Tazobactam for injection','RS BY HPLC MP-B','TZ0424025A','E4CS24000528','Diluent compatibility','QC-PHMTZ-007','CX1-15733','1','NULL','205058','C. Srinivasulu',2,1,101,'pH','Piper Tazo',1,'205208','S. Siddhartha Kumar Reddy','2024-11-14 12:31:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   74.8 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1730,'2024-11-14','10:24:22','90.00000','105.00000','Piperacillin and Tazobactam for injection','RS BY HPLC MP-A','TZ0424025A','E4CS24000488','Diluent compatibility','QC-PHMTZ-007','CX1-15733','1','NULL','205058','C. Srinivasulu',2,0,101,'pH','Piper Tazo',1,'205208','S. Siddhartha Kumar Reddy','2024-11-14 12:32:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   74.7 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1731,'2024-11-14','10:35:58','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0324001','E4PP24002076','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZ',1,'205208','S. Siddhartha Kumar Reddy','2024-11-14 11:10:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.56 pH\r\nmV                   73.8 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1732,'2024-11-14','10:41:41','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0324001','E4PP24002079','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZ',1,'205208','S. Siddhartha Kumar Reddy','2024-11-14 11:10:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   77.1 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1733,'2024-11-14','10:49:14','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC MP A','TT0424016','E4FP24000750','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','87697','M. Raghavendr Reddy',2,0,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-11-14 12:33:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.02 pH\r\nmV                   47.3 mV\r\nTemperature          29.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\nLog #                1052\r\n\r\n\r\n'),
(1734,'2024-11-14','11:29:26','90.00000','105.00000','TAZOBACTAM','NA','TR0024005','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-003','CW1-12031','1','NULL','88332','P.V.A.K. Raju',2,0,105,'pH','1',1,'90365','Abhilash','2024-11-14 11:43:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   4.40 pH\r\nmV                   153.8 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        95.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             14-11-24,01:27\r\n\r\n\r\n'),
(1735,'2024-11-14','11:57:28','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TZ0024108','E4IP24001349','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','pH (Bulk)',1,'205208','S. Siddhartha Kumar Reddy','2024-11-14 12:34:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.18 pH\r\nmV                   36.8 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1736,'2024-11-14','12:59:35','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005970','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-11-14 13:57:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   300.5 mV\r\nTemperature          26.7 C (ATC)\r\nAverage Slope        97.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             14-11-24,12:53\r\n\r\n\r\n'),
(1737,'2024-11-14','13:01:40','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005970','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-11-14 13:57:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   163.9 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        97.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             14-11-24,12:53\r\n\r\n\r\n'),
(1738,'2024-11-14','13:03:35','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005970','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-11-14 13:57:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   -7.6 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        97.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             14-11-24,12:53\r\n\r\n\r\n'),
(1739,'2024-11-14','13:06:04','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005970','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-11-14 13:58:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.02 pH\r\nmV                   -181.4 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        97.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             14-11-24,12:53\r\n\r\n\r\n'),
(1740,'2024-11-14','13:09:37','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005970','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-11-14 13:58:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.47 pH\r\nmV                   -321.6 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        97.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             14-11-24,12:53\r\n\r\n\r\n'),
(1741,'2024-11-14','13:59:33','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424124','E4FP24000753','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-11-14 14:23:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   63.2 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1742,'2024-11-14','14:12:22','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24006168','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90018','Ch. Satyam','2024-11-14 15:31:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.73 pH\r\nmV                   63.4 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1743,'2024-11-14','14:16:05','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24006169','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90018','Ch. Satyam','2024-11-14 15:32:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.67 pH\r\nmV                   67.1 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1744,'2024-11-14','14:19:16','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24006170','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90018','Ch. Satyam','2024-11-14 15:32:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.62 pH\r\nmV                   70.0 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1745,'2024-11-14','14:24:04','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24006171','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90018','Ch. Satyam','2024-11-14 15:33:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.60 pH\r\nmV                   71.4 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1746,'2024-11-14','14:28:04','90.00000','105.00000','Water For Injection','pH','WFI/SP33','E4RM24006172','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90018','Ch. Satyam','2024-11-14 15:32:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   75.3 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1747,'2024-11-14','14:33:03','90.00000','105.00000','Pure steam','pH','PSC/SP21','E4RM24006173','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'90018','Ch. Satyam','2024-11-14 15:32:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.64 pH\r\nmV                   69.0 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1748,'2024-11-14','14:37:05','90.00000','105.00000','Purified water','pH','PW/SP24','E4RM24006174','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90018','Ch. Satyam','2024-11-14 15:32:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   65.1 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1749,'2024-11-14','14:54:51','90.00000','105.00000','Water For Injection','pH','WFI/SP25','E4RM24006175','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90018','Ch. Satyam','2024-11-14 15:32:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.61 pH\r\nmV                   70.3 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1750,'2024-11-14','15:05:55','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024005','E4PP24002081','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'90018','Ch. Satyam','2024-11-14 15:40:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.39 pH\r\nmV                   142.5 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1751,'2024-11-14','16:42:49','90.00000','105.00000','Tryptone soya agar TSA with 0.05%soya lecithin&0.5% tween 80 and beta lactamase 55mm plates(RODAC)','NA','RS039/24015','E4SO24000844','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','205207','P. Hanmanthu',2,0,103,'pH','RUTSA',1,'205143','Ch. Prasad','2024-11-14 17:01:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.22 pH\r\nmV                   -20.2 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        97.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             14-11-24,12:53\r\nLog #                430\r\n\r\n\r\n'),
(1752,'2024-11-14','16:49:51','90.00000','105.00000','PIPERACILLIN','NA','TR0024005','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-003','CW1-12031','1','NULL','92595','K. Murali Krishna',2,0,105,'pH','2',1,'50654','A Suresh','2024-11-14 17:01:46','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   6.32 pH\r\nmV                   45.2 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        95.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             14-11-24,01:27\r\n\r\n\r\n'),
(1753,'2024-11-14','17:34:26','90.00000','105.00000','Peptone','NA','RS007/24048','E4SO24000845','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'88842','R. Mallesh','2024-11-15 19:17:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.92 pH\r\nmV                   -3.0 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        97.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             14-11-24,12:53\r\n\r\n\r\n'),
(1754,'2024-11-14','18:04:14','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024005','E4PP24002083','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','PIPERTAZ',1,'90018','Ch. Satyam','2024-11-14 20:52:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.23 pH\r\nmV                   33.8 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             14-11-24,07:07\r\n\r\n\r\n'),
(1755,'2024-11-14','19:36:19','90.00000','105.00000','Piperacillin and Tazobactam for Injection','NA','TR0024005','NA','BULK','PR-PHMTZ-003','CW1-12031','1','NULL','85705','P. Prakash Rao',2,0,105,'pH','pH (Bulk)',1,'50654','A Suresh','2024-11-14 19:58:10','verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   6.25 pH\r\nmV                   49.2 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        95.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             14-11-24,01:27\r\n\r\n\r\n'),
(1756,'2024-11-14','20:51:30','90.00000','105.00000','Piperacillin and Tazobactam for Injection','NA','TR0024005','NA','BULK','PR-PHMTZ-003','CW1-12031','1','NULL','85705','P. Prakash Rao',2,0,105,'pH','pH (Bulk)',1,'88515','Prashanth B','2024-11-14 21:55:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   6.28 pH\r\nmV                   47.1 mV\r\nTemperature          20.3 C (ATC)\r\nAverage Slope        95.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             14-11-24,01:27\r\n\r\n\r\n'),
(1757,'2024-11-14','21:25:50','90.00000','105.00000','Peptone','NA','RS007/24048','E4SO24000845','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'88842','R. Mallesh','2024-11-15 19:19:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.92 pH\r\nmV                   -2.7 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        97.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             14-11-24,12:53\r\n\r\n\r\n'),
(1758,'2024-11-14','23:09:42','90.00000','105.00000','Piperacillin and Tazobactam for Injection','NA','TR0024005','NA','BULK','PR-PHMTZ-003','CW1-12031','1','NULL','85705','P. Prakash Rao',2,0,105,'pH','pH (Bulk)',1,'88629','Raghuram Reddy','2024-11-14 23:35:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   6.29 pH\r\nmV                   46.7 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        95.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             14-11-24,01:27\r\n\r\n\r\n'),
(1759,'2024-11-15','07:06:52','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005979','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','1.68 pH',1,'84378','Shivanand. W','2024-11-15 07:31:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   302.3 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1760,'2024-11-15','07:09:26','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005979','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','4.01pH',1,'84378','Shivanand. W','2024-11-15 07:32:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   164.4 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\nLog #                1066\r\n\r\n\r\n'),
(1761,'2024-11-15','07:14:33','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005979','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','7.00pH',1,'84378','Shivanand. W','2024-11-15 07:32:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -11.4 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1762,'2024-11-15','07:18:40','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005979','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','10.01 pH',1,'84378','Shivanand. W','2024-11-15 07:33:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.96 pH\r\nmV                   -185.6 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\nLog #                1068\r\n\r\n\r\n'),
(1763,'2024-11-15','07:21:10','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005979','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','12.46 pH',1,'84378','Shivanand. W','2024-11-15 07:33:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.48 pH\r\nmV                   -334.1 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1764,'2024-11-15','08:49:38','90.00000','105.00000','TAZOBACTAM','NA','TZ0024109','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','92936','Sameer',2,0,104,'pH','1',1,'88515','Prashanth B','2024-11-15 09:36:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.47 pH\r\nmV                   127.7 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             15-11-24,07:06\r\n\r\n\r\n'),
