
CREATE TABLE `tbl_audit_users` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `dt` date NOT NULL,
  `tm` time NOT NULL,
  `userid` varchar(50) NOT NULL DEFAULT 'NULL',
  `username` varchar(255) NOT NULL DEFAULT 'NULL',
  `OldRole` varchar(50) NOT NULL DEFAULT 'NA',
  `NewRole` varchar(50) NOT NULL DEFAULT 'NA',
  `ACT` varchar(50) NOT NULL DEFAULT 'NULL',
  `KeyCode` varchar(50) NOT NULL DEFAULT 'NA',
  `KeyValue` varchar(50) NOT NULL DEFAULT 'NA',
  `RightAdded` varchar(5000) NOT NULL DEFAULT 'NA',
  `RightRemoved` varchar(5000) NOT NULL DEFAULT 'NA',
  `Remark` varchar(500) NOT NULL DEFAULT 'NULL',
  `OldDepartment` varchar(150) DEFAULT 'NA',
  `NewDepartment` varchar(150) DEFAULT 'NA',
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB AUTO_INCREMENT=93 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_audit_users` */

insert  into `tbl_audit_users`(`RecNo`,`dt`,`tm`,`userid`,`username`,`OldRole`,`NewRole`,`ACT`,`KeyCode`,`KeyValue`,`RightAdded`,`RightRemoved`,`Remark`,`OldDepartment`,`NewDepartment`) values 
(1,'2024-09-12','13:08:18','85395','Ravi Kumar','NA','Service engineer','New User Added','V-trident','Trident','NA','NA','AS part of OQ New user ID created','NA','Service'),
(2,'2024-09-12','14:46:12','85395','Ravi Kumar','Service engineer','Service engineer','User Temporarily Disabled','V-trident','Trident','NA','NA','AS pert of OQ User ID temporary disabled','Service','Service'),
(3,'2024-09-12','14:49:05','85395','Ravi Kumar','Service engineer','Service engineer','User Enabled','V-trident','Trident','NA','NA','As part of OQ User ID enabled','Service','Service'),
(4,'2024-09-12','14:52:43','85395','Ravi Kumar','Service engineer','Service engineer','User Edited','V-trident','Trident','NA','Edit User','AS part of OQ edit user roles','Service','Service'),
(5,'2024-09-12','16:06:32','85395','Ravi Kumar','Service engineer','Service engineer','User Edited','V-trident','Trident','Edit User','NA','As part of OQ','Service','Service'),
(6,'2024-09-27','17:13:44','85395','Ravi Kumar','NA','Analyst','New User Added','205066','Sreeram Jakkaraiah','NA','NA','New user ID was created as per Approved request','NA','QC-Chemical'),
(7,'2024-09-27','17:17:56','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','205208','S. Siddhartha Kumar Reddy','NA','NA','New User ID was created as per approved request','NA','QC-Chemical'),
(8,'2024-09-27','17:19:19','85395','Ravi Kumar','NA','Manager-QC','New User Added','89007','N. Gopi Krishna','NA','NA','New User ID was created as per approved request','NA','QC-Chemical'),
(9,'2024-09-27','17:40:05','85395','Ravi Kumar','NA','Analyst','New User Added','88599','MD Mahamad','NA','NA','New User ID was created as per approved request','NA','QC-Chemical'),
(10,'2024-09-27','17:41:09','85395','Ravi Kumar','NA','Analyst','New User Added','88571','V. Ramu','NA','NA','New User ID was created as per approved request','NA','QC-Chemical'),
(11,'2024-09-27','17:42:26','85395','Ravi Kumar','NA','Analyst','New User Added','205090','M. Srikrishnudu','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(12,'2024-09-27','17:43:22','85395','Ravi Kumar','NA','Analyst','New User Added','205075','G. Shyam Kumar','NA','NA','New User ID was created as per approved request','NA','QC-Chemical'),
(13,'2024-09-27','17:44:20','85395','Ravi Kumar','NA','Analyst','New User Added','205089','M. Venkata Ramana','NA','NA','New User ID was created as per approved request','NA','QC-Chemical'),
(14,'2024-09-27','17:45:41','85395','Ravi Kumar','NA','Analyst','New User Added','205216','C. narsimulu','NA','NA','New User ID was created as per approved request','NA','QC-Chemical'),
(15,'2024-09-27','17:46:56','85395','Ravi Kumar','NA','Analyst','New User Added','205247','R. Puneetha','NA','NA','New User ID was created as per approved request','NA','QC-Chemical'),
(16,'2024-09-27','17:47:57','85395','Ravi Kumar','NA','Analyst','New User Added','89180','R. Harish Reddy','NA','NA','New User ID was created as per approved request','NA','QC-Chemical'),
(17,'2024-09-27','17:49:15','85395','Ravi Kumar','NA','Analyst','New User Added','205210','B. Anitha','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(18,'2024-09-27','17:50:14','85395','Ravi Kumar','NA','Analyst','New User Added','80903','D. naga Prasad','NA','NA','New User ID was created as per approved request','NA','QC-Chemical'),
(19,'2024-09-27','17:51:10','85395','Ravi Kumar','NA','Analyst','New User Added','87697','M. Raghavendr Reddy','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(20,'2024-09-27','17:52:08','85395','Ravi Kumar','NA','Analyst','New User Added','205189','P. Ravi Kumar','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(21,'2024-09-27','17:53:07','85395','Ravi Kumar','NA','Analyst','New User Added','205058','C. Srinivasulu','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(22,'2024-09-27','17:55:00','85395','Ravi Kumar','NA','Analyst','New User Added','205211','S. Sailatha','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(23,'2024-09-30','09:31:25','85395','Ravi Kumar','NA','Administrator','New User Added','91011','K. Vishwa Mohan Reddy','NA','NA','New Administrator User ID was created as per approved request.','NA','Production'),
(24,'2024-09-30','09:36:53','85395','Ravi Kumar','NA','Administrator','New User Added','205170','P. Satish','NA','NA','New Administrator user ID created as per approved request.','NA','Production'),
(25,'2024-09-30','14:52:23','85395','Ravi Kumar','NA','Analyst','New User Added','95681','T. Sivanarayana','NA','NA','New user ID created as per request','NA','Production'),
(26,'2024-09-30','14:53:23','85395','Ravi Kumar','NA','Analyst','New User Added','205190','B. Prabhudas','NA','NA','New user ID created as per approved request','NA','Production'),
(27,'2024-09-30','15:08:05','85395','Ravi Kumar','Analyst','Manager-QC','User Edited','95681','T. Sivanarayana','Print Audit Trail,Edit Product Type,View Activity Log,Add Product Type,View Audit Trail','View Report(Self),Reprint All Report,Reprint Report(Self),Calibration,Test,Verify Report','User Role updated as per request','Production','Production'),
(28,'2024-09-30','15:09:08','85395','Ravi Kumar','NA','Analyst','New User Added','84342','G.S.A. Naidu','NA','NA','New user ID created as per approved request','NA','Production'),
(29,'2024-09-30','15:10:11','85395','Ravi Kumar','NA','Analyst','New User Added','205223','Jithendra','NA','NA','New user ID created as per approved request','NA','Production'),
(30,'2024-09-30','15:11:18','85395','Ravi Kumar','NA','Analyst','New User Added','205006','A. Srikanth','NA','NA','New user ID created as per approved request','NA','Production'),
(31,'2024-09-30','15:12:24','85395','Ravi Kumar','NA','Analyst','New User Added','88332','P.V.A.K. Raju','NA','NA','New user ID created as per approved request','NA','Production'),
(32,'2024-09-30','15:13:26','85395','Ravi Kumar','NA','Analyst','New User Added','205169','M. Venkatesham','NA','NA','New user ID was created as per approved request','NA','Production'),
(33,'2024-09-30','15:14:25','85395','Ravi Kumar','NA','Analyst','New User Added','92595','K. Murali Krishna','NA','NA','New User ID was created as per approved request','NA','Production'),
(34,'2024-09-30','15:15:43','85395','Ravi Kumar','NA','Analyst','New User Added','205104','Siva Shankar Reddy','NA','NA','New user ID was created as per approved request','NA','Production'),
(35,'2024-09-30','15:33:28','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','90018','Ch. Satyam','NA','NA','New user ID created as per approved request','NA','QC-Chemical'),
(36,'2024-09-30','15:34:22','85395','Ravi Kumar','NA','Analyst','New User Added','89853','J.Ravi','NA','NA','New user ID created as per approved request','NA','QC-Chemical'),
(37,'2024-09-30','15:35:15','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','90778','B. Bhaskar','NA','NA','New user ID created as per approved request','NA','QC-Chemical'),
(38,'2024-09-30','15:36:14','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','82735','M.D. Babar Qhadri','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(39,'2024-09-30','15:37:10','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','86253','D. Jayapal Reddy','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(40,'2024-09-30','15:38:06','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','84378','Shivanand. W','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(41,'2024-09-30','15:39:51','85395','Ravi Kumar','NA','Analyst','New User Added','205196','P. Mounika','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(42,'2024-09-30','15:41:29','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','84914','D. Vital','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(43,'2024-09-30','15:42:25','85395','Ravi Kumar','NA','Analyst','New User Added','88819','M. Ramya','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(44,'2024-09-30','15:43:23','85395','Ravi Kumar','NA','Analyst','New User Added','205207','P. Hanmanthu','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(45,'2024-09-30','15:45:02','85395','Ravi Kumar','NA','Analyst','New User Added','95422','M. Shankar','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(46,'2024-09-30','15:45:58','85395','Ravi Kumar','NA','Analyst','New User Added','92513','Ch. Chinna Rao','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(47,'2024-09-30','16:10:10','85395','Ravi Kumar','NA','Manager-QC','New User Added','88588','. Srisailam','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(48,'2024-09-30','16:52:20','85395','Ravi Kumar','NA','Analyst','New User Added','205007','T. Siva Ranganath','NA','NA','New User ID was created as per approved request','NA','Production'),
(49,'2024-09-30','16:54:08','85395','Ravi Kumar','NA','Analyst','New User Added','85705','P. Prakash Rao','NA','NA','New user ID was created as per approved request','NA','Production'),
(50,'2024-09-30','16:55:13','85395','Ravi Kumar','NA','Analyst','New User Added','205117','V. Seenaiah','NA','NA','New User ID was created as per approved request','NA','Production'),
(51,'2024-09-30','16:56:12','85395','Ravi Kumar','NA','Analyst','New User Added','91700','Ch. Ramarao','NA','NA','New User ID was created as per approved request','NA','Production'),
(52,'2024-09-30','17:23:27','85395','Ravi Kumar','NA','Analyst','New User Added','205187','B. Ramvikas','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(53,'2024-09-30','17:24:40','85395','Ravi Kumar','NA','Analyst','New User Added','205252','Y. Sivaiah','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(54,'2024-09-30','17:35:04','85395','Ravi Kumar','NA','Supervisor','New User Added','88515','Prashanth B','NA','NA','New user ID was created as per approved request','NA','Production'),
(55,'2024-09-30','17:46:39','85395','Ravi Kumar','NA','Supervisor','New User Added','88629','Raghuram Reddy','NA','NA','New user ID was created as per approved request','NA','Production'),
(56,'2024-09-30','17:48:08','85395','Ravi Kumar','NA','Supervisor','New User Added','84472','G. Ashok','NA','NA','New user ID was created as per approved request','NA','Production'),
(57,'2024-09-30','17:49:14','85395','Ravi Kumar','NA','Supervisor','New User Added','205044','Prem kumar','NA','NA','New user ID was created as per approved request','NA','Production'),
(58,'2024-10-01','07:17:13','91011','K. Vishwa Mohan Reddy','Supervisor','Analyst','User Edited','205044','Prem kumar','Reprint All Report,Reprint Report(Self),Calibration','Print Audit Trail,View Audit Trail,View Activity Log,Add Product Type,Edit Product Type','updated','Production','Production'),
(59,'2024-10-01','07:18:27','91011','K. Vishwa Mohan Reddy','Manager-QC','Supervisor','User Edited','95681','T. Sivanarayana','NA','Calibration','updated','Production','Production'),
(60,'2024-10-01','10:42:34','85395','Ravi Kumar','NA','Analyst','New User Added','88655','V. Prashanth Kumar','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(61,'2024-10-01','10:43:07','85395','Ravi Kumar','NA','Analyst','New User Added','205048','P. Nagesh','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(62,'2024-10-01','11:45:31','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','205143','Ch. Prasad','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(63,'2024-10-01','11:46:27','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','88842','R. Mallesh','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(64,'2024-10-03','11:27:39','85395','Ravi Kumar','NA','Analyst','New User Added','205119','S.M. Subhani','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(65,'2024-10-03','11:29:17','85395','Ravi Kumar','NA','Analyst','New User Added','205043','T.V. Subbaiah','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(66,'2024-10-03','17:00:59','85395','Ravi Kumar','NA','Analyst','New User Added','205232','K. Narender Reddy','NA','NA','New user ID created as per approved request','NA','QC-Chemical'),
(67,'2024-10-03','17:02:37','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','85152','V. Chandra Shekara Achari','NA','NA','New User ID created as per approved request','NA','QC-Chemical'),
(68,'2024-10-05','15:40:10','85395','Ravi Kumar','NA','Manager-QC','New User Added','89454','L. Veera babu','NA','NA','New user ID created as per approved request','NA','QC-Chemical'),
(69,'2024-10-05','15:41:04','85395','Ravi Kumar','NA','Manager-QC','New User Added','82416','B. Somasekhar','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(70,'2024-10-05','15:42:10','85395','Ravi Kumar','NA','Manager-QC','New User Added','82528','M. Sureshbabu','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(71,'2024-10-05','17:17:57','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','89384','M.V.V. Satish','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(72,'2024-10-07','11:10:08','85395','Ravi Kumar','NA','Analyst','New User Added','90017','G. Srishailam','NA','NA','New User ID was created as per approved request','NA','QC-Chemical'),
(73,'2024-10-07','11:58:51','85395','Ravi Kumar','NA','Analyst','New User Added','205266','K. Vara prasad','NA','NA','New User ID created as per approved request','NA','QC-Chemical'),
(74,'2024-10-07','15:07:34','85395','Ravi Kumar','NA','Analyst','New User Added','87624','N. Nani babu','NA','NA','New user ID was created as per approved request','NA','QC-Chemical'),
(75,'2024-10-08','16:01:36','205170','P. Satish','Analyst','Analyst','User Edited','205104','N. Siva Sekhar Reddy','NA','NA','User details updated','Production','Production'),
(76,'2024-10-11','09:05:17','85395','Ravi Kumar','NA','Analyst','New User Added','205224','K. Uma Mahesh','NA','NA','New user ID was created as per approved request','NA','Production'),
(77,'2024-10-15','09:57:42','85395','Ravi Kumar','NA','Analyst','New User Added','92936','Sameer','NA','NA','New user ID created as per approved request','NA','Production'),
(78,'2024-10-18','12:39:20','85395','Ravi Kumar','NA','Analyst','New User Added','92574','Abhinav.G','NA','NA','New user created as per approved request','NA','QC-Chemical'),
(79,'2024-11-01','17:41:53','85395','Ravi Kumar','NA','Analyst','New User Added','205269','P. Srinivas varma','NA','NA','New user ID createdas per request','NA','QC-Chemical'),
(80,'2024-11-05','15:39:16','85395','Ravi Kumar','NA','Analyst','New User Added','86252','N.V.S.R.K.Raju','NA','NA','New user ID created as per approved request','NA','QC-Chemical'),
(81,'2024-11-12','08:58:56','85395','Ravi Kumar','NA','Supervisor','New User Added','82546','A. Avinash','NA','NA','New user ID created as per approved request ','NA','Production'),
(82,'2024-11-12','09:00:06','85395','Ravi Kumar','NA','Supervisor','New User Added','205267','B.H. Hemanth kumar','NA','NA','New user ID created as per approved request','NA','Production'),
(83,'2024-11-12','15:14:04','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','91406','Sridhar','NA','NA','New user creation','NA','Production'),
(84,'2024-11-12','15:14:50','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','91408','Anand','NA','NA','New user creation','NA','Production'),
(85,'2024-11-12','15:23:38','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','90358','Krishna','NA','NA','new id creation','NA','Production'),
(86,'2024-11-12','15:24:16','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','50654','A Suresh','NA','NA','new id creation','NA','Production'),
(87,'2024-11-12','15:25:16','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','88522','Purnachandra','NA','NA','new id creation','NA','Production'),
(88,'2024-11-12','15:26:06','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','84534','M Viswa','NA','NA','new id creation','NA','Production'),
(89,'2024-11-12','15:26:55','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','90365','Abhilash','NA','NA','new id creation','NA','Production'),
(90,'2024-11-12','15:27:58','85395','Ravi Kumar','NA','Reviewer-QA','New User Added','205240','R Anil kumar','NA','NA','new id creation','NA','Production'),
(91,'2024-11-15','15:58:50','85395','Ravi Kumar','NA','Supervisor','New User Added','205192','L. Venkata naga prasad','NA','NA','New user ID created as per approved request','NA','Production'),
(92,'2024-12-28','11:52:55','11','85395@tantrasoftsolutions.com','Analyst','Service engineer','User Edited','2','205066@CORP.AUROBINDO.COM','Set All Parameters,Add User,Manage User,Edit User,Add Department,Edit Department,Edit Role,Activate / Deactivate,View Audit Trail,Print Audit Trail,View Activity Log,Add Meter,Add Product Type,Edit Product Type,Console Setting,Edit Meter,Verify Report','Print Report','asfsf','QC-Chemical','QC-Chemical');