(1765,'2024-11-15','09:54:07','90.00000','105.00000','Piperacillin and Tazobactam For injection','ASSAY','TT0424016','E4FP24000750','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','90017','G. Srishailam',2,0,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-11-15 14:33:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   78.4 mV\r\nTemperature          32.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1766,'2024-11-15','10:22:25','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005987','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','1.68 pH',1,'85152','V. Chandra Shekara Achari','2024-11-15 11:04:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.65 pH\r\nmV                   300.7 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        97.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             15-11-24,10:20\r\nLog #                433\r\n\r\n\r\n'),
(1767,'2024-11-15','10:24:06','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005987','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','4.01pH',1,'85152','V. Chandra Shekara Achari','2024-11-15 11:04:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   162.3 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        97.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             15-11-24,10:20\r\n\r\n\r\n'),
(1768,'2024-11-15','10:27:04','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','ph ASSAY','TR0024005','E4PP24002085','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,1,101,'pH','PIPERTAZ',1,'205208','S. Siddhartha Kumar Reddy','2024-11-15 14:28:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   78.1 mV\r\nTemperature          32.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1769,'2024-11-15','10:26:50','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005987','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','7.00pH',1,'85152','V. Chandra Shekara Achari','2024-11-15 11:04:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.99 pH\r\nmV                   -9.1 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        97.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             15-11-24,10:20\r\nLog #                435\r\n\r\n\r\n'),
(1770,'2024-11-15','10:34:04','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005987','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','10.01 pH',1,'85152','V. Chandra Shekara Achari','2024-11-15 11:05:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   9.98 pH\r\nmV                   -182.5 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        97.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             15-11-24,10:20\r\nLog #                437\r\n\r\n\r\n'),
(1771,'2024-11-15','10:35:33','90.00000','105.00000','Piperacillin And Tazobactam','IDENTIFICATION','2449201230','E4RM24006087','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205232','K. Narender Reddy',2,1,101,'pH','PT',1,'84378','Shivanand. W','2024-11-15 11:29:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   78.7 mV\r\nTemperature          32.2 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\nLog #                1072\r\n\r\n\r\n'),
(1772,'2024-11-15','10:38:19','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005987','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','12.46 pH',1,'85152','V. Chandra Shekara Achari','2024-11-15 11:05:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.43 pH\r\nmV                   -321.8 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        97.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             15-11-24,10:20\r\nLog #                438\r\n\r\n\r\n'),
(1773,'2024-11-15','10:49:31','90.00000','105.00000','Piperacillin and Tazobactam for injection','Cleaning Verification','TR0024005','E4CV24000314','Cleaning Validation','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,0,101,'pH','PTCV',1,'84378','Shivanand. W','2024-11-15 11:30:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   79.8 mV\r\nTemperature          32.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1774,'2024-11-15','11:15:52','90.00000','105.00000','AMPISULBACTAM','ASSAY','2449102402','E4RM24006176','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205189','P. Ravi Kumar',2,1,101,'pH','Assay buffer pH',1,'84378','Shivanand. W','2024-11-15 11:25:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.01 pH\r\nmV                   109.1 mV\r\nTemperature          32.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1775,'2024-11-15','11:20:44','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','pH','TR0024005','E4PP24002084','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','3',1,'84378','Shivanand. W','2024-11-15 12:02:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.67 pH\r\nmV                   67.7 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1776,'2024-11-15','11:24:09','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','pH','TR0024005','E4PP24002086','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','3',1,'84378','Shivanand. W','2024-11-15 12:03:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.91 pH\r\nmV                   53.1 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1777,'2024-11-15','11:27:03','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','pH','TR0024005','E4PP24002087','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','3',1,'84378','Shivanand. W','2024-11-15 12:03:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.89 pH\r\nmV                   54.7 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1778,'2024-11-15','11:30:10','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','pH','TR0024005','E4PP24002089','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','3',1,'84378','Shivanand. W','2024-11-15 12:03:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.88 pH\r\nmV                   55.3 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1779,'2024-11-15','11:43:20','90.00000','105.00000','PIPERACILLIN','NA','TZ0024109','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'88515','Prashanth B','2024-11-15 13:59:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   48.1 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             15-11-24,07:06\r\n\r\n\r\n'),
(1780,'2024-11-15','12:15:37','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24022','E4SO24000849','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SCDBIH',1,'205143','Ch. Prasad','2024-11-15 13:47:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.36 pH\r\nmV                   -30.7 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        97.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             15-11-24,10:20\r\n\r\n\r\n'),
(1781,'2024-11-15','12:17:53','90.00000','105.00000','SCDM WITH 4%Tween80 &0.5 Soya lecthin','NA','RS011/24010','E4SO24000850','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','NSCDBIH',1,'205143','Ch. Prasad','2024-11-15 13:48:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.37 pH\r\nmV                   -31.3 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        97.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             15-11-24,10:20\r\n\r\n\r\n'),
(1782,'2024-11-15','12:20:07','90.00000','105.00000','SCDM with 1% Tween 80','NA','RS034/24009','E4SO24000851','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','NSSCDBSIH',1,'205143','Ch. Prasad','2024-11-15 13:49:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.36 pH\r\nmV                   -30.6 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        97.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             15-11-24,10:20\r\n\r\n\r\n'),
(1783,'2024-11-15','12:26:35','90.00000','105.00000','PIPERACILLIN','NA','TZ0024109','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'88515','Prashanth B','2024-11-15 14:01:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.86 pH\r\nmV                   46.6 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             15-11-24,07:06\r\n\r\n\r\n'),
(1784,'2024-11-15','13:12:03','90.00000','105.00000','PIPERACILLIN','NA','TZ0024109','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'88515','Prashanth B','2024-11-15 14:01:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.20 pH\r\nmV                   26.2 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             15-11-24,07:06\r\n\r\n\r\n'),
(1785,'2024-11-15','13:51:45','90.00000','105.00000','Ampicillin and sulbactam for Injection','pH','AS0124001A','E4STFP24000706','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Ampisulbactam',1,'84378','Shivanand. W','2024-11-15 14:27:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.10 pH\r\nmV                   -134.7 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1786,'2024-11-15','13:56:19','90.00000','105.00000','Ampicillin and sulbactam for Injection','pH','AS0324002A','E4STFP24000707','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Ampisulbactam',1,'84378','Shivanand. W','2024-11-15 14:28:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.08 pH\r\nmV                   -133.9 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1787,'2024-11-15','13:58:04','90.00000','105.00000','PIPERACILLIN','NA','TZ0024109','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'88515','Prashanth B','2024-11-15 14:02:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.48 pH\r\nmV                   9.9 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             15-11-24,07:06\r\n\r\n\r\n'),
(1788,'2024-11-15','14:02:31','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0324002','E4PP24002100','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'84378','Shivanand. W','2024-11-15 14:28:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   76.5 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1789,'2024-11-15','14:06:45','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0324002','E4PP24002101','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'84378','Shivanand. W','2024-11-15 14:29:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.62 pH\r\nmV                   70.3 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1790,'2024-11-15','14:13:06','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0324002','E4PP24002102','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'84378','Shivanand. W','2024-11-15 14:29:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.64 pH\r\nmV                   69.1 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1791,'2024-11-15','14:50:51','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0324002','E4PP24002103','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'84378','Shivanand. W','2024-11-15 14:55:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.79 pH\r\nmV                   60.6 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1792,'2024-11-15','15:08:17','90.00000','105.00000','Sodium Hydrogen Carbonate','pH','SL/214/173','E4RM24006125','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205266','K. Vara prasad',2,1,101,'pH','pH   Test',1,'90778','B. Bhaskar','2024-11-15 15:19:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   8.05 pH\r\nmV                   -72.7 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1793,'2024-11-15','15:14:57','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24006195','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-11-15 15:19:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.13 pH\r\nmV                   40.0 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1794,'2024-11-15','15:18:25','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24006196','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-11-15 15:49:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.09 pH\r\nmV                   42.9 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1795,'2024-11-15','15:22:01','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24006197','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90778','B. Bhaskar','2024-11-15 15:49:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.60 pH\r\nmV                   71.6 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1796,'2024-11-15','15:26:05','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24006198','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90778','B. Bhaskar','2024-11-15 15:49:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   77.1 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\nLog #                1089\r\n\r\n\r\n'),
(1797,'2024-11-15','15:27:53','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024109','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','3',1,'88629','Raghuram Reddy','2024-11-15 16:06:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.27 pH\r\nmV                   21.7 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             15-11-24,07:06\r\n\r\n\r\n'),
(1798,'2024-11-15','15:32:27','90.00000','105.00000','Purified water','pH','PW/SP34','E4RM24006199','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-11-15 15:50:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.58 pH\r\nmV                   72.5 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1799,'2024-11-15','15:35:16','90.00000','105.00000','Purified water','pH','PW/SP36','E4RM24006200','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-11-15 15:50:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   75.2 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1800,'2024-11-15','15:38:12','90.00000','105.00000','Purified water','pH','PW/SP40','E4RM24006201','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-11-15 15:50:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.56 pH\r\nmV                   74.1 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1801,'2024-11-15','15:41:13','90.00000','105.00000','Pure steam','pH','PSC/SP04','E4RM24006202','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'90778','B. Bhaskar','2024-11-15 15:51:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   76.2 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1802,'2024-11-15','15:47:28','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024109','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','3',1,'88629','Raghuram Reddy','2024-11-15 16:06:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.23 pH\r\nmV                   24.3 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             15-11-24,07:06\r\n\r\n\r\n'),
(1803,'2024-11-15','17:09:44','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24022','E4SO24000849','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDBIH',1,'88842','R. Mallesh','2024-11-15 19:33:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.31 pH\r\nmV                   -27.7 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        97.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             15-11-24,10:20\r\n\r\n\r\n'),
(1804,'2024-11-15','17:17:30','90.00000','105.00000','SCDM WITH 4%Tween80 &0.5 Soya lecthin','NA','RS011/24010','E4SO24000850','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','NSCDBIH',1,'88842','R. Mallesh','2024-11-15 19:37:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.25 pH\r\nmV                   -24.1 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        97.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             15-11-24,10:20\r\n\r\n\r\n'),
(1805,'2024-11-15','17:19:48','90.00000','105.00000','SCDM with 1% Tween 80','NA','RS034/24009','E4SO24000851','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','NSSCDBSIH',1,'88842','R. Mallesh','2024-11-15 22:35:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.26 pH\r\nmV                   -24.7 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        97.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             15-11-24,10:20\r\n\r\n\r\n'),
(1806,'2024-11-15','17:22:04','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0324002','E4PP24002106','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZ',1,'90778','B. Bhaskar','2024-11-15 17:24:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.87 pH\r\nmV                   55.5 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             15-11-24,07:03\r\n\r\n\r\n'),
(1807,'2024-11-15','17:29:21','90.00000','105.00000','Soyabean casean digest agar SCDA With 0.05%soya lecithin 05% tween 80 and beta lactamase (Finger dab)','NA','RS041/24008','E4SO24000855','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','RUSCDAF',1,'88842','R. Mallesh','2024-11-15 22:38:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.24 pH\r\nmV                   -23.7 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        97.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             15-11-24,10:20\r\n\r\n\r\n'),
(1808,'2024-11-16','01:09:03','90.00000','105.00000','TAZOBACTAM','NA','TR0024006','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-003','CW1-12031','1','NULL','205169','M. Venkatesham',2,0,105,'pH','1',1,'95681','T. Sivanarayana','2024-11-16 05:55:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   4.39 pH\r\nmV                   151.9 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        95.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             16-11-24,00:07\r\n\r\n\r\n'),
(1809,'2024-11-16','02:52:05','90.00000','105.00000','PIPERACILLIN','NA','TR0024006','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-003','CW1-12031','1','NULL','85705','P. Prakash Rao',2,0,105,'pH','2',1,'95681','T. Sivanarayana','2024-11-16 05:55:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   6.39 pH\r\nmV                   39.3 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        95.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             16-11-24,00:07\r\n\r\n\r\n'),
(1810,'2024-11-16','07:05:10','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TR0024006','NA','BULK','PR-PHMTZ-003','CW1-12031','1','NULL','88332','P.V.A.K. Raju',2,0,105,'pH','3',1,'90365','Abhilash','2024-11-16 07:14:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   6.35 pH\r\nmV                   41.5 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        95.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             16-11-24,00:07\r\n\r\n\r\n'),
(1811,'2024-11-16','07:15:33','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005991','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-11-16 07:37:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   302.2 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1812,'2024-11-16','07:18:46','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005991','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-11-16 07:37:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   164.1 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1813,'2024-11-16','07:22:07','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005991','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-11-16 07:37:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   -11.3 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1814,'2024-11-16','07:25:39','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005991','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-11-16 07:37:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.99 pH\r\nmV                   -187.8 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1815,'2024-11-16','07:28:43','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005991','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-11-16 07:38:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.50 pH\r\nmV                   -335.7 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1816,'2024-11-16','07:54:02','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TR0024006','NA','BULK','PR-PHMTZ-003','CW1-12031','1','NULL','88332','P.V.A.K. Raju',2,0,105,'pH','3',1,'90365','Abhilash','2024-11-16 08:28:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   6.36 pH\r\nmV                   41.3 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        95.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             16-11-24,00:07\r\n\r\n\r\n'),
(1817,'2024-11-16','09:15:43','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TZ0024109','E4IP24001354','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','pH (Bulk)',1,'82735','M.D. Babar Qhadri','2024-11-16 10:19:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.21 pH\r\nmV                   35.3 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1818,'2024-11-16','09:21:29','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PJ0422001A','E4STFP24000704','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-11-16 11:21:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.30 pH\r\nmV                   89.5 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1819,'2024-11-16','09:27:05','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TR0424001A','E4STFP24000708','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-11-16 11:21:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.95 pH\r\nmV                   51.1 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1820,'2024-11-16','10:19:48','90.00000','105.00000','Ampicillin and Sulbactam for injection','ID','2449101502','E4RM24006059','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','90017','G. Srishailam',2,0,101,'pH','Ampi&Sulbactam',1,'82735','M.D. Babar Qhadri','2024-11-16 11:24:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.02 pH\r\nmV                   108.5 mV\r\nTemperature          32.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1821,'2024-11-16','10:46:44','90.00000','105.00000','pH Buffer','NA','NA','E4INS24006007','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-11-16 11:46:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   295.9 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        97.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             16-11-24,10:41\r\n\r\n\r\n'),
(1822,'2024-11-16','10:48:31','90.00000','105.00000','pH Buffer','NA','NA','E4INS24006007','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-11-16 11:46:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   162.6 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        97.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             16-11-24,10:41\r\n\r\n\r\n'),
(1823,'2024-11-16','10:50:21','90.00000','105.00000','pH Buffer','NA','NA','E4INS24006007','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-11-16 11:47:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   -9.0 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        97.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             16-11-24,10:41\r\n\r\n\r\n'),
(1824,'2024-11-16','10:52:46','90.00000','105.00000','pH Buffer','NA','NA','E4INS24006007','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-11-16 11:47:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   9.99 pH\r\nmV                   -182.5 mV\r\nTemperature          26.9 C (ATC)\r\nAverage Slope        97.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             16-11-24,10:41\r\n\r\n\r\n'),
(1825,'2024-11-16','11:04:07','90.00000','105.00000','Ampcillin for injection','RS','AP1024001A','E4STFP24000690','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','Ampi',1,'82735','M.D. Babar Qhadri','2024-11-16 11:21:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.02 pH\r\nmV                   107.0 mV\r\nTemperature          28.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1826,'2024-11-16','11:15:01','90.00000','105.00000','Piperacillin and Tazobactam For injection','UOD','PT0424124','E4FP24000753','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,0,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-16 11:22:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.48 pH\r\nmV                   80.5 mV\r\nTemperature          32.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1827,'2024-11-16','11:26:08','90.00000','105.00000','pH Buffer','NA','NA','E4INS24006007','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-11-16 11:47:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.48 pH\r\nmV                   -322.6 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        97.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             16-11-24,10:41\r\nLog #                450\r\n\r\n\r\n'),
(1828,'2024-11-16','11:22:53','90.00000','105.00000','Ampicillin and Sulbactam','RS BY HPLC MP A','AS0324086','E4FP24000745','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','87697','M. Raghavendr Reddy',2,0,101,'pH','AS',1,'82735','M.D. Babar Qhadri','2024-11-16 11:58:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.00 pH\r\nmV                   108.5 mV\r\nTemperature          28.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\nLog #                1106\r\n\r\n\r\n'),
(1829,'2024-11-16','11:41:03','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','NA','TR0024006','E4PP24002108','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,2,101,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-11-16 11:59:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.72 pH\r\nmV                   122.9 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1830,'2024-11-16','11:44:54','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024006','E4PP24002110','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-11-16 12:00:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.14 pH\r\nmV                   39.4 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1831,'2024-11-16','11:47:41','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024006','E4PP24002111','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-11-16 12:00:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.95 pH\r\nmV                   50.7 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1832,'2024-11-16','12:03:47','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324087','E4FP24000752','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,2,101,'pH','AS',1,'90018','Ch. Satyam','2024-11-16 13:33:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.18 pH\r\nmV                   -139.8 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1833,'2024-11-16','12:06:55','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324088','E4FP24000755','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,2,101,'pH','AS',1,'90018','Ch. Satyam','2024-11-16 13:34:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.19 pH\r\nmV                   -140.8 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1834,'2024-11-16','12:12:29','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424128','E4FP24000754','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,2,101,'pH','Pipertazoinj',1,'90018','Ch. Satyam','2024-11-16 13:35:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.72 pH\r\nmV                   64.3 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1835,'2024-11-16','12:19:45','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TZ0424105','E4FP24000751','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,2,101,'pH','Pipertazoinj',1,'90018','Ch. Satyam','2024-11-16 13:36:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.28 pH\r\nmV                   31.5 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1836,'2024-11-16','12:24:19','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24006225','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,2,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-11-16 13:37:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.57 pH\r\nmV                   73.1 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\nLog #                1114\r\n\r\n\r\n'),