/*Table structure for table `tbl_biometric` */


CREATE TABLE `tbl_biometric` (
  `id` int(11) NOT NULL AUTO_INCREMENT,
  `userId` varchar(50) DEFAULT NULL,
  `template1` blob,
  `template_id` int(50) NOT NULL,
  KEY `id` (`id`)
) ENGINE=InnoDB DEFAULT CHARSET=utf8;

/*Data for the table `tbl_biometric` */

/*Table structure for table `tbl_calibration_detail_conductivity` */


CREATE TABLE `tbl_calibration_detail_conductivity` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `RepSerNo` int(11) DEFAULT NULL,
  `RecSeqNo` int(11) DEFAULT NULL,
  `Conductivity` varchar(100) DEFAULT '0',
  `Conductance` varchar(100) DEFAULT '0',
  `Temp` varchar(100) DEFAULT '0',
  `CalibType` varchar(50) DEFAULT 'NULL',
  `K` varchar(100) DEFAULT '0',
  `AvgK` varchar(100) DEFAULT '0',
  `Offset` varchar(100) DEFAULT '0',
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_calibration_detail_conductivity` */

/*Table structure for table `tbl_calibration_detail_ph` */


CREATE TABLE `tbl_calibration_detail_ph` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `RepSerNo` int(11) DEFAULT NULL,
  `RecSeqNo` int(11) DEFAULT NULL,
  `Slope` varchar(100) DEFAULT '0',
  `Eo` varchar(100) DEFAULT '0',
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB AUTO_INCREMENT=840 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_calibration_detail_ph` */