(1837,'2024-11-16','12:27:34','90.00000','105.00000','Soyabean casein digest broth for sterilty','NA','RS033/24011','E4SO24000861','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SCDBSIH',1,'88842','R. Mallesh','2024-11-16 13:37:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.38 pH\r\nmV                   -30.8 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        97.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             16-11-24,10:41\r\nLog #                451\r\n\r\n\r\n'),
(1838,'2024-11-16','12:29:07','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24006226','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,2,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-11-16 13:38:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.60 pH\r\nmV                   71.5 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\nLog #                1115\r\n\r\n\r\n'),
(1839,'2024-11-16','12:32:11','90.00000','105.00000','Peptone','NA','RS007/24049','E4SO24000859','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','PEPIH',1,'88842','R. Mallesh','2024-11-16 13:40:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -10.4 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        97.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             16-11-24,10:41\r\nLog #                452\r\n\r\n\r\n'),
(1840,'2024-11-16','12:34:28','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24006227','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,2,101,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-11-16 13:38:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   74.9 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1841,'2024-11-16','12:37:01','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24006228','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,2,101,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-11-16 13:39:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   75.6 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1842,'2024-11-16','12:40:33','90.00000','105.00000','Water For Injection','pH','WFI/SP34','E4RM24006229','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,2,101,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-11-16 13:39:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   77.5 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\nLog #                1118\r\n\r\n\r\n'),
(1843,'2024-11-16','12:46:57','90.00000','105.00000','Purified water','pH','PW/SP45','E4RM24006230','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,2,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-11-16 13:41:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.57 pH\r\nmV                   72.8 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1844,'2024-11-16','12:49:31','90.00000','105.00000','Purified water','pH','PW/SP09','E4RM24006231','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-16 14:45:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   74.8 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1845,'2024-11-16','13:12:09','90.00000','105.00000','Fluid thioglycollate meddiuam','NA','RS026/24012','E4SO24000860','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','FTGMIH',1,'205143','Ch. Prasad','2024-11-16 14:35:12','Adjust as per SOP','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.32 pH\r\nmV                   -28.1 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        97.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             16-11-24,10:41\r\n\r\n\r\n'),
(1846,'2024-11-16','14:30:00','90.00000','105.00000','Fluid thioglycollate meddiuam','NA','RS026/24012','E4SO24000860','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','FTGMIH',1,'205143','Ch. Prasad','2024-11-16 14:35:39','Adjusted as per SOP','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.09 pH\r\nmV                   -14.7 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        97.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             16-11-24,10:41\r\n\r\n\r\n'),
(1847,'2024-11-16','15:14:26','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0222002A','E4STFP24000711','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','PIPERTAZO',1,'205208','S. Siddhartha Kumar Reddy','2024-11-16 15:34:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.34 pH\r\nmV                   86.8 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1848,'2024-11-16','15:21:08','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0222003A','E4STFP24000712','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','PIPERTAZO',1,'205208','S. Siddhartha Kumar Reddy','2024-11-16 15:36:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   65.9 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1849,'2024-11-16','15:24:43','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0222001A','E4STFP24000713','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','PIPERTAZO',1,'205208','S. Siddhartha Kumar Reddy','2024-11-16 15:37:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.32 pH\r\nmV                   88.3 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1850,'2024-11-16','15:29:12','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0222004B','E4STFP24000714','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','PIPERTAZO',1,'205208','S. Siddhartha Kumar Reddy','2024-11-16 15:38:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.72 pH\r\nmV                   64.3 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1851,'2024-11-16','15:56:01','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024006','E4PP24002114','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','PIPERTAZ',1,'205208','S. Siddhartha Kumar Reddy','2024-11-16 17:06:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.95 pH\r\nmV                   50.4 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1852,'2024-11-16','15:59:04','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024006','E4PP24002115','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','PIPERTAZ',1,'205208','S. Siddhartha Kumar Reddy','2024-11-16 17:06:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.96 pH\r\nmV                   50.5 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1853,'2024-11-16','16:02:00','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024006','E4PP24002117','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','PIPERTAZ',1,'205208','S. Siddhartha Kumar Reddy','2024-11-16 17:05:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.94 pH\r\nmV                   51.3 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1854,'2024-11-16','16:08:30','90.00000','105.00000','Glass Vials for pH','pH(Vial-1)','NA','E4EQ24000401','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:01:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.37 pH\r\nmV                   261.7 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1855,'2024-11-16','16:10:31','90.00000','105.00000','Glass Vials for pH','pH(Vial-2)','NA','E4EQ24000401','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:01:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.38 pH\r\nmV                   261.2 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1856,'2024-11-16','16:12:49','90.00000','105.00000','Glass Vials for pH','pH(Vial-3)','NA','E4EQ24000401','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:01:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.43 pH\r\nmV                   258.6 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1857,'2024-11-16','16:14:51','90.00000','105.00000','Glass Vials for pH','pH(Vial-4)','NA','E4EQ24000401','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:01:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.48 pH\r\nmV                   255.0 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1858,'2024-11-16','16:16:45','90.00000','105.00000','Glass Vials for pH','pH(Vial-5)','NA','E4EQ24000401','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:01:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.60 pH\r\nmV                   247.8 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1859,'2024-11-16','16:18:43','90.00000','105.00000','Glass Vials for pH','pH(Vial-6)','NA','E4EQ24000401','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:02:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.68 pH\r\nmV                   243.7 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1860,'2024-11-16','16:20:38','90.00000','105.00000','Glass Vials for pH','pH(Vial-7)','NA','E4EQ24000401','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:02:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.74 pH\r\nmV                   240.3 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1861,'2024-11-16','16:22:51','90.00000','105.00000','Glass Vials for pH','pH(Vial-8)','NA','E4EQ24000401','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:02:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.30 pH\r\nmV                   266.2 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1862,'2024-11-16','16:25:09','90.00000','105.00000','Glass Vials for pH','pH(Vial-9)','NA','E4EQ24000401','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:03:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.40 pH\r\nmV                   260.0 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1863,'2024-11-16','16:27:17','90.00000','105.00000','Glass Vials for pH','pH(Vial-10)','NA','E4EQ24000401','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:03:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.48 pH\r\nmV                   255.3 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1864,'2024-11-16','16:29:53','90.00000','105.00000','Glass Vials for pH','pH(Vial-11)','NA','E4EQ24000401','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:03:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.58 pH\r\nmV                   249.0 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1865,'2024-11-16','16:32:44','90.00000','105.00000','Glass Vials for pH','pH(Vial-12)','NA','E4EQ24000401','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:04:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.37 pH\r\nmV                   261.6 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1866,'2024-11-16','16:34:37','90.00000','105.00000','Glass Vials for pH','pH(Vial-13)','NA','E4EQ24000401','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:04:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.42 pH\r\nmV                   258.6 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1867,'2024-11-16','16:36:42','90.00000','105.00000','Glass Vials for pH','pH(Vial-14)','NA','E4EQ24000401','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:04:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.51 pH\r\nmV                   253.3 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1868,'2024-11-16','16:38:46','90.00000','105.00000','Glass Vials for pH','pH(Vial-15)','NA','E4EQ24000401','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:04:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.57 pH\r\nmV                   249.9 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1869,'2024-11-16','16:40:35','90.00000','105.00000','Glass Vials for pH','pH(Vial-16)','NA','E4EQ24000401','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:04:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.61 pH\r\nmV                   247.7 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1870,'2024-11-16','16:50:30','90.00000','105.00000','Glass Vials for pH','pH(Vial-1)','NA','E4EQ24000402','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:05:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.61 pH\r\nmV                   247.3 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1871,'2024-11-16','16:53:09','90.00000','105.00000','Glass Vials for pH','pH(Vial-2)','NA','E4EQ24000402','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:05:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.56 pH\r\nmV                   251.0 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1872,'2024-11-16','16:55:17','90.00000','105.00000','Glass Vials for pH','pH(Vial-3)','NA','E4EQ24000402','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:06:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.21 pH\r\nmV                   271.1 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1873,'2024-11-16','16:57:34','90.00000','105.00000','Glass Vials for pH','pH(Vial-4)','NA','E4EQ24000402','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:06:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.37 pH\r\nmV                   262.0 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1874,'2024-11-16','16:59:41','90.00000','105.00000','Glass Vials for pH','pH(Vial-5)','NA','E4EQ24000402','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:06:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.45 pH\r\nmV                   257.3 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1875,'2024-11-16','17:02:01','90.00000','105.00000','Glass Vials for pH','pH(Vial-6)','NA','E4EQ24000402','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-18 11:52:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.53 pH\r\nmV                   252.1 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1876,'2024-11-16','17:04:06','90.00000','105.00000','Glass Vials for pH','pH(Vial-7)','NA','E4EQ24000402','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:07:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.57 pH\r\nmV                   249.4 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1877,'2024-11-16','17:06:52','90.00000','105.00000','Glass Vials for pH','pH(Vial-8)','NA','E4EQ24000402','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:07:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.66 pH\r\nmV                   244.1 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1878,'2024-11-16','17:09:21','90.00000','105.00000','Glass Vials for pH','pH(Vial-9)','NA','E4EQ24000402','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:07:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.77 pH\r\nmV                   238.2 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1879,'2024-11-16','17:11:32','90.00000','105.00000','Glass Vials for pH','pH(Vial-10)','NA','E4EQ24000402','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:07:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.39 pH\r\nmV                   260.5 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1880,'2024-11-16','17:13:52','90.00000','105.00000','Glass Vials for pH','pH(Vial-11)','NA','E4EQ24000402','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:07:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.54 pH\r\nmV                   251.4 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1881,'2024-11-16','17:16:17','90.00000','105.00000','Glass Vials for pH','pH(Vial-12)','NA','E4EQ24000402','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:08:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.78 pH\r\nmV                   237.0 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1882,'2024-11-16','17:18:36','90.00000','105.00000','Glass Vials for pH','pH(Vial-13)','NA','E4EQ24000402','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:08:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.55 pH\r\nmV                   250.6 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1883,'2024-11-16','17:21:03','90.00000','105.00000','Glass Vials for pH','pH(Vial-14)','NA','E4EQ24000402','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:08:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.62 pH\r\nmV                   246.6 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1884,'2024-11-16','17:23:17','90.00000','105.00000','Glass Vials for pH','pH( ial-15)','NA','E4EQ24000402','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-18 11:51:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.69 pH\r\nmV                   242.7 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1885,'2024-11-16','17:25:27','90.00000','105.00000','Glass Vials for pH','pH(Vial-16)','NA','E4EQ24000402','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-11-17 07:08:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.27 pH\r\nmV                   267.1 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             16-11-24,07:11\r\n\r\n\r\n'),
(1886,'2024-11-16','20:54:58','90.00000','105.00000','Peptone','NA','RS007/24049','E4SO24000859','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-11-18 12:15:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -10.8 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        97.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             16-11-24,10:41\r\n\r\n\r\n'),
(1887,'2024-11-16','20:58:48','90.00000','105.00000','Fluid thioglycollate meddiuam','NA','RS026/24012','E4SO24000860','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','FTGMIH',1,'205143','Ch. Prasad','2024-11-18 12:17:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.23 pH\r\nmV                   -22.6 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        97.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             16-11-24,10:41\r\n\r\n\r\n'),
(1888,'2024-11-16','21:00:50','90.00000','105.00000','Soyabean casein digest broth for sterilty','NA','RS033/24011','E4SO24000861','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDBSIH',1,'205143','Ch. Prasad','2024-11-18 12:20:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.31 pH\r\nmV                   -27.1 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        97.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             16-11-24,10:41\r\n\r\n\r\n'),
(1889,'2024-11-17','08:13:14','90.00000','105.00000','pH Buffer','NA','NA','E4INS24006018','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-11-17 09:10:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.67 pH\r\nmV                   302.6 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             17-11-24,08:00\r\n\r\n\r\n'),
(1890,'2024-11-17','08:18:55','90.00000','105.00000','pH Buffer','NA','NA','E4INS24006018','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-11-17 09:10:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   164.6 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             17-11-24,08:00\r\n\r\n\r\n'),
(1891,'2024-11-17','08:23:43','90.00000','105.00000','pH Buffer','NA','NA','E4INS24006018','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-11-17 09:10:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -11.6 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             17-11-24,08:00\r\n\r\n\r\n'),
(1892,'2024-11-17','08:30:11','90.00000','105.00000','pH Buffer','NA','NA','E4INS24006018','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-11-17 09:11:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.02 pH\r\nmV                   -187.4 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             17-11-24,08:00\r\n\r\n\r\n'),
(1893,'2024-11-17','08:39:00','90.00000','105.00000','pH Buffer','NA','NA','E4INS24006018','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-11-17 09:11:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.50 pH\r\nmV                   -334.2 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             17-11-24,08:00\r\n\r\n\r\n'),
(1894,'2024-11-17','10:13:54','90.00000','105.00000','Piperacillin and Tazobactam For injection','ASSAY','PT0424123','E4FP24000749','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205232','K. Narender Reddy',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-17 12:08:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   80.2 mV\r\nTemperature          31.7 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             17-11-24,08:00\r\n\r\n\r\n'),
(1895,'2024-11-17','11:34:25','90.00000','105.00000','Piperacillin and Tazobactam for injection','CV','TR0024006','E4CV24000333','Cleaning Validation','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,0,101,'pH','PTCV',1,'82735','M.D. Babar Qhadri','2024-11-17 12:08:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   79.4 mV\r\nTemperature          32.2 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             17-11-24,08:00\r\n\r\n\r\n'),
(1896,'2024-11-17','11:39:14','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','ph','TR0324003','E4PP24002168','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-11-17 12:05:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.64 pH\r\nmV                   69.7 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             17-11-24,08:00\r\n\r\n\r\n'),
(1897,'2024-11-17','11:42:48','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','ph','TR0324003','E4PP24002169','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-11-17 12:05:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.78 pH\r\nmV                   62.0 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             17-11-24,08:00\r\n\r\n\r\n'),
(1898,'2024-11-17','11:46:11','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','ph','TR0324003','E4PP24002170','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-11-17 12:06:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.82 pH\r\nmV                   59.3 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             17-11-24,08:00\r\n\r\n\r\n'),
(1899,'2024-11-17','11:50:01','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','ph','TR0324003','E4PP24002171','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-11-17 12:06:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   58.6 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             17-11-24,08:00\r\n\r\n\r\n'),
(1900,'2024-11-17','11:53:26','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','ph','TR0324003','E4PP24002183','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-11-17 12:06:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.85 pH\r\nmV                   57.7 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             17-11-24,08:00\r\n\r\n\r\n'),
(1901,'2024-11-17','12:19:32','90.00000','105.00000','TAZOBACTAM','NA','TR0024007','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-003','CW1-12031','1','NULL','88332','P.V.A.K. Raju',2,0,105,'pH','1',1,'90365','Abhilash','2024-11-17 12:32:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   4.51 pH\r\nmV                   144.9 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        95.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             17-11-24,09:55\r\n\r\n\r\n'),
(1902,'2024-11-17','18:54:36','90.00000','105.00000','PIPERACILLIN','NA','TR0024007','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-003','CW1-12031','1','NULL','205169','M. Venkatesham',2,0,105,'pH','2',1,'84472','G. Ashok','2024-11-17 22:29:45','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   6.48 pH\r\nmV                   34.2 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        95.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             17-11-24,09:55\r\n\r\n\r\n'),
(1903,'2024-11-17','20:45:07','90.00000','105.00000','Piperacillin and Tazobactam for Injection','NA','TR0024007','NA','BULK','PR-PHMTZ-003','CW1-12031','1','NULL','205169','M. Venkatesham',2,0,105,'pH','pH (Bulk)',1,'84472','G. Ashok','2024-11-17 22:30:11','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   6.34 pH\r\nmV                   41.9 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        95.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             17-11-24,09:55\r\n\r\n\r\n'),
(1904,'2024-11-17','21:20:16','90.00000','105.00000','Piperacillin and Tazobactam for Injection','NA','TR0024007','NA','BULK','PR-PHMTZ-003','CW1-12031','1','NULL','91700','Ch. Ramarao',2,0,105,'pH','pH (Bulk)',1,'84472','G. Ashok','2024-11-17 22:30:31','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   6.35 pH\r\nmV                   41.3 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        95.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             17-11-24,09:55\r\n\r\n\r\n'),
(1905,'2024-11-18','06:51:31','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24006024','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-11-18 08:01:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.67 pH\r\nmV                   302.6 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1906,'2024-11-18','06:56:13','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24006024','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-11-18 08:01:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   164.7 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1907,'2024-11-18','06:59:48','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24006024','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-11-18 08:01:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -11.4 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1908,'2024-11-18','07:04:32','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24006024','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-11-18 08:01:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.03 pH\r\nmV                   -187.4 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1909,'2024-11-18','07:42:42','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24006024','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-11-18 08:02:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.50 pH\r\nmV                   -332.5 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\nLog #                1176\r\n\r\n\r\n'),
(1910,'2024-11-18','09:07:32','90.00000','105.00000','TAZOBACTAM','NA','TZ0024110','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','1',1,'88522','Purnachandra','2024-11-18 13:18:45','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.51 pH\r\nmV                   126.1 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        100.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             18-11-24,07:16\r\n\r\n\r\n'),
(1911,'2024-11-18','10:19:41','90.00000','105.00000','Ampicillin and sulbactam for Injection','RS','AS0123009A','E4STFP24000683','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','Ampisulbactam',1,'82735','M.D. Babar Qhadri','2024-11-18 11:55:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.04 pH\r\nmV                   107.2 mV\r\nTemperature          30.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1912,'2024-11-18','11:06:01','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC MP A','TZ0424105','E4FP24000751','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','87697','M. Raghavendr Reddy',2,0,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-18 11:55:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   77.2 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\nLog #                1178\r\n\r\n\r\n'),
(1913,'2024-11-18','11:15:26','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC MP B','TZ0424105','E4FP24000751','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','87697','M. Raghavendr Reddy',2,0,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-18 11:56:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   78.2 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\nLog #                1179\r\n\r\n\r\n'),
(1914,'2024-11-18','11:22:27','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24006040','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-11-18 12:11:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.68 pH\r\nmV                   301.4 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-11-24,11:17\r\nLog #                458\r\n\r\n\r\n'),
(1915,'2024-11-18','11:27:30','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24006040','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-11-18 12:12:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.05 pH\r\nmV                   163.7 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-11-24,11:17\r\n\r\n\r\n'),