insert  into `tbl_calibration_detail_ph`(`RecNo`,`RepSerNo`,`RecSeqNo`,`Slope`,`Eo`) values 
(1,12,1,'98.5 %\r','1.6 mV\r'),
(2,12,2,'101.3 %\r','-3.5 mV\r'),
(3,12,3,'95.7 %\r','-3.5 mV\r'),
(4,12,4,'98.8 %\r','2.1 mV\r'),
(5,13,1,'99.7 %\r','0.1 mV\r'),
(6,13,2,'99.7 %\r','0.2 mV\r'),
(7,13,3,'97.1 %\r','0.2 mV\r'),
(8,13,4,'100.8 %\r','6.9 mV\r'),
(9,14,1,'99.9 %\r','-5.7 mV\r'),
(10,14,2,'98.9 %\r','-4.0 mV\r'),
(11,14,3,'94.7 %\r','-4.0 mV\r'),
(12,14,4,'97.7 %\r','1.4 mV\r'),
(13,15,1,'102.0 %\r','-5.9 mV\r'),
(14,15,2,'98.5 %\r','0.4 mV\r'),
(15,15,3,'96.6 %\r','0.4 mV\r'),
(16,15,4,'101.7 %\r','9.5 mV\r'),
(17,16,1,'98.5 %\r','3.8 mV\r'),
(18,16,2,'100.2 %\r','0.8 mV\r'),
(19,16,3,'97.0 %\r','0.8 mV\r'),
(20,16,4,'100.5 %\r','7.0 mV\r'),
(21,17,1,'100.3 %\r','2.8 mV\r'),
(22,17,2,'83.2 %\r','33.1 mV\r'),
(23,17,3,'113.0 %\r','32.9 mV\r'),
(24,17,4,'97.3 %\r','4.9 mV\r'),
(25,18,1,'100.6 %\r','1.5 mV\r'),
(26,18,2,'99.3 %\r','3.7 mV\r'),
(27,18,3,'97.3 %\r','3.7 mV\r'),
(28,18,4,'95.3 %\r','0.0 mV\r'),
(29,19,1,'102.8 %\r','-8.0 mV\r'),
(30,19,2,'96.0 %\r','3.9 mV\r'),
(31,19,3,'95.0 %\r','4.0 mV\r'),
(32,19,4,'99.2 %\r','11.3 mV\r'),
(33,20,1,'100.8 %\r','-4.9 mV\r'),
(34,20,2,'96.6 %\r','2.5 mV\r'),
(35,20,3,'97.1 %\r','2.5 mV\r'),
(36,20,4,'99.1 %\r','6.1 mV\r'),
(37,21,1,'100.5 %\r','-5.3 mV\r'),
(38,21,2,'99.0 %\r','-2.6 mV\r'),
(39,21,3,'95.9 %\r','-2.6 mV\r'),
(40,21,4,'102.2 %\r','8.6 mV\r'),
(41,22,1,'105.9 %\r','-7.4 mV\r'),
(42,22,2,'99.1 %\r','4.7 mV\r'),
(43,22,3,'96.5 %\r','4.7 mV\r'),
(44,22,4,'101.0 %\r','12.7 mV\r'),
(45,23,1,'99.9 %\r','3.9 mV\r'),
(46,23,2,'99.4 %\r','4.9 mV\r'),
(47,23,3,'97.2 %\r','4.9 mV\r'),
(48,23,4,'98.7 %\r','7.6 mV\r'),
(49,24,1,'102.4 %\r','-8.2 mV\r'),
(50,24,2,'96.1 %\r','2.9 mV\r'),
(51,24,3,'96.6 %\r','2.9 mV\r'),
(52,24,4,'100.0 %\r','8.8 mV\r'),
(53,25,1,'101.0 %\r','-3.4 mV\r'),
(54,25,2,'99.1 %\r','-0.0 mV\r'),
(55,25,3,'94.8 %\r','0.0 mV\r'),
(56,25,4,'104.8 %\r','17.7 mV\r'),
(57,26,1,'99.5 %\r','4.2 mV\r'),
(58,26,2,'99.1 %\r','4.8 mV\r'),
(59,26,3,'96.9 %\r','4.8 mV\r'),
(60,26,4,'100.3 %\r','10.8 mV\r'),
(61,27,1,'99.6 %\r','-1.5 mV\r'),
(62,27,2,'98.9 %\r','-0.3 mV\r'),
(63,27,3,'94.3 %\r','-0.3 mV\r'),
(64,27,4,'106.1 %\r','20.7 mV\r'),
(65,28,1,'100.7 %\r','-6.6 mV\r'),
(66,28,2,'98.5 %\r','-2.9 mV\r'),
(67,28,3,'96.3 %\r','-2.8 mV\r'),
(68,28,4,'100.9 %\r','5.2 mV\r'),
(69,29,1,'101.2 %\r','1.2 mV\r'),
(70,29,2,'99.2 %\r','4.6 mV\r'),
(71,29,3,'97.0 %\r','4.6 mV\r'),
(72,29,4,'100.3 %\r','10.5 mV\r'),
(73,30,1,'104.8 %\r','-33.7 mV\r'),
(74,30,2,'96.2 %\r','-18.6 mV\r'),
(75,30,3,'96.3 %\r','-18.6 mV\r'),
(76,30,4,'101.0 %\r','-10.1 mV\r'),
(77,31,1,'100.9 %\r','-5.3 mV\r'),
(78,31,2,'99.1 %\r','-2.1 mV\r'),
(79,31,3,'96.2 %\r','-2.0 mV\r'),
(80,31,4,'102.8 %\r','9.7 mV\r'),
(81,32,1,'103.7 %\r','-29.2 mV\r'),
(82,32,2,'96.4 %\r','-16.5 mV\r'),
(83,32,3,'97.4 %\r','-16.5 mV\r'),
(84,32,4,'102.4 %\r','-7.6 mV\r'),
(85,33,1,'100.6 %\r','-4.2 mV\r'),
(86,33,2,'99.8 %\r','-2.8 mV\r'),
(87,33,3,'96.4 %\r','-2.8 mV\r'),
(88,33,4,'102.7 %\r','8.4 mV\r'),
(89,34,1,'98.1 %\r','2.1 mV\r'),
(90,34,2,'99.3 %\r','0.0 mV\r'),
(91,34,3,'97.5 %\r','0.0 mV\r'),
(92,34,4,'101.1 %\r','6.3 mV\r'),
(93,35,1,'103.8 %\r','-30.2 mV\r'),
(94,35,2,'96.5 %\r','-17.2 mV\r'),
(95,35,3,'96.8 %\r','-17.2 mV\r'),
(96,35,4,'103.3 %\r','-5.7 mV\r'),
(97,36,1,'99.9 %\r','-2.3 mV\r'),
(98,36,2,'99.6 %\r','-1.8 mV\r'),
(99,36,3,'95.1 %\r','-1.8 mV\r'),
(100,36,4,'103.2 %\r','12.7 mV\r'),
(101,37,1,'100.5 %\r','0.2 mV\r'),
(102,37,2,'99.4 %\r','2.2 mV\r'),
(103,37,3,'97.4 %\r','2.3 mV\r'),
(104,37,4,'98.8 %\r','4.7 mV\r'),
(105,38,1,'0','0'),
(106,38,2,'0','0'),
(107,38,3,'0','0'),
(108,38,4,'0','0'),
(109,38,5,'0','0'),
(110,39,1,'0','0'),
(111,39,2,'0','0'),
(112,39,3,'0','0'),
(113,39,4,'0','0'),
(114,39,5,'0','0'),
(115,40,1,'0','0'),
(116,40,2,'0','0'),
(117,40,3,'0','0'),
(118,40,4,'0','0'),
(119,40,5,'0','0'),
(120,41,1,'0','0'),
(121,41,2,'0','0'),
(122,41,3,'0','0'),
(123,41,4,'0','0'),
(124,41,5,'0','0'),
(125,42,1,'0','0'),
(126,42,2,'0','0'),
(127,42,3,'0','0'),
(128,42,4,'0','0'),
(129,42,5,'0','0'),
(130,43,1,'0','0'),
(131,43,2,'0','0'),
(132,43,3,'0','0'),
(133,43,4,'0','0'),
(134,43,5,'0','0'),
(135,44,1,'0','0'),
(136,44,2,'0','0'),
(137,44,3,'0','0'),
(138,44,4,'0','0'),
(139,44,5,'0','0'),
(140,45,1,'0','0'),
(141,45,2,'0','0'),
(142,45,3,'0','0'),
(143,45,4,'0','0'),
(144,45,5,'0','0'),
(145,46,1,'0','0'),
(146,46,2,'0','0'),
(147,46,3,'0','0'),
(148,46,4,'0','0'),
(149,46,5,'0','0'),
(150,47,1,'0','0'),
(151,47,2,'0','0'),
(152,47,3,'0','0'),
(153,47,4,'0','0'),
(154,47,5,'0','0'),
(155,48,1,'0','0'),
(156,48,2,'0','0'),
(157,48,3,'0','0'),
(158,48,4,'0','0'),
(159,48,5,'0','0'),
(160,49,1,'0','0'),
(161,49,2,'0','0'),
(162,49,3,'0','0'),
(163,49,4,'0','0'),
(164,49,5,'0','0'),
(165,50,1,'0','0'),
(166,50,2,'0','0'),
(167,50,3,'0','0'),
(168,50,4,'0','0'),
(169,50,5,'0','0'),
(170,51,1,'101.3 %\r','-5.0 mV\r'),
(171,51,2,'99.3 %\r','-1.5 mV\r'),
(172,51,3,'95.4 %\r','-1.5 mV\r'),
(173,51,4,'101.3 %\r','9.1 mV\r'),
(174,51,5,'0','0'),
(175,52,1,'0','0'),
(176,52,2,'0','0'),
(177,52,3,'0','0'),
(178,52,4,'0','0'),
(179,52,5,'0','0'),
(180,53,1,'0','0'),
(181,53,2,'0','0'),
(182,53,3,'0','0'),
(183,53,4,'0','0'),
(184,53,5,'0','0'),
(185,54,1,'0','0'),
(186,54,2,'0','0'),
(187,54,3,'0','0'),
(188,54,4,'0','0'),
(189,54,5,'0','0'),
(190,55,1,'0','0'),
(191,55,2,'0','0'),
(192,55,3,'0','0'),
(193,55,4,'0','0'),
(194,55,5,'0','0'),
(195,56,1,'0','0'),
(196,56,2,'0','0'),
(197,56,3,'0','0'),
(198,56,4,'0','0'),
(199,56,5,'0','0'),
(200,57,1,'0','0'),
(201,57,2,'0','0'),
(202,57,3,'0','0'),
(203,57,4,'0','0'),
(204,57,5,'0','0'),
(205,58,1,'0','0'),
(206,58,2,'0','0'),
(207,58,3,'0','0'),
(208,58,4,'0','0'),
(209,58,5,'0','0'),
(210,59,1,'0','0'),
(211,59,2,'0','0'),
(212,59,3,'0','0'),
(213,59,4,'0','0'),
(214,59,5,'0','0'),
(215,60,1,'0','0'),
(216,60,2,'0','0'),
(217,60,3,'0','0'),
(218,60,4,'0','0'),
(219,60,5,'0','0'),
(220,61,1,'0','0'),
(221,61,2,'0','0'),
(222,61,3,'0','0'),
(223,61,4,'0','0'),
(224,61,5,'0','0'),
(225,62,1,'0','0'),
(226,62,2,'0','0'),
(227,62,3,'0','0'),
(228,62,4,'0','0'),
(229,62,5,'0','0'),
(230,63,1,'0','0'),
(231,63,2,'0','0'),
(232,63,3,'0','0'),
(233,63,4,'0','0'),
(234,63,5,'0','0'),
(235,64,1,'0','0'),
(236,64,2,'0','0'),
(237,64,3,'0','0'),
(238,64,4,'0','0'),
(239,64,5,'0','0'),
(240,65,1,'0','0'),
(241,65,2,'0','0'),
(242,65,3,'0','0'),
(243,65,4,'0','0'),
(244,65,5,'0','0'),
(245,66,1,'0','0'),
(246,66,2,'0','0'),
(247,66,3,'0','0'),
(248,66,4,'0','0'),
(249,66,5,'0','0'),
(250,67,1,'0','0'),
(251,67,2,'0','0'),
(252,67,3,'0','0'),
(253,67,4,'0','0'),
(254,67,5,'0','0'),
(255,68,1,'0','0'),
(256,68,2,'0','0'),
(257,68,3,'0','0'),
(258,68,4,'0','0'),
(259,68,5,'0','0'),
(260,69,1,'0','0'),
(261,69,2,'0','0'),
(262,69,3,'0','0'),
(263,69,4,'0','0'),
(264,69,5,'0','0'),
(265,70,1,'0','0'),
(266,70,2,'0','0'),
(267,70,3,'0','0'),
(268,70,4,'0','0'),
(269,70,5,'0','0'),
(270,71,1,'0','0'),
(271,71,2,'0','0'),
(272,71,3,'0','0'),
(273,71,4,'0','0'),
(274,71,5,'0','0'),
(275,72,1,'0','0'),
(276,72,2,'0','0'),
(277,72,3,'0','0'),
(278,72,4,'0','0'),
(279,72,5,'0','0'),
(280,73,1,'0','0'),
(281,73,2,'0','0'),
(282,73,3,'0','0'),
(283,73,4,'0','0'),
(284,73,5,'0','0'),
(285,74,1,'0','0'),
(286,74,2,'0','0'),
(287,74,3,'0','0'),
(288,74,4,'0','0'),
(289,74,5,'0','0'),
(290,75,1,'0','0'),
(291,75,2,'0','0'),
(292,75,3,'0','0'),
(293,75,4,'0','0'),
(294,75,5,'0','0'),
(295,76,1,'0','0'),
(296,76,2,'0','0'),
(297,76,3,'0','0'),
(298,76,4,'0','0'),
(299,76,5,'0','0'),
(300,77,1,'0','0'),
(301,77,2,'0','0'),
(302,77,3,'0','0'),
(303,77,4,'0','0'),
(304,77,5,'0','0'),
(305,78,1,'0','0'),
(306,78,2,'0','0'),
(307,78,3,'0','0'),
(308,78,4,'0','0'),
(309,78,5,'0','0'),
(310,79,1,'0','0'),
(311,79,2,'0','0'),
(312,79,3,'0','0'),
(313,79,4,'0','0'),
(314,79,5,'0','0'),
(315,80,1,'0','0'),
(316,80,2,'0','0'),
(317,80,3,'0','0'),
(318,80,4,'0','0'),
(319,80,5,'0','0'),
(320,81,1,'0','0'),
(321,81,2,'0','0'),
(322,81,3,'0','0'),
(323,81,4,'0','0'),
(324,81,5,'0','0'),
(325,82,1,'0','0'),
(326,82,2,'0','0'),
(327,82,3,'0','0'),
(328,82,4,'0','0'),
(329,82,5,'0','0'),
(330,83,1,'0','0'),
(331,83,2,'0','0'),
(332,83,3,'0','0'),
(333,83,4,'0','0'),
(334,83,5,'0','0'),
(335,84,1,'0','0'),
(336,84,2,'0','0'),
(337,84,3,'0','0'),
(338,84,4,'0','0'),
(339,84,5,'0','0'),
(340,85,1,'0','0'),
(341,85,2,'0','0'),
(342,85,3,'0','0'),
(343,85,4,'0','0'),
(344,85,5,'0','0'),
(345,86,1,'0','0'),
(346,86,2,'0','0'),
(347,86,3,'0','0'),
(348,86,4,'0','0'),
(349,86,5,'0','0'),
(350,87,1,'0','0'),
(351,87,2,'0','0'),
(352,87,3,'0','0'),
(353,87,4,'0','0'),
(354,87,5,'0','0'),
(355,88,1,'0','0'),
(356,88,2,'0','0'),
(357,88,3,'0','0'),
(358,88,4,'0','0'),
(359,88,5,'0','0'),
(360,89,1,'0','0'),
(361,89,2,'0','0'),
(362,89,3,'0','0'),
(363,89,4,'0','0'),
(364,89,5,'0','0'),
(365,90,1,'0','0'),
(366,90,2,'0','0'),
(367,90,3,'0','0'),
(368,90,4,'0','0'),
(369,90,5,'0','0'),
(370,91,1,'0','0'),
(371,91,2,'0','0'),
(372,91,3,'0','0'),
(373,91,4,'0','0'),
(374,91,5,'0','0'),
(375,92,1,'0','0'),
(376,92,2,'0','0'),
(377,92,3,'0','0'),
(378,92,4,'0','0'),
(379,92,5,'0','0'),
(380,93,1,'0','0'),
(381,93,2,'0','0'),
(382,93,3,'0','0'),
(383,93,4,'0','0'),
(384,93,5,'0','0'),
(385,94,1,'0','0'),
(386,94,2,'0','0'),
(387,94,3,'0','0'),
(388,94,4,'0','0'),
(389,94,5,'0','0'),
(390,95,1,'0','0'),
(391,95,2,'0','0'),
(392,95,3,'0','0'),
(393,95,4,'0','0'),
(394,95,5,'0','0'),
(395,96,1,'0','0'),
(396,96,2,'0','0'),
(397,96,3,'0','0'),
(398,96,4,'0','0'),
(399,96,5,'0','0'),
(400,97,1,'0','0'),
(401,97,2,'0','0'),
(402,97,3,'0','0'),
(403,97,4,'0','0'),
(404,97,5,'0','0'),
(405,98,1,'0','0'),
(406,98,2,'0','0'),
(407,98,3,'0','0'),
(408,98,4,'0','0'),
(409,98,5,'0','0'),
(410,99,1,'0','0'),
(411,99,2,'0','0'),
(412,99,3,'0','0'),
(413,99,4,'0','0'),
(414,99,5,'0','0'),
(415,100,1,'0','0'),
(416,100,2,'0','0'),
(417,100,3,'0','0'),
(418,100,4,'0','0'),
(419,100,5,'0','0'),
(420,101,1,'0','0'),
(421,101,2,'0','0'),
(422,101,3,'0','0'),
(423,101,4,'0','0'),
(424,101,5,'0','0'),
(425,102,1,'0','0'),
(426,102,2,'0','0'),
(427,102,3,'0','0'),
(428,102,4,'0','0'),
(429,102,5,'0','0'),
(430,103,1,'0','0'),
(431,103,2,'0','0'),
(432,103,3,'0','0'),
(433,103,4,'0','0'),
(434,103,5,'0','0'),
(435,104,1,'0','0'),
(436,104,2,'0','0'),
(437,104,3,'0','0'),
(438,104,4,'0','0'),
(439,104,5,'0','0'),
(440,105,1,'0','0'),
(441,105,2,'0','0'),
(442,105,3,'0','0'),
(443,105,4,'0','0'),
(444,105,5,'0','0'),
(445,106,1,'0','0'),
(446,106,2,'0','0'),
(447,106,3,'0','0'),
(448,106,4,'0','0'),
(449,106,5,'0','0'),
(450,107,1,'0','0'),
(451,107,2,'0','0'),
(452,107,3,'0','0'),
(453,107,4,'0','0'),
(454,107,5,'0','0'),
(455,108,1,'0','0'),
(456,108,2,'0','0'),
(457,108,3,'0','0'),
(458,108,4,'0','0'),
(459,108,5,'0','0'),
(460,109,1,'0','0'),
(461,109,2,'0','0'),
(462,109,3,'0','0'),
(463,109,4,'0','0'),
(464,109,5,'0','0'),
(465,110,1,'0','0'),
(466,110,2,'0','0'),
(467,110,3,'0','0'),
(468,110,4,'0','0'),
(469,110,5,'0','0'),
(470,111,1,'0','0'),
(471,111,2,'0','0'),
(472,111,3,'0','0'),
(473,111,4,'0','0'),
(474,111,5,'0','0'),
(475,112,1,'0','0'),
(476,112,2,'0','0'),
(477,112,3,'0','0'),
(478,112,4,'0','0'),
(479,112,5,'0','0'),
(480,113,1,'0','0'),
(481,113,2,'0','0'),
(482,113,3,'0','0'),
(483,113,4,'0','0'),
(484,113,5,'0','0'),
(485,114,1,'0','0'),
(486,114,2,'0','0'),
(487,114,3,'0','0'),
(488,114,4,'0','0'),
(489,114,5,'0','0'),
(490,115,1,'0','0'),
(491,115,2,'0','0'),
(492,115,3,'0','0'),
(493,115,4,'0','0'),
(494,115,5,'0','0'),
(495,116,1,'0','0'),
(496,116,2,'0','0'),
(497,116,3,'0','0'),
(498,116,4,'0','0'),
(499,116,5,'0','0'),
(500,117,1,'0','0'),
(501,117,2,'0','0'),
(502,117,3,'0','0'),
(503,117,4,'0','0'),
(504,117,5,'0','0'),
(505,118,1,'0','0'),
(506,118,2,'0','0'),
(507,118,3,'0','0'),
(508,118,4,'0','0'),
(509,118,5,'0','0'),
(510,119,1,'0','0'),
(511,119,2,'0','0'),
(512,119,3,'0','0'),
(513,119,4,'0','0'),
(514,119,5,'0','0'),
(515,120,1,'0','0'),
(516,120,2,'0','0'),
(517,120,3,'0','0'),
(518,120,4,'0','0'),
(519,120,5,'0','0'),
(520,121,1,'0','0'),
(521,121,2,'0','0'),
(522,121,3,'0','0'),
(523,121,4,'0','0'),
(524,121,5,'0','0'),
(525,122,1,'0','0'),
(526,122,2,'0','0'),
(527,122,3,'0','0'),
(528,122,4,'0','0'),
(529,122,5,'0','0'),
(530,123,1,'0','0'),
(531,123,2,'0','0'),
(532,123,3,'0','0'),
(533,123,4,'0','0'),
(534,123,5,'0','0'),
(535,124,1,'0','0'),
(536,124,2,'0','0'),
(537,124,3,'0','0'),
(538,124,4,'0','0'),
(539,124,5,'0','0'),
(540,125,1,'0','0'),
(541,125,2,'0','0'),
(542,125,3,'0','0'),
(543,125,4,'0','0'),
(544,125,5,'0','0'),
(545,126,1,'0','0'),
(546,126,2,'0','0'),
(547,126,3,'0','0'),
(548,126,4,'0','0'),
(549,126,5,'0','0'),
(550,127,1,'0','0'),
(551,127,2,'0','0'),
(552,127,3,'0','0'),
(553,127,4,'0','0'),
(554,127,5,'0','0'),
(555,128,1,'0','0'),
(556,128,2,'0','0'),
(557,128,3,'0','0'),
(558,128,4,'0','0'),
(559,128,5,'0','0'),
(560,129,1,'0','0'),
(561,129,2,'0','0'),
(562,129,3,'0','0'),
(563,129,4,'0','0'),
(564,129,5,'0','0'),
(565,130,1,'0','0'),
(566,130,2,'0','0'),
(567,130,3,'0','0'),
(568,130,4,'0','0'),
(569,130,5,'0','0'),
(570,131,1,'0','0'),
(571,131,2,'0','0'),
(572,131,3,'0','0'),
(573,131,4,'0','0'),
(574,131,5,'0','0'),
(575,132,1,'0','0'),
(576,132,2,'0','0'),
(577,132,3,'0','0'),
(578,132,4,'0','0'),
(579,132,5,'0','0'),
(580,133,1,'0','0'),
(581,133,2,'0','0'),
(582,133,3,'0','0'),
(583,133,4,'0','0'),
(584,133,5,'0','0'),
(585,134,1,'0','0'),
(586,134,2,'0','0'),
(587,134,3,'0','0'),
(588,134,4,'0','0'),
(589,134,5,'0','0'),
(590,135,1,'0','0'),
(591,135,2,'0','0'),
(592,135,3,'0','0'),
(593,135,4,'0','0'),
(594,135,5,'0','0'),
(595,136,1,'0','0'),
(596,136,2,'0','0'),
(597,136,3,'0','0'),
(598,136,4,'0','0'),
(599,136,5,'0','0'),
(600,137,1,'0','0'),
(601,137,2,'0','0'),
(602,137,3,'0','0'),
(603,137,4,'0','0'),
(604,137,5,'0','0'),
(605,138,1,'0','0'),
(606,138,2,'0','0'),
(607,138,3,'0','0'),
(608,138,4,'0','0'),
(609,138,5,'0','0'),
(610,139,1,'0','0'),
(611,139,2,'0','0'),
(612,139,3,'0','0'),
(613,139,4,'0','0'),
(614,139,5,'0','0'),
(615,140,1,'0','0'),
(616,140,2,'0','0'),
(617,140,3,'0','0'),
(618,140,4,'0','0'),
(619,140,5,'0','0'),
(620,141,1,'0','0'),
(621,141,2,'0','0'),
(622,141,3,'0','0'),
(623,141,4,'0','0'),
(624,141,5,'0','0'),
(625,142,1,'0','0'),
(626,142,2,'0','0'),
(627,142,3,'0','0'),
(628,142,4,'0','0'),
(629,142,5,'0','0'),
(630,143,1,'0','0'),
(631,143,2,'0','0'),
(632,143,3,'0','0'),
(633,143,4,'0','0'),
(634,143,5,'0','0'),
(635,144,1,'0','0'),
(636,144,2,'0','0'),
(637,144,3,'0','0'),
(638,144,4,'0','0'),
(639,144,5,'0','0'),
(640,145,1,'0','0'),
(641,145,2,'0','0'),
(642,145,3,'0','0'),
(643,145,4,'0','0'),
(644,145,5,'0','0'),
(645,146,1,'0','0'),
(646,146,2,'0','0'),
(647,146,3,'0','0'),
(648,146,4,'0','0'),
(649,146,5,'0','0'),
(650,147,1,'0','0'),
(651,147,2,'0','0'),
(652,147,3,'0','0'),
(653,147,4,'0','0'),
(654,147,5,'0','0'),
(655,148,1,'0','0'),
(656,148,2,'0','0'),
(657,148,3,'0','0'),
(658,148,4,'0','0'),
(659,148,5,'0','0'),
(660,149,1,'0','0'),
(661,149,2,'0','0'),
(662,149,3,'0','0'),
(663,149,4,'0','0'),
(664,149,5,'0','0'),
(665,150,1,'0','0'),
(666,150,2,'0','0'),
(667,150,3,'0','0'),
(668,150,4,'0','0'),
(669,150,5,'0','0'),
(670,151,1,'0','0'),
(671,151,2,'0','0'),
(672,151,3,'0','0'),
(673,151,4,'0','0'),
(674,151,5,'0','0'),
(675,152,1,'0','0'),
(676,152,2,'0','0'),
(677,152,3,'0','0'),
(678,152,4,'0','0'),
(679,152,5,'0','0'),
(680,153,1,'0','0'),
(681,153,2,'0','0'),
(682,153,3,'0','0'),
(683,153,4,'0','0'),
(684,153,5,'0','0'),
(685,154,1,'0','0'),
(686,154,2,'0','0'),
(687,154,3,'0','0'),
(688,154,4,'0','0'),
(689,154,5,'0','0'),
(690,155,1,'0','0'),
(691,155,2,'0','0'),
(692,155,3,'0','0'),
(693,155,4,'0','0'),
(694,155,5,'0','0'),
(695,156,1,'0','0'),
(696,156,2,'0','0'),
(697,156,3,'0','0'),
(698,156,4,'0','0'),
(699,156,5,'0','0'),
(700,157,1,'0','0'),
(701,157,2,'0','0'),
(702,157,3,'0','0'),
(703,157,4,'0','0'),
(704,157,5,'0','0'),
(705,158,1,'0','0'),
(706,158,2,'0','0'),
(707,158,3,'0','0'),
(708,158,4,'0','0'),
(709,158,5,'0','0'),
(710,159,1,'0','0'),
(711,159,2,'0','0'),
(712,159,3,'0','0'),
(713,159,4,'0','0'),
(714,159,5,'0','0'),
(715,160,1,'0','0'),
(716,160,2,'0','0'),
(717,160,3,'0','0'),
(718,160,4,'0','0'),
(719,160,5,'0','0'),
(720,161,1,'0','0'),
(721,161,2,'0','0'),
(722,161,3,'0','0'),
(723,161,4,'0','0'),
(724,161,5,'0','0'),
(725,162,1,'0','0'),
(726,162,2,'0','0'),
(727,162,3,'0','0'),
(728,162,4,'0','0'),
(729,162,5,'0','0'),
(730,163,1,'0','0'),
(731,163,2,'0','0'),
(732,163,3,'0','0'),
(733,163,4,'0','0'),
(734,163,5,'0','0'),
(735,164,1,'0','0'),
(736,164,2,'0','0'),
(737,164,3,'0','0'),
(738,164,4,'0','0'),
(739,164,5,'0','0'),
(740,165,1,'0','0'),
(741,165,2,'0','0'),
(742,165,3,'0','0'),
(743,165,4,'0','0'),
(744,165,5,'0','0'),
(745,166,1,'0','0'),
(746,166,2,'0','0'),
(747,166,3,'0','0'),
(748,166,4,'0','0'),
(749,166,5,'0','0'),
(750,167,1,'0','0'),
(751,167,2,'0','0'),
(752,167,3,'0','0'),
(753,167,4,'0','0'),
(754,167,5,'0','0'),
(755,168,1,'0','0'),
(756,168,2,'0','0'),
(757,168,3,'0','0'),
(758,168,4,'0','0'),
(759,168,5,'0','0'),
(760,169,1,'0','0'),
(761,169,2,'0','0'),
(762,169,3,'0','0'),
(763,169,4,'0','0'),
(764,169,5,'0','0'),
(765,170,1,'0','0'),
(766,170,2,'0','0'),
(767,170,3,'0','0'),
(768,170,4,'0','0'),
(769,170,5,'0','0'),
(770,171,1,'0','0'),
(771,171,2,'0','0'),
(772,171,3,'0','0'),
(773,171,4,'0','0'),
(774,171,5,'0','0'),
(775,172,1,'0','0'),
(776,172,2,'0','0'),
(777,172,3,'0','0'),
(778,172,4,'0','0'),
(779,172,5,'0','0'),
(780,173,1,'0','0'),
(781,173,2,'0','0'),
(782,173,3,'0','0'),
(783,173,4,'0','0'),
(784,173,5,'0','0'),
(785,174,1,'0','0'),
(786,174,2,'0','0'),
(787,174,3,'0','0'),
(788,174,4,'0','0'),
(789,174,5,'0','0'),
(790,175,1,'0','0'),
(791,175,2,'0','0'),
(792,175,3,'0','0'),
(793,175,4,'0','0'),
(794,175,5,'0','0'),
(795,176,1,'0','0'),
(796,176,2,'0','0'),
(797,176,3,'0','0'),
(798,176,4,'0','0'),
(799,176,5,'0','0'),
(800,177,1,'0','0'),
(801,177,2,'0','0'),
(802,177,3,'0','0'),
(803,177,4,'0','0'),
(804,177,5,'0','0'),
(805,178,1,'0','0'),
(806,178,2,'0','0'),
(807,178,3,'0','0'),
(808,178,4,'0','0'),
(809,178,5,'0','0'),
(810,179,1,'0','0'),
(811,179,2,'0','0'),
(812,179,3,'0','0'),
(813,179,4,'0','0'),
(814,179,5,'0','0'),
(815,180,1,'0','0'),
(816,180,2,'0','0'),
(817,180,3,'0','0'),
(818,180,4,'0','0'),
(819,180,5,'0','0'),
(820,181,1,'0','0'),
(821,181,2,'0','0'),
(822,181,3,'0','0'),
(823,181,4,'0','0'),
(824,181,5,'0','0'),
(825,182,1,'0','0'),
(826,182,2,'0','0'),
(827,182,3,'0','0'),
(828,182,4,'0','0'),
(829,182,5,'0','0'),
(830,183,1,'0','0'),
(831,183,2,'0','0'),
(832,183,3,'0','0'),
(833,183,4,'0','0'),
(834,183,5,'0','0'),
(835,184,1,'0','0'),
(836,184,2,'0','0'),
(837,184,3,'0','0'),
(838,184,4,'0','0'),
(839,184,5,'0','0');