(1916,'2024-11-18','11:29:29','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24006040','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-11-18 12:12:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   -8.1 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-11-24,11:17\r\n\r\n\r\n'),
(1917,'2024-11-18','11:30:56','90.00000','105.00000','pH Buffer','NA','pH verication','E4INS24006040','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-11-18 12:12:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.02 pH\r\nmV                   -183.9 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-11-24,11:17\r\n\r\n\r\n'),
(1918,'2024-11-18','11:32:38','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24006040','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-11-18 12:13:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.43 pH\r\nmV                   -324.1 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-11-24,11:17\r\n\r\n\r\n'),
(1919,'2024-11-18','11:44:10','90.00000','105.00000','Ampicillin and Sulbactam','RS','2449102402','E4RM24006176','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205189','P. Ravi Kumar',2,0,101,'pH','RS Mobile phase A',1,'82735','M.D. Babar Qhadri','2024-11-18 11:56:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.01 pH\r\nmV                   108.7 mV\r\nTemperature          29.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1920,'2024-11-18','11:59:20','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC','PT0424124','E4FP24000753','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205058','C. Srinivasulu',2,0,101,'pH','Pipertazoinj',1,'90018','Ch. Satyam','2024-11-18 19:37:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.03 pH\r\nmV                   47.3 mV\r\nTemperature          27.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1921,'2024-11-18','12:16:42','90.00000','105.00000','Ampicillin and Sulbactam','UOD','AS0324087','E4FP24000752','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205075','G. Shyam Kumar',2,1,101,'pH','AS',1,'90018','Ch. Satyam','2024-11-18 19:15:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.99 pH\r\nmV                   109.9 mV\r\nTemperature          29.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1922,'2024-11-18','12:32:03','90.00000','105.00000','Piperacillin and Tazobactam for Injection','IDENTIFICATION','2449202465','E4RM24006000','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,0,101,'pH','Mobile phase pH',1,'90018','Ch. Satyam','2024-11-18 19:34:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   80.5 mV\r\nTemperature          34.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\nLog #                1183\r\n\r\n\r\n'),
(1923,'2024-11-18','12:41:53','90.00000','105.00000','PIPERACILLIN','NA','TZ0024110','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'88522','Purnachandra','2024-11-18 13:19:18','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   48.1 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        100.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             18-11-24,07:16\r\n\r\n\r\n'),
(1924,'2024-11-18','12:44:46','90.00000','105.00000','Ampicillin and Sulbactam for injection','pH','2449102402','E4RM24006176','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','Ampi&Sulbactam',1,'90018','Ch. Satyam','2024-11-18 14:29:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   8.97 pH\r\nmV                   -126.0 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\nLog #                1184\r\n\r\n\r\n'),
(1925,'2024-11-18','12:51:10','90.00000','105.00000','Ampicillin and Sulbactam for injection','pH','2449102403','E4RM24006177','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','Ampi&Sulbactam',1,'90018','Ch. Satyam','2024-11-18 14:29:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   8.96 pH\r\nmV                   -125.1 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1926,'2024-11-18','12:55:07','90.00000','105.00000','Ampicillin and Sulbactam for injection','pH','2449102404','E4RM24006178','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','Ampi&Sulbactam',1,'90018','Ch. Satyam','2024-11-18 14:29:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   8.97 pH\r\nmV                   -126.0 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1927,'2024-11-18','12:58:21','90.00000','105.00000','Ampicillin and Sulbactam for injection','pH','2449102673','E4RM24006179','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','Ampi&Sulbactam',1,'90018','Ch. Satyam','2024-11-18 14:30:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   8.95 pH\r\nmV                   -124.6 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1928,'2024-11-18','13:02:33','90.00000','105.00000','Ampicillin and Sulbactam for injection','pH','2449102674','E4RM24006180','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','Ampi&Sulbactam',1,'90018','Ch. Satyam','2024-11-18 14:30:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   8.98 pH\r\nmV                   -126.2 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1929,'2024-11-18','13:23:46','90.00000','105.00000','PIPERACILLIN','NA','TZ0024110','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'88522','Purnachandra','2024-11-18 14:55:16','verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.90 pH\r\nmV                   43.5 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        100.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             18-11-24,07:16\r\n\r\n\r\n'),
(1930,'2024-11-18','13:43:17','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24006262','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-18 15:03:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.29 pH\r\nmV                   89.7 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\nLog #                1189\r\n\r\n\r\n'),
(1931,'2024-11-18','13:47:07','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24006263','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-18 15:04:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   66.4 mV\r\nTemperature          23.1 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1932,'2024-11-18','13:50:31','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24006264','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-18 15:04:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   75.5 mV\r\nTemperature          23.1 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1933,'2024-11-18','13:54:03','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24006265','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-18 15:05:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   77.4 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1934,'2024-11-18','13:57:08','90.00000','105.00000','Purified water','pH','PW/SP03','E4RM24006266','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-18 15:05:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.63 pH\r\nmV                   70.2 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1935,'2024-11-18','14:00:52','90.00000','105.00000','Pure steam','pH','PSC/SP09','E4RM24006267','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'205208','S. Siddhartha Kumar Reddy','2024-11-18 15:05:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   75.2 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1936,'2024-11-18','14:03:54','90.00000','105.00000','Purified water','pH','PW/SP29','E4RM24006268','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-18 15:06:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   76.7 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1937,'2024-11-18','14:07:18','90.00000','105.00000','Water For Injection','pH','WFI/SP28','E4RM24006269','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-18 15:06:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.56 pH\r\nmV                   74.2 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1938,'2024-11-18','14:11:06','90.00000','105.00000','PIPERACILLIN','NA','TZ0024110','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'88522','Purnachandra','2024-11-18 14:55:40','verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.16 pH\r\nmV                   28.3 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        100.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             18-11-24,07:16\r\n\r\n\r\n'),
(1939,'2024-11-18','14:51:10','90.00000','105.00000','Peptone','NA','RS007/24050','E4SO24000867','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-11-18 18:06:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -13.5 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-11-24,11:17\r\nLog #                463\r\n\r\n\r\n'),
(1940,'2024-11-18','14:54:41','90.00000','105.00000','PIPERACILLIN','NA','TZ0024110','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-18 19:12:45','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.46 pH\r\nmV                   11.1 mV\r\nTemperature          26.7 C (ATC)\r\nAverage Slope        100.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             18-11-24,07:16\r\n\r\n\r\n'),
(1941,'2024-11-18','16:34:50','90.00000','105.00000','Ampicillin and Sulbactam','UOD','AS0324088','E4FP24000755','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','AS',1,'90018','Ch. Satyam','2024-11-18 19:36:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.01 pH\r\nmV                   109.3 mV\r\nTemperature          30.2 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\nLog #                1198\r\n\r\n\r\n'),
(1942,'2024-11-18','17:02:22','90.00000','105.00000','Ampicillin and Sulbactam','UOD SST','AS0324087','E4FP24000752','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205075','G. Shyam Kumar',2,1,101,'pH','AS',1,'90018','Ch. Satyam','2024-11-18 19:36:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.01 pH\r\nmV                   107.4 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1943,'2024-11-18','16:43:57','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024110','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','3',1,'84472','G. Ashok','2024-11-18 19:13:05','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.25 pH\r\nmV                   23.6 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        100.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             18-11-24,07:16\r\nLog #                333\r\n\r\n\r\n'),
(1944,'2024-11-18','17:12:03','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024110','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','3',1,'84472','G. Ashok','2024-11-18 19:13:22','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.28 pH\r\nmV                   21.4 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        100.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             18-11-24,07:16\r\n\r\n\r\n'),
(1945,'2024-11-18','17:39:43','90.00000','105.00000','Peptone','NA','RS007/24050','E4SO24000867','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-11-18 18:06:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.07 pH\r\nmV                   -12.2 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-11-24,11:17\r\n\r\n\r\n'),
(1946,'2024-11-18','18:55:38','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TR0024007','E4IP24001359','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,0,101,'pH','pH (Bulk)',1,'90018','Ch. Satyam','2024-11-18 19:36:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.89 pH\r\nmV                   55.4 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             18-11-24,06:46\r\n\r\n\r\n'),
(1947,'2024-11-19','02:50:54','90.00000','105.00000','TAZOBACTAM','NA','TR0024008','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-003','CW1-12031','1','NULL','205117','V. Seenaiah',2,0,105,'pH','1',1,'95681','T. Sivanarayana','2024-11-19 06:17:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   4.44 pH\r\nmV                   148.4 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        95.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             19-11-24,00:45\r\n\r\n\r\n'),
(1948,'2024-11-19','05:06:00','90.00000','105.00000','PIPERACILLIN','NA','TR0024008','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-003','CW1-12031','1','NULL','205007','T. Siva Ranganath',2,0,105,'pH','2',1,'95681','T. Sivanarayana','2024-11-19 06:17:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   6.40 pH\r\nmV                   38.5 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        95.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             19-11-24,00:45\r\n\r\n\r\n'),
(1949,'2024-11-19','07:13:31','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24006045','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','1.68 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-19 07:36:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.67 pH\r\nmV                   301.5 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1950,'2024-11-19','07:16:55','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24006045','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,0,101,'pH','4.01pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-19 07:36:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   163.8 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1951,'2024-11-19','07:20:07','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24006045','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,0,101,'pH','7.00pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-19 07:36:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -11.5 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1952,'2024-11-19','07:23:14','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24006045','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,0,101,'pH','10.01 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-19 07:36:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.01 pH\r\nmV                   -187.6 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1953,'2024-11-19','07:27:23','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24006045','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,0,101,'pH','12.46 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-19 07:36:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.47 pH\r\nmV                   -332.6 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1954,'2024-11-19','08:34:56','90.00000','105.00000','Piperacillin and Tazobactam For injection','Assay','TR0324001','E4FP24000763','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','89853','J.Ravi',2,0,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-11-19 11:36:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   78.4 mV\r\nTemperature          31.3 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1955,'2024-11-19','10:10:15','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24006061','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-11-19 12:30:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.68 pH\r\nmV                   298.9 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-11-24,10:03\r\n\r\n\r\n'),