/*Table structure for table `tbl_calibration_detail_ph_ise` */


CREATE TABLE `tbl_calibration_detail_ph_ise` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `RepSerNo` int(11) DEFAULT NULL,
  `RecSeqNo` int(11) DEFAULT NULL,
  `Slope` varchar(100) DEFAULT '0',
  `Eo` varchar(100) DEFAULT '0',
  `Concentration` varchar(100) DEFAULT '0',
  `mVValue` varchar(100) DEFAULT '0',
  `Temp` varchar(100) DEFAULT '0',
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_calibration_detail_ph_ise` */

/*Table structure for table `tbl_calibration_detail_temperature` */


CREATE TABLE `tbl_calibration_detail_temperature` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `RepSerNo` int(11) DEFAULT NULL,
  `RecSeqNo` int(11) DEFAULT NULL,
  `TempType` varchar(100) DEFAULT '0',
  `RefTemp` varchar(100) DEFAULT '0',
  `ActualTemp` varchar(100) DEFAULT '0',
  `Offset` varchar(100) DEFAULT '0',
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_calibration_detail_temperature` */

/*Table structure for table `tbl_calibration_master_conductivity` */


CREATE TABLE `tbl_calibration_master_conductivity` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `PrDate` date DEFAULT NULL,
  `PrTime` time DEFAULT NULL,
  `MeterID` varchar(500) DEFAULT 'NULL',
  `ProbID` varchar(500) DEFAULT 'NULL',
  `StdValue` varchar(30) DEFAULT '0',
  `NegTol` varchar(30) DEFAULT NULL,
  `PosTol` varchar(30) DEFAULT NULL,
  `DP` int(5) DEFAULT '0',
  `Remark` varchar(50) DEFAULT 'NULL',
  `UserID` varchar(500) DEFAULT 'NULL',
  `UserName` varchar(500) DEFAULT 'NULL',
  `ChannelNo` int(11) DEFAULT '0',
  `IDSNo` int(11) DEFAULT '0',
  `ProbType` varchar(50) DEFAULT 'Conductivity',
  `ProdID` varchar(500) DEFAULT 'NULL',
  `PrinterName` varchar(500) DEFAULT 'NULL',
  `PrintNo` int(11) DEFAULT '0',
  `CheckedByID` varchar(25) DEFAULT 'NULL',
  `CheckedByName` varchar(200) DEFAULT 'NULL',
  `CheckedByDate` datetime DEFAULT '1992-12-08 00:00:00',
  `VerifyRemark` varchar(500) DEFAULT 'NULL',
  `AllData` text,
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_calibration_master_conductivity` */

/*Table structure for table `tbl_calibration_master_do` */


CREATE TABLE `tbl_calibration_master_do` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `PrDate` date DEFAULT NULL,
  `PrTime` time DEFAULT NULL,
  `MeterID` varchar(500) DEFAULT 'NULL',
  `ProbID` varchar(500) DEFAULT 'NULL',
  `NegTol` varchar(30) DEFAULT NULL,
  `PosTol` varchar(30) DEFAULT NULL,
  `DP` int(5) DEFAULT '0',
  `Slope` varchar(30) DEFAULT '0',
  `Remark` varchar(50) DEFAULT 'NULL',
  `UserID` varchar(500) DEFAULT 'NULL',
  `UserName` varchar(500) DEFAULT 'NULL',
  `ChannelNo` int(11) DEFAULT '0',
  `IDSNo` int(11) DEFAULT '0',
  `ProbType` varchar(50) DEFAULT 'DO',
  `ProdID` varchar(500) DEFAULT 'NULL',
  `PrinterName` varchar(500) DEFAULT 'NULL',
  `PrintNo` int(11) DEFAULT '0',
  `Saturation` varchar(50) DEFAULT '0',
  `Current` varchar(50) DEFAULT '0',
  `ZeroCurrent` varchar(50) DEFAULT '0',
  `TempSolution` varchar(50) DEFAULT '0',
  `TempMembrane` varchar(50) DEFAULT '0',
  `CalibType` varchar(100) DEFAULT 'NULL',
  `BioPressure` varchar(50) DEFAULT '0',
  `CheckedByID` varchar(25) DEFAULT 'NULL',
  `CheckedByName` varchar(200) DEFAULT 'NULL',
  `CheckedByDate` datetime DEFAULT '1992-12-08 00:00:00',
  `VerifyRemark` varchar(500) DEFAULT 'NULL',
  `AllData` text,
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_calibration_master_do` */

/*Table structure for table `tbl_calibration_master_ph` */


CREATE TABLE `tbl_calibration_master_ph` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `PrDate` date DEFAULT NULL,
  `PrTime` time DEFAULT NULL,
  `MeterID` varchar(500) DEFAULT 'NULL',
  `ProbID` varchar(500) DEFAULT 'NULL',
  `NegTol` varchar(30) DEFAULT NULL,
  `PosTol` varchar(30) DEFAULT NULL,
  `DP` int(5) DEFAULT '0',
  `AvgSlope` varchar(30) DEFAULT '0',
  `Remark` varchar(50) DEFAULT 'NULL',
  `UserID` varchar(500) DEFAULT 'NULL',
  `UserName` varchar(500) DEFAULT 'NULL',
  `ChannelNo` int(11) DEFAULT '0',
  `IDSNo` int(11) DEFAULT '0',
  `ProbType` varchar(50) DEFAULT 'pH',
  `ProdID` varchar(500) DEFAULT 'NULL',
  `PrinterName` varchar(500) DEFAULT 'NULL',
  `PrintNo` int(11) DEFAULT '0',
  `CheckedByID` varchar(25) DEFAULT 'NULL',
  `CheckedByName` varchar(200) DEFAULT 'NULL',
  `CheckedByDate` datetime DEFAULT '1992-12-08 00:00:00',
  `VerifyRemark` varchar(500) DEFAULT 'NULL',
  `AllData` text,
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB AUTO_INCREMENT=185 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_calibration_master_ph` */