(1956,'2024-11-19','10:13:01','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24006061','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-11-19 12:31:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.00 pH\r\nmV                   165.6 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-11-24,10:03\r\n\r\n\r\n'),
(1957,'2024-11-19','10:15:49','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24006061','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-11-19 12:32:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   -8.6 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-11-24,10:03\r\n\r\n\r\n'),
(1958,'2024-11-19','10:25:23','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24006061','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-11-19 12:34:34','Out of limit observed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   9.94 pH\r\nmV                   -181.5 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-11-24,10:03\r\n\r\n\r\n'),
(1959,'2024-11-19','10:29:09','90.00000','105.00000','Piperacillin and Tazobactam for injection','Cleaning  Validation','TR0324002','E4CV24000339','Cleaning Validation','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,0,101,'pH','PTCV',1,'205208','S. Siddhartha Kumar Reddy','2024-11-19 11:47:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   79.7 mV\r\nTemperature          32.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1960,'2024-11-19','10:42:17','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TR0024008','NA','BULK','PR-PHMTZ-003','CW1-12031','1','NULL','88332','P.V.A.K. Raju',2,0,105,'pH','3',1,'91408','Anand','2024-11-19 11:01:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   6.36 pH\r\nmV                   41.0 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        95.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             19-11-24,00:45\r\n\r\n\r\n'),
(1961,'2024-11-19','10:48:08','90.00000','105.00000','Ampicillin and Sulbactam','UOD','AS0324089','E4FP24000759','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205075','G. Shyam Kumar',2,0,101,'pH','AS',1,'205208','S. Siddhartha Kumar Reddy','2024-11-19 11:48:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.01 pH\r\nmV                   108.6 mV\r\nTemperature          31.9 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1962,'2024-11-19','10:54:14','90.00000','105.00000','Piperacillin And Tazobactam','IDENTIFICATION','2449201774','E4RM24006187','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205232','K. Narender Reddy',2,0,101,'pH','PT',1,'205208','S. Siddhartha Kumar Reddy','2024-11-19 11:49:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   77.8 mV\r\nTemperature          29.0 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1963,'2024-11-19','11:16:59','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TZ0024110','E4IP24001364','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,0,101,'pH','pH (Bulk)',1,'205208','S. Siddhartha Kumar Reddy','2024-11-19 11:50:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.24 pH\r\nmV                   33.5 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1964,'2024-11-19','11:33:24','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24006061','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-11-19 12:34:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.42 pH\r\nmV                   -321.8 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-11-24,10:03\r\n\r\n\r\n'),
(1965,'2024-11-19','11:58:31','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TR0024008','NA','BULK','PR-PHMTZ-003','CW1-12031','1','NULL','88332','P.V.A.K. Raju',2,0,105,'pH','3',1,'91408','Anand','2024-11-19 12:39:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   6.34 pH\r\nmV                   42.0 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        95.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             19-11-24,00:45\r\n\r\n\r\n'),
(1966,'2024-11-19','12:15:08','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324090','E4FP24000760','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,0,101,'pH','AS',1,'205208','S. Siddhartha Kumar Reddy','2024-11-19 14:09:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.19 pH\r\nmV                   -139.5 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1967,'2024-11-19','12:55:13','90.00000','105.00000','Piperacillin And Tazobactam','pH','2449203398','E4RM24006298','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,0,101,'pH','PT',1,'205208','S. Siddhartha Kumar Reddy','2024-11-19 14:09:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.44 pH\r\nmV                   22.2 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1968,'2024-11-19','12:59:36','90.00000','105.00000','Piperacillin And Tazobactam','pH','2449203407','E4RM24006299','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,0,101,'pH','PT',1,'205208','S. Siddhartha Kumar Reddy','2024-11-19 14:10:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.28 pH\r\nmV                   31.6 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1969,'2024-11-19','13:03:42','90.00000','105.00000','Piperacillin And Tazobactam','pH','2449203408','E4RM24006300','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,0,101,'pH','PT',1,'205208','S. Siddhartha Kumar Reddy','2024-11-19 14:11:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.19 pH\r\nmV                   36.8 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1970,'2024-11-19','13:43:43','90.00000','105.00000','Piperacillin And Tazobactam','RS Mobilephase','2449203398','E4RM24006298','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205266','K. Vara prasad',2,0,101,'pH','PT',1,'205208','S. Siddhartha Kumar Reddy','2024-11-19 14:12:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   3.80 pH\r\nmV                   178.7 mV\r\nTemperature          27.8 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1971,'2024-11-19','14:01:02','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24006288','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-11-19 15:32:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.48 pH\r\nmV                   78.2 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1972,'2024-11-19','14:04:33','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24006289','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-11-19 15:34:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.55 pH\r\nmV                   74.0 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1973,'2024-11-19','14:07:20','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24006291','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-11-19 15:34:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   76.0 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1974,'2024-11-19','14:10:24','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24006292','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-11-19 15:35:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   74.9 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1975,'2024-11-19','14:13:41','90.00000','105.00000','Water For Injection','pH','WFI/SP13','E4RM24006293','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-11-19 15:35:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.48 pH\r\nmV                   78.1 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1976,'2024-11-19','14:16:35','90.00000','105.00000','Pure steam','pH','PSC/SP10','E4RM24006294','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'84378','Shivanand. W','2024-11-19 15:36:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.48 pH\r\nmV                   78.2 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1977,'2024-11-19','14:20:13','90.00000','105.00000','Purified water','pH','PW/SP30','E4RM24006295','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-11-19 15:36:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   75.2 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1978,'2024-11-19','14:23:35','90.00000','105.00000','Purified water','pH','PW/SP50','E4RM24006296','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-11-19 15:36:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   76.0 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\nLog #                1225\r\n\r\n\r\n'),
(1979,'2024-11-19','14:28:16','90.00000','105.00000','Water For Injection','pH','WFI/SP36','E4RM24006297','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-11-19 15:37:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   75.6 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1980,'2024-11-19','15:39:17','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24006061','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-11-19 17:10:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.72 pH\r\nmV                   297.1 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        96.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             19-11-24,14:37\r\nLog #                470\r\n\r\n\r\n'),
(1981,'2024-11-19','15:47:38','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24006061','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-11-19 17:10:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   169.5 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        96.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             19-11-24,14:37\r\nLog #                471\r\n\r\n\r\n'),
(1982,'2024-11-19','15:49:19','90.00000','105.00000','Piperacillin And Tazobactam','ASSAY','2449203398','E4RM24006298','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205048','P. Nagesh',2,0,101,'pH','PT',1,'84378','Shivanand. W','2024-11-19 16:52:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.49 pH\r\nmV                   81.0 mV\r\nTemperature          36.7 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\nLog #                1227\r\n\r\n\r\n'),
(1983,'2024-11-19','16:10:54','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24006061','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-11-19 17:10:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -4.2 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        96.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             19-11-24,14:37\r\nLog #                472\r\n\r\n\r\n'),
(1984,'2024-11-19','16:14:04','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TR0324002','E4PP24002259','Process performance qualification','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PPQ pH',1,'84378','Shivanand. W','2024-11-19 16:54:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.81 pH\r\nmV                   59.1 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1985,'2024-11-19','16:17:13','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24006061','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-11-19 17:11:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.03 pH\r\nmV                   -175.5 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        96.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             19-11-24,14:37\r\nLog #                473\r\n\r\n\r\n'),
(1986,'2024-11-19','16:17:13','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TR0324001','E4PP24002260','Process performance qualification','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PPQ pH',1,'84378','Shivanand. W','2024-11-19 16:55:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   77.6 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\nLog #                1229\r\n\r\n\r\n'),
(1987,'2024-11-19','16:21:29','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TR0324001','E4PP24002261','Process performance qualification','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PPQ pH',1,'84378','Shivanand. W','2024-11-19 16:55:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.55 pH\r\nmV                   74.6 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1988,'2024-11-19','16:23:32','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24006061','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-11-19 17:11:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.44 pH\r\nmV                   -312.8 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        96.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             19-11-24,14:37\r\nLog #                474\r\n\r\n\r\n'),
(1989,'2024-11-19','16:24:19','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TR0324001','E4PP24002262','Process performance qualification','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PPQ pH',1,'84378','Shivanand. W','2024-11-19 16:55:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.48 pH\r\nmV                   78.3 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\nLog #                1231\r\n\r\n\r\n'),