insert  into `tbl_calibration_master_ph`(`RecNo`,`PrDate`,`PrTime`,`MeterID`,`ProbID`,`NegTol`,`PosTol`,`DP`,`AvgSlope`,`Remark`,`UserID`,`UserName`,`ChannelNo`,`IDSNo`,`ProbType`,`ProdID`,`PrinterName`,`PrintNo`,`CheckedByID`,`CheckedByName`,`CheckedByDate`,`VerifyRemark`,`AllData`) values 
(1,'2024-09-13','09:36:30','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'100.3 %','Pass','V-trident','Trident',1,101,'pH','NULL','NULL',2,'V-trident','Trident','2024-09-13 10:05:07','As part of OQ','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        13-09-24,09:35\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      7.00\r\nmV                      0.0\r\nTemperature             25.0 C (MAN)\r\nCalibration Type        Auto\r\n\r\nAverage Slope           100.3 %\r\nCal #                   103\r\n\r\n'),
(2,'2024-09-13','09:47:50','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'100.3 %','Pass','V-trident','Trident',1,101,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        13-09-24,09:35\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      7.00\r\nmV                      0.0\r\nTemperature             25.0 C (MAN)\r\nCalibration Type        Auto\r\n\r\nAverage Slope           100.3 %\r\nCal #                   103\r\n\r\n'),
(3,'2024-09-17','15:03:17','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'100.3 %','Pass','V-trident','Trident',1,101,'pH','NULL','NULL',1,'V-trident','Trident','2024-09-17 15:32:30','As part of PQ','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        17-09-24,15:01\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      7.00\r\nmV                      0.0\r\nTemperature             25.0 C (MAN)\r\nCalibration Type        Auto\r\n\r\nAverage Slope           100.3 %\r\nCal #                   104\r\n\r\n'),
(4,'2024-09-17','15:15:56','QC-PHMTZ-009','CX1-16015','90.00 %','105.00 %',0,'100.0 %','Pass','V-trident','Trident',1,102,'pH','NULL','NULL',1,'V-trident','Trident','2024-09-17 15:33:46','As part of PQ','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA32174\r\nMethod                  100\r\nCalibration Time        17-09-24,15:07\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      7.00\r\nmV                      0.0\r\nTemperature             25.0 C (MAN)\r\nCalibration Type        Auto\r\n\r\nAverage Slope           100.0 %\r\nCal #                   103\r\n\r\n'),
(5,'2024-09-17','16:34:53','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'100.3 %','Pass','V-trident','Trident',1,103,'pH','NULL','NULL',1,'V-trident','Trident','2024-09-17 16:41:00','As part of PQ','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        17-09-24,16:34\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      7.00\r\nmV                      0.0\r\nTemperature             25.0 C (MAN)\r\nCalibration Type        Auto\r\n\r\nAverage Slope           100.3 %\r\nCal #                   103\r\n\r\n'),
(6,'2024-09-18','14:30:09','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'99.4 %','Pass','V-trident','Trident',1,105,'pH','NULL','NULL',1,'V-trident','Trident','2024-09-18 15:49:42','As part of PQ','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        18-09-24,14:16\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      7.00\r\nmV                      0.0\r\nTemperature             25.0 C (MAN)\r\nCalibration Type        Auto\r\n\r\nAverage Slope           99.4 %\r\nCal #                   103\r\n\r\n'),
(7,'2024-09-18','15:41:50','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'100.0 %','Pass','V-trident','Trident',1,104,'pH','NULL','NULL',1,'V-trident','Trident','2024-09-18 15:50:12','As part of PQ','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        18-09-24,15:40\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      7.00\r\nmV                      0.0\r\nTemperature             25.0 C (MAN)\r\nCalibration Type        Auto\r\n\r\nAverage Slope           100.0 %\r\nCal #                   103\r\n\r\n'),
(8,'2024-09-19','10:39:13','QC-PHMTZ-009','CX1-16015','90.00 %','105.00 %',0,'100.0 %','Pass','V-trident','Trident',1,102,'pH','NULL','NULL',0,'V-trident','Trident','2024-09-19 10:46:07','As part of DEMO','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA32174\r\nMethod                  100\r\nCalibration Time        19-09-24,10:37\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      7.00\r\nmV                      11.5\r\nTemperature             23.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nAverage Slope           100.0 %\r\nCal #                   104\r\n\r\n'),
(9,'2024-09-19','12:01:55','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'100.3 %','Pass','V-trident','Trident',1,101,'pH','NULL','NULL',0,'V-trident','Trident','2024-09-19 12:08:19','As part of DEMO','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        19-09-24,12:00\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      7.00\r\nmV                      -3.1\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nAverage Slope           100.3 %\r\nCal #                   105\r\n\r\n'),
(10,'2024-09-19','12:30:37','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'100.3 %','Pass','V-trident','Trident',1,101,'pH','NULL','NULL',0,'V-trident','Trident','2024-09-19 12:50:38','As part of DEMO','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        19-09-24,12:28\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      7.00\r\nmV                      -0.3\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nAverage Slope           100.3 %\r\nCal #                   106\r\n\r\n'),
(11,'2024-09-30','16:13:13','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'100.3 %','Pass','V-trident','Trident',1,103,'pH','NULL','NULL',0,'V-trident','Trident','2024-09-30 16:16:06','As part of demo to user by vendor','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        30-09-24,16:12\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      7.00\r\nmV                      0.0\r\nTemperature             25.0 C (MAN)\r\nCalibration Type        Auto\r\n\r\nAverage Slope           100.3 %\r\nCal #                   104\r\n\r\n'),
(12,'2024-09-30','16:51:44','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'98.6 %','Pass','205089','M. Venkata Ramana',1,101,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-09-30 17:53:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        30-09-24,16:49\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      310.9\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      175.6\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -3.5\r\nTemperature             23.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -174.1\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -317.1\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                98.5 %\r\nSlope 2                101.3 %\r\nSlope 3                95.7 %\r\nSlope 4                98.8 %\r\nEo    1                1.6 mV\r\nEo    2                -3.5 mV\r\nEo    3                -3.5 mV\r\nEo    4                2.1 mV\r\nAverage Slope           98.6 %\r\nCal #                   107\r\n\r\n'),
(13,'2024-09-30','16:56:45','QC-PHMTZ-009','CX1-16015','90.00 %','105.00 %',0,'99.3 %','Pass','205066','Sreeram Jakkaraiah',1,102,'pH','NULL','NULL',1,'86253','D. Jayapal Reddy','2024-09-30 17:41:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA32174\r\nMethod                  100\r\nCalibration Time        30-09-24,16:49\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      312.5\r\nTemperature             23.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      176.0\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      0.2\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -172.6\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -318.7\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                99.7 %\r\nSlope 2                99.7 %\r\nSlope 3                97.1 %\r\nSlope 4                100.8 %\r\nEo    1                0.1 mV\r\nEo    2                0.2 mV\r\nEo    3                0.2 mV\r\nEo    4                6.9 mV\r\nAverage Slope           99.3 %\r\nCal #                   105\r\n\r\n'),
(14,'2024-10-01','07:14:25','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'97.8 %','Pass','88571','V. Ramu',1,101,'pH','NULL','NULL',1,'90778','B. Bhaskar','2024-10-01 07:49:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        01-10-24,07:10\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      309.9\r\nTemperature             26.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      170.6\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -4.0\r\nTemperature             23.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -172.5\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -314.2\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                99.9 %\r\nSlope 2                98.9 %\r\nSlope 3                94.7 %\r\nSlope 4                97.7 %\r\nEo    1                -5.7 mV\r\nEo    2                -4.0 mV\r\nEo    3                -4.0 mV\r\nEo    4                1.4 mV\r\nAverage Slope           97.8 %\r\nCal #                   108\r\n\r\n'),
(15,'2024-10-01','07:20:21','QC-PHMTZ-009','CX1-16015','90.00 %','105.00 %',0,'99.7 %','Pass','205089','M. Venkata Ramana',1,102,'pH','NULL','NULL',1,'90778','B. Bhaskar','2024-10-01 07:48:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA32174\r\nMethod                  100\r\nCalibration Time        01-10-24,07:18\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      316.2\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      174.3\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      0.4\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -172.1\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -319.1\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                102.0 %\r\nSlope 2                98.5 %\r\nSlope 3                96.6 %\r\nSlope 4                101.7 %\r\nEo    1                -5.9 mV\r\nEo    2                0.4 mV\r\nEo    3                0.4 mV\r\nEo    4                9.5 mV\r\nAverage Slope           99.7 %\r\nCal #                   106\r\n\r\n'),
(16,'2024-10-01','09:13:46','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'99.0 %','Pass','205044','Prem kumar',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        01-10-24,09:12\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      312.6\r\nTemperature             23.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      177.1\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      0.8\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -171.8\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -317.5\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                98.5 %\r\nSlope 2                100.2 %\r\nSlope 3                97.0 %\r\nSlope 4                100.5 %\r\nEo    1                3.8 mV\r\nEo    2                0.8 mV\r\nEo    3                0.8 mV\r\nEo    4                7.0 mV\r\nAverage Slope           99.0 %\r\nCal #                   104\r\n\r\n'),
(17,'2024-10-01','09:23:43','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.5 %','Fail','95422','M. Shankar',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-01 11:56:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        01-10-24,09:22\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      318.9\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      180.6\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      6.99\r\nmV                      33.6\r\nTemperature             26.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.00\r\nmV                      -168.6\r\nTemperature             26.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.42\r\nmV                      -308.3\r\nTemperature             26.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                100.3 %\r\nSlope 2                83.2 %\r\nSlope 3                113.0 %\r\nSlope 4                97.3 %\r\nEo    1                2.8 mV\r\nEo    2                33.1 mV\r\nEo    3                32.9 mV\r\nEo    4                4.9 mV\r\nAverage Slope           98.5 %\r\nCal #                   105\r\n\r\n'),
(18,'2024-10-01','09:49:18','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.1 %','Pass','95422','M. Shankar',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-01 12:01:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        01-10-24,09:47\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      319.1\r\nTemperature             26.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      180.1\r\nTemperature             26.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      6.99\r\nmV                      4.3\r\nTemperature             26.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.00\r\nmV                      -169.8\r\nTemperature             26.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.42\r\nmV                      -306.7\r\nTemperature             26.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                100.6 %\r\nSlope 2                99.3 %\r\nSlope 3                97.3 %\r\nSlope 4                95.3 %\r\nEo    1                1.5 mV\r\nEo    2                3.7 mV\r\nEo    3                3.7 mV\r\nEo    4                0.0 mV\r\nAverage Slope           98.1 %\r\nCal #                   106\r\n\r\n'),
(19,'2024-10-02','08:22:35','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'98.3 %','Pass','205044','Prem kumar',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        02-10-24,08:21\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      314.5\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.8\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      4.0\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -165.5\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -308.8\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                102.8 %\r\nSlope 2                96.0 %\r\nSlope 3                95.0 %\r\nSlope 4                99.2 %\r\nEo    1                -8.0 mV\r\nEo    2                3.9 mV\r\nEo    3                4.0 mV\r\nEo    4                11.3 mV\r\nAverage Slope           98.3 %\r\nCal #                   105\r\n\r\n'),
(20,'2024-10-03','00:54:27','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'98.4 %','Pass','205104','Siva Shankar Reddy',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        03-10-24,00:53\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      310.9\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      172.7\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      2.5\r\nTemperature             23.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -169.6\r\nTemperature             23.8 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -312.6\r\nTemperature             23.7 C (ATC)\r\nCalibration Type        Manual\r\n\r\nSlope 1                100.8 %\r\nSlope 2                96.6 %\r\nSlope 3                97.1 %\r\nSlope 4                99.1 %\r\nEo    1                -4.9 mV\r\nEo    2                2.5 mV\r\nEo    3                2.5 mV\r\nEo    4                6.1 mV\r\nAverage Slope           98.4 %\r\nCal #                   106\r\n\r\n'),
(21,'2024-10-03','07:46:53','QC-PHMTZ-009','CX1-16015','90.00 %','105.00 %',0,'99.4 %','Pass','205211','S. Sailatha',1,102,'pH','NULL','NULL',0,'84378','Shivanand. W','2024-10-03 08:14:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA32174\r\nMethod                  100\r\nCalibration Time        03-10-24,07:44\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      311.8\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      172.9\r\nTemperature             25.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.6\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -173.8\r\nTemperature             25.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -321.5\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                100.5 %\r\nSlope 2                99.0 %\r\nSlope 3                95.9 %\r\nSlope 4                102.2 %\r\nEo    1                -5.3 mV\r\nEo    2                -2.6 mV\r\nEo    3                -2.6 mV\r\nEo    4                8.6 mV\r\nAverage Slope           99.4 %\r\nCal #                   107\r\n\r\n'),
(22,'2024-10-03','10:34:18','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'100.6 %','Fail','92513','Ch. Chinna Rao',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-03 12:16:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        03-10-24,10:33\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      326.0\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      180.1\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      4.7\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -167.1\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.49\r\nmV                      -314.4\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                105.9 %\r\nSlope 2                99.1 %\r\nSlope 3                96.5 %\r\nSlope 4                101.0 %\r\nEo    1                -7.4 mV\r\nEo    2                4.7 mV\r\nEo    3                4.7 mV\r\nEo    4                12.7 mV\r\nAverage Slope           100.6 %\r\nCal #                   107\r\n\r\n'),
(23,'2024-10-03','11:48:47','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.8 %','Pass','92513','Ch. Chinna Rao',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-03 12:16:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        03-10-24,11:46\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      318.3\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      180.8\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      4.9\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -168.2\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -311.2\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                99.9 %\r\nSlope 2                99.4 %\r\nSlope 3                97.2 %\r\nSlope 4                98.7 %\r\nEo    1                3.9 mV\r\nEo    2                4.9 mV\r\nEo    3                4.9 mV\r\nEo    4                7.6 mV\r\nAverage Slope           98.8 %\r\nCal #                   109\r\n\r\n'),
(24,'2024-10-04','03:36:08','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'98.8 %','Pass','205223','Jithendra',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        04-10-24,03:34\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      312.5\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      172.0\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      2.9\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -168.2\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -312.3\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Manual\r\n\r\nSlope 1                102.4 %\r\nSlope 2                96.1 %\r\nSlope 3                96.6 %\r\nSlope 4                100.0 %\r\nEo    1                -8.2 mV\r\nEo    2                2.9 mV\r\nEo    3                2.9 mV\r\nEo    4                8.8 mV\r\nAverage Slope           98.8 %\r\nCal #                   107\r\n\r\n'),
(25,'2024-10-04','07:48:29','QC-PHMTZ-009','CX1-16015','90.00 %','105.00 %',0,'99.9 %','Pass','205187','B. Ramvikas',1,102,'pH','NULL','NULL',1,'205208','S. Siddhartha Kumar Reddy','2024-10-04 08:30:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA32174\r\nMethod                  100\r\nCalibration Time        04-10-24,07:44\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      314.1\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      175.0\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      0.0\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -168.8\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.49\r\nmV                      -321.7\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                101.0 %\r\nSlope 2                99.1 %\r\nSlope 3                94.8 %\r\nSlope 4                104.8 %\r\nEo    1                -3.4 mV\r\nEo    2                -0.0 mV\r\nEo    3                0.0 mV\r\nEo    4                17.7 mV\r\nAverage Slope           99.9 %\r\nCal #                   108\r\n\r\n'),
(26,'2024-10-04','10:28:36','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'99.0 %','Pass','92513','Ch. Chinna Rao',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-04 10:59:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        04-10-24,10:27\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      317.3\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      180.5\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      4.8\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -168.2\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.49\r\nmV                      -314.1\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                99.5 %\r\nSlope 2                99.1 %\r\nSlope 3                96.9 %\r\nSlope 4                100.3 %\r\nEo    1                4.2 mV\r\nEo    2                4.8 mV\r\nEo    3                4.8 mV\r\nEo    4                10.8 mV\r\nAverage Slope           99.0 %\r\nCal #                   110\r\n\r\n'),