(1990,'2024-11-19','16:31:17','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TR0324002','E4PP24002346','Process performance qualification','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PPQ pH',1,'84378','Shivanand. W','2024-11-19 16:56:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.81 pH\r\nmV                   59.1 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1991,'2024-11-19','16:33:50','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TR0324002','E4PP24002347','Process performance qualification','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PPQ pH',1,'84378','Shivanand. W','2024-11-19 17:06:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   58.0 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1992,'2024-11-19','17:00:34','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TR0024008','E4IP24001369','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','pH (Bulk)',1,'84378','Shivanand. W','2024-11-19 17:07:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.19 pH\r\nmV                   36.4 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             19-11-24,07:10\r\n\r\n\r\n'),
(1993,'2024-11-19','18:20:56','90.00000','105.00000','FLUID-A','NA','RS012/24013','E4SO24000872','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Fluid-A',1,'85152','V. Chandra Shekara Achari','2024-11-19 19:37:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -3.0 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        96.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             19-11-24,14:37\r\n\r\n\r\n'),
(1994,'2024-11-19','18:23:09','90.00000','105.00000','Fluid-D','NA','RS010/24011','E4SO24000871','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Fluid-D',1,'85152','V. Chandra Shekara Achari','2024-11-19 19:37:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.06 pH\r\nmV                   -5.3 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        96.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             19-11-24,14:37\r\n\r\n\r\n'),
(1995,'2024-11-19','22:10:37','90.00000','105.00000','FLUID-A','NA','RS012/24013','E4SO24000872','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Fluid-A',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.05 pH\r\nmV                   -4.8 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        96.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             19-11-24,14:37\r\n\r\n\r\n'),
(1996,'2024-11-19','22:17:58','90.00000','105.00000','Fluid-D','NA','RS010/24011','E4SO24000871','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Fluid-D',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.08 pH\r\nmV                   -6.6 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        96.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             19-11-24,14:37\r\n\r\n\r\n'),
(1997,'2024-11-20','03:37:23','90.00000','105.00000','TAZOBACTAM','NA','TZ0024111','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','92936','Sameer',2,0,104,'pH','1',1,'95681','T. Sivanarayana','2024-11-20 06:25:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.60 pH\r\nmV                   121.7 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             20-11-24,00:35\r\n\r\n\r\n'),
(1998,'2024-11-20','07:05:14','90.00000','105.00000','pH Buffer','verification','NA','E4INS24006078','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-11-20 07:57:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.67 pH\r\nmV                   302.0 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             20-11-24,07:01\r\n\r\n\r\n'),
(1999,'2024-11-20','07:08:28','90.00000','105.00000','pH Buffer','verification','NA','E4INS24006078','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-11-20 07:57:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   164.3 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             20-11-24,07:01\r\n\r\n\r\n'),
(2000,'2024-11-20','07:12:14','90.00000','105.00000','pH Buffer','verification','NA','E4INS24006078','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-11-20 07:57:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -11.1 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             20-11-24,07:01\r\n\r\n\r\n'),
(2001,'2024-11-20','07:15:57','90.00000','105.00000','pH Buffer','verification','NA','E4INS24006078','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-11-20 07:57:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.01 pH\r\nmV                   -186.7 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             20-11-24,07:01\r\n\r\n\r\n'),
(2002,'2024-11-20','07:19:06','90.00000','105.00000','pH Buffer','verification','NA','E4INS24006078','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-11-20 07:57:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.48 pH\r\nmV                   -331.6 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             20-11-24,07:01\r\n\r\n\r\n'),
(2003,'2024-11-20','08:53:13','90.00000','105.00000','PIPERACILLIN','NA','TZ0024111','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','2',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.88 pH\r\nmV                   46.8 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             20-11-24,00:35\r\n\r\n\r\n'),
(2004,'2024-11-20','09:37:24','90.00000','105.00000','PIPERACILLIN','NA','TZ0024111','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','2',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.90 pH\r\nmV                   45.8 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             20-11-24,00:35\r\n\r\n\r\n'),
(2005,'2024-11-20','09:54:01','90.00000','105.00000','Piperacillin and Tazobactam for injection','Cleaning Verification','TR0324003','E4CV24000342','Cleaning Validation','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,0,101,'pH','PTCV',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   79.1 mV\r\nTemperature          29.9 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             20-11-24,07:01\r\n\r\n\r\n'),
(2006,'2024-11-20','10:04:19','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS  MP_A','TZ0424105','E4FP24000751','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205043','T.V. Subbaiah',2,0,101,'pH','Pipertazoinj',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   76.2 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             20-11-24,07:01\r\n\r\n\r\n'),
(2007,'2024-11-20','10:15:21','90.00000','105.00000','Ampicillin and Sulbactam for injection','ID','2449102395','E4RM24006214','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','90017','G. Srishailam',2,0,101,'pH','Ampi&Sulbactam',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.00 pH\r\nmV                   109.5 mV\r\nTemperature          32.0 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             20-11-24,07:01\r\n\r\n\r\n'),
(2008,'2024-11-20','10:22:05','90.00000','105.00000','PIPERACILLIN','NA','TZ0024111','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','2',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.24 pH\r\nmV                   26.2 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             20-11-24,00:35\r\n\r\n\r\n'),
(2009,'2024-11-20','10:27:59','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS_MP_B','TZ0424105','E4FP24000751','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205043','T.V. Subbaiah',2,0,101,'pH','Pipertazoinj',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   76.9 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             20-11-24,07:01\r\n\r\n\r\n'),
(2010,'2024-11-20','10:40:20','90.00000','105.00000','Piperacillin and Tazobactam for Injection','RS','PN0324079A','E4STFP24000703','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','PIPERTAZO',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.99 pH\r\nmV                   50.5 mV\r\nTemperature          31.9 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             20-11-24,07:01\r\n\r\n\r\n'),
(2011,'2024-11-20','11:04:04','90.00000','105.00000','PIPERACILLIN','NA','TZ0024111','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','2',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.45 pH\r\nmV                   13.6 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             20-11-24,00:35\r\n\r\n\r\n'),
(2012,'2024-11-20','11:15:14','90.00000','105.00000','Piperacillin and Tazobactam For injection','Mobile phase','TR0324001','E4FP24000763','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,0,101,'pH','Pipertazoinj',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.00 pH\r\nmV                   57.9 mV\r\nTemperature          29.4 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(2013,'2024-11-20','11:43:59','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24006314','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,0,101,'pH','pH Test',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.71 pH\r\nmV                   73.4 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\nLog #                1247\r\n\r\n\r\n'),
(2014,'2024-11-20','11:49:02','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24006315','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,0,101,'pH','pH Test',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   74.6 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\nLog #                1248\r\n\r\n\r\n'),
(2015,'2024-11-20','11:51:06','90.00000','105.00000','TAZOBACTAM','NA','TR0024009','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-003','CW1-12031','1','NULL','205169','M. Venkatesham',2,0,105,'pH','1',1,'91408','Anand','2024-11-20 12:11:17','verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   4.49 pH\r\nmV                   147.9 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        95.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             20-11-24,05:47\r\n\r\n\r\n'),
(2016,'2024-11-20','11:53:14','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24006316','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,0,101,'pH','pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.66 pH\r\nmV                   76.0 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\nLog #                1249\r\n\r\n\r\n'),
(2017,'2024-11-20','11:57:51','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24006317','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,0,101,'pH','pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.73 pH\r\nmV                   71.9 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(2018,'2024-11-20','12:00:10','90.00000','105.00000','Purified water','pH','PW/SP05','E4RM24006318','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,0,101,'pH','pH Test',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   71.8 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(2019,'2024-11-20','12:03:11','90.00000','105.00000','Pure steam','pH','PSC/SP20','E4RM24006319','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,0,101,'pH','pH_Pure Steam',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.95 pH\r\nmV                   59.5 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(2020,'2024-11-20','12:05:54','90.00000','105.00000','Purified water','pH','PW/SP31','E4RM24006320','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,0,101,'pH','pH Test',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.94 pH\r\nmV                   59.8 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(2021,'2024-11-20','12:09:01','90.00000','105.00000','Water For Injection','pH','WFI/SP37','E4RM24006321','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,0,101,'pH','pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   74.3 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(2022,'2024-11-20','12:12:21','90.00000','105.00000','Water For Injection','pH','WFI/SP42','E4RM24006322','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,0,101,'pH','pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   74.5 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(2023,'2024-11-20','12:23:45','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024111','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','3',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.20 pH\r\nmV                   28.6 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             20-11-24,00:35\r\n\r\n\r\n'),
(2024,'2024-11-20','12:30:01','90.00000','105.00000','pH Buffer','NA','NA','	E4INS24006094','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','1.68 pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.65 pH\r\nmV                   307.9 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             20-11-24,12:22\r\nLog #                479\r\n\r\n\r\n'),
(2025,'2024-11-20','12:32:43','90.00000','105.00000','pH Buffer','NA','NA','E4INS24006094','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','4.01pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   169.3 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             20-11-24,12:22\r\n\r\n\r\n'),
(2026,'2024-11-20','12:34:12','90.00000','105.00000','pH Buffer','NA','NA','E4INS24006094','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','7.00pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -6.2 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             20-11-24,12:22\r\n\r\n\r\n'),
(2027,'2024-11-20','12:36:19','90.00000','105.00000','pH Buffer','NA','NA','E4INS24006094','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','10.01 pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.01 pH\r\nmV                   -180.8 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             20-11-24,12:22\r\n\r\n\r\n'),
(2028,'2024-11-20','12:42:00','90.00000','105.00000','pH Buffer','NA','NA','E4INS24006094','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','12.46 pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.44 pH\r\nmV                   -323.4 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             20-11-24,12:22\r\n\r\n\r\n');

/*Table structure for table `tbl_weighment_master_ph_ise` */