(27,'2024-10-05','07:52:03','QC-PHMTZ-009','CX1-16015','90.00 %','105.00 %',0,'99.7 %','Fail','205210','B. Anitha',1,102,'pH','NULL','NULL',0,'86253','D. Jayapal Reddy','2024-10-05 12:23:49','Reviewed (Refer PNC No: PNC-EP4-QC-24-0132)','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA32174\r\nMethod                  100\r\nCalibration Time        05-10-24,07:50\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      312.3\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      174.9\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -0.3\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -168.3\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -322.0\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                99.6 %\r\nSlope 2                98.9 %\r\nSlope 3                94.3 %\r\nSlope 4                106.1 %\r\nEo    1                -1.5 mV\r\nEo    2                -0.3 mV\r\nEo    3                -0.3 mV\r\nEo    4                20.7 mV\r\nAverage Slope           99.7 %\r\nCal #                   109\r\n\r\n'),
(28,'2024-10-05','08:51:16','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.1 %','Pass','205210','B. Anitha',1,101,'pH','NULL','NULL',0,'86253','D. Jayapal Reddy','2024-10-05 09:18:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        05-10-24,08:47\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      309.2\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      170.9\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.8\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -174.4\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.47\r\nmV                      -320.9\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                100.7 %\r\nSlope 2                98.5 %\r\nSlope 3                96.3 %\r\nSlope 4                100.9 %\r\nEo    1                -6.6 mV\r\nEo    2                -2.9 mV\r\nEo    3                -2.8 mV\r\nEo    4                5.2 mV\r\nAverage Slope           99.1 %\r\nCal #                   109\r\n\r\n'),
(29,'2024-10-05','10:41:49','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'99.4 %','Pass','88819','M. Ramya',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-05 10:53:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        05-10-24,10:39\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      319.5\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      180.1\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      4.6\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -168.1\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -313.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                101.2 %\r\nSlope 2                99.2 %\r\nSlope 3                97.0 %\r\nSlope 4                100.3 %\r\nEo    1                1.2 mV\r\nEo    2                4.6 mV\r\nEo    3                4.6 mV\r\nEo    4                10.5 mV\r\nAverage Slope           99.4 %\r\nCal #                   111\r\n\r\n'),
(30,'2024-10-05','12:26:01','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'99.5 %','Pass','84342','G.S.A. Naidu',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        05-10-24,12:24\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      293.9\r\nTemperature             23.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      150.4\r\nTemperature             23.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -18.6\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -189.0\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -334.7\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Manual\r\n\r\nSlope 1                104.8 %\r\nSlope 2                96.2 %\r\nSlope 3                96.3 %\r\nSlope 4                101.0 %\r\nEo    1                -33.7 mV\r\nEo    2                -18.6 mV\r\nEo    3                -18.6 mV\r\nEo    4                -10.1 mV\r\nAverage Slope           99.5 %\r\nCal #                   108\r\n\r\n'),
(31,'2024-10-05','14:44:51','QC-PHMTZ-009','CX1-16015','90.00 %','105.00 %',0,'99.7 %','Pass','205210','B. Anitha',1,102,'pH','NULL','NULL',0,'205208','S. Siddhartha Kumar Reddy','2024-10-05 16:14:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA32174\r\nMethod                  100\r\nCalibration Time        05-10-24,14:43\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      311.4\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      172.6\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.0\r\nTemperature             23.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -173.6\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -322.3\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                100.9 %\r\nSlope 2                99.1 %\r\nSlope 3                96.2 %\r\nSlope 4                102.8 %\r\nEo    1                -5.3 mV\r\nEo    2                -2.1 mV\r\nEo    3                -2.0 mV\r\nEo    4                9.7 mV\r\nAverage Slope           99.7 %\r\nCal #                   110\r\n\r\n'),
(32,'2024-10-06','00:40:01','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'100.0 %','Pass','205190','B. Prabhudas',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        06-10-24,00:39\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      294.7\r\nTemperature             22.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      152.8\r\nTemperature             22.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -16.5\r\nTemperature             22.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -188.6\r\nTemperature             22.8 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -335.8\r\nTemperature             22.7 C (ATC)\r\nCalibration Type        Manual\r\n\r\nSlope 1                103.7 %\r\nSlope 2                96.4 %\r\nSlope 3                97.4 %\r\nSlope 4                102.4 %\r\nEo    1                -29.2 mV\r\nEo    2                -16.5 mV\r\nEo    3                -16.5 mV\r\nEo    4                -7.6 mV\r\nAverage Slope           100.0 %\r\nCal #                   109\r\n\r\n'),
(33,'2024-10-06','08:46:26','QC-PHMTZ-009','CX1-16015','90.00 %','105.00 %',0,'99.9 %','Pass','205187','B. Ramvikas',1,102,'pH','NULL','NULL',1,'90018','Ch. Satyam','2024-10-06 09:49:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA32174\r\nMethod                  100\r\nCalibration Time        06-10-24,08:43\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      312.3\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.4\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.8\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -174.6\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -323.4\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                100.6 %\r\nSlope 2                99.8 %\r\nSlope 3                96.4 %\r\nSlope 4                102.7 %\r\nEo    1                -4.2 mV\r\nEo    2                -2.8 mV\r\nEo    3                -2.8 mV\r\nEo    4                8.4 mV\r\nAverage Slope           99.9 %\r\nCal #                   111\r\n\r\n'),
(34,'2024-10-07','07:10:43','QC-PHMTZ-009','CX1-16015','90.00 %','105.00 %',0,'99.0 %','Pass','205089','M. Venkata Ramana',1,102,'pH','NULL','NULL',0,'90018','Ch. Satyam','2024-10-07 07:49:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA32174\r\nMethod                  100\r\nCalibration Time        07-10-24,07:08\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      309.7\r\nTemperature             23.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      175.1\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      0.0\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -173.6\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -320.3\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                98.1 %\r\nSlope 2                99.3 %\r\nSlope 3                97.5 %\r\nSlope 4                101.1 %\r\nEo    1                2.1 mV\r\nEo    2                0.0 mV\r\nEo    3                0.0 mV\r\nEo    4                6.3 mV\r\nAverage Slope           99.0 %\r\nCal #                   112\r\n\r\n'),
(35,'2024-10-07','07:42:12','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'100.1 %','Pass','205223','Jithendra',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        07-10-24,07:41\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      294.6\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      152.4\r\nTemperature             23.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -17.2\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -188.4\r\nTemperature             23.1 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.45\r\nmV                      -336.7\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Manual\r\n\r\nSlope 1                103.8 %\r\nSlope 2                96.5 %\r\nSlope 3                96.8 %\r\nSlope 4                103.3 %\r\nEo    1                -30.2 mV\r\nEo    2                -17.2 mV\r\nEo    3                -17.2 mV\r\nEo    4                -5.7 mV\r\nAverage Slope           100.1 %\r\nCal #                   110\r\n\r\n'),
(36,'2024-10-07','08:28:28','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.4 %','Pass','205089','M. Venkata Ramana',1,101,'pH','NULL','NULL',1,'90018','Ch. Satyam','2024-10-07 09:03:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        07-10-24,08:25\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      310.6\r\nTemperature             23.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      174.1\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -1.8\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -171.6\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -320.8\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                99.9 %\r\nSlope 2                99.6 %\r\nSlope 3                95.1 %\r\nSlope 4                103.2 %\r\nEo    1                -2.3 mV\r\nEo    2                -1.8 mV\r\nEo    3                -1.8 mV\r\nEo    4                12.7 mV\r\nAverage Slope           99.4 %\r\nCal #                   110\r\n\r\n'),
(37,'2024-10-07','11:49:25','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'99.1 %','Pass','92513','Ch. Chinna Rao',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-07 13:48:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        07-10-24,11:47\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      316.1\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      177.9\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      2.3\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -171.7\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -314.7\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                100.5 %\r\nSlope 2                99.4 %\r\nSlope 3                97.4 %\r\nSlope 4                98.8 %\r\nEo    1                0.2 mV\r\nEo    2                2.2 mV\r\nEo    3                2.3 mV\r\nEo    4                4.7 mV\r\nAverage Slope           99.1 %\r\nCal #                   112\r\n\r\n'),
(38,'2024-10-07','18:01:12','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'99.0 %','Pass','92513','Ch. Chinna Rao',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-07 18:10:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        07-10-24,17:59\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      313.0\r\nTemperature             23.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      175.9\r\nTemperature             23.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      1.4\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -174.3\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -317.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    2.1 mV\r\nAverage Slope           99.0 %\r\nCal #                   115\r\n\r\n'),
(39,'2024-10-08','07:18:36','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'98.8 %','Pass','205089','M. Venkata Ramana',1,101,'pH','NULL','NULL',1,'90018','Ch. Satyam','2024-10-08 08:03:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        08-10-24,07:15\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      311.5\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.7\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -172.8\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -320.8\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -0.4 mV\r\nAverage Slope           98.8 %\r\nCal #                   111\r\n\r\n'),
(40,'2024-10-08','10:10:38','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'99.3 %','Pass','88819','M. Ramya',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-08 10:31:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        08-10-24,10:09\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      312.0\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      174.8\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -0.6\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.03\r\nmV                      -175.8\r\nTemperature             23.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.48\r\nmV                      -321.4\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    0.1 mV\r\nAverage Slope           99.3 %\r\nCal #                   116\r\n\r\n'),
(41,'2024-10-08','16:56:45','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'98.8 %','Pass','205104','N. Siva Sekhar Reddy',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        08-10-24,16:55\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      293.9\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      157.2\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -17.4\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.03\r\nmV                      -187.7\r\nTemperature             23.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -336.7\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -15.7 mV\r\nAverage Slope           98.8 %\r\nCal #                   112\r\n\r\n'),
(42,'2024-10-09','00:32:51','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'98.8 %','Pass','205044','Prem kumar',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        09-10-24,00:31\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      294.4\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      160.0\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -17.5\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -187.5\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -335.4\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -14.9 mV\r\nAverage Slope           98.8 %\r\nCal #                   113\r\n\r\n'),
(43,'2024-10-09','07:15:50','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'98.7 %','Pass','88599','MD Mahamad',1,101,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-10-09 07:58:20','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        09-10-24,07:12\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      309.9\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.8\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -1.6\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -172.5\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -322.1\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -0.7 mV\r\nAverage Slope           98.7 %\r\nCal #                   112\r\n\r\n'),
(44,'2024-10-09','08:47:12','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.9 %','Pass','92513','Ch. Chinna Rao',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-09 10:01:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        09-10-24,08:45\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      309.8\r\nTemperature             23.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      172.8\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.3\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -177.8\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -320.0\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -1.4 mV\r\nAverage Slope           98.9 %\r\nCal #                   117\r\n\r\n'),
(45,'2024-10-10','05:18:39','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'99.1 %','Pass','205223','Jithendra',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        10-10-24,05:17\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      293.6\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      158.7\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -18.3\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -188.2\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -336.7\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -16.2 mV\r\nAverage Slope           99.1 %\r\nCal #                   114\r\n\r\n'),
(46,'2024-10-10','07:33:03','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'98.8 %','Pass','205075','G. Shyam Kumar',1,101,'pH','NULL','NULL',1,'84378','Shivanand. W','2024-10-10 08:02:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        10-10-24,07:29\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      310.3\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      170.9\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.0\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -173.8\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -321.2\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -1.0 mV\r\nAverage Slope           98.8 %\r\nCal #                   113\r\n\r\n'),
(47,'2024-10-10','08:47:49','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'99.1 %','Pass','92513','Ch. Chinna Rao',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-10 09:13:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        10-10-24,08:46\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      311.7\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      172.9\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.8\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -177.4\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -321.1\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -1.6 mV\r\nAverage Slope           99.1 %\r\nCal #                   118\r\n\r\n'),
(48,'2024-10-11','00:30:29','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'98.8 %','Pass','88571','V. Ramu',1,101,'pH','NULL','NULL',1,'90018','Ch. Satyam','2024-10-11 01:23:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        11-10-24,00:27\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      312.7\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.0\r\nTemperature             25.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -1.2\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -173.5\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -320.1\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -0.0 mV\r\nAverage Slope           98.8 %\r\nCal #                   114\r\n\r\n'),
(49,'2024-10-11','07:14:59','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'99.1 %','Pass','88819','M. Ramya',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-11 09:50:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        11-10-24,07:12\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      313.2\r\nTemperature             26.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      172.5\r\nTemperature             26.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      6.99\r\nmV                      -3.4\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.00\r\nmV                      -178.5\r\nTemperature             26.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.43\r\nmV                      -320.3\r\nTemperature             26.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -2.1 mV\r\nAverage Slope           99.1 %\r\nCal #                   119\r\n\r\n'),
(50,'2024-10-11','14:19:44','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'98.4 %','Pass','205104','N. Siva Sekhar Reddy',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        11-10-24,14:18\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      293.3\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      156.2\r\nTemperature             23.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -19.0\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -187.4\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.48\r\nmV                      -335.7\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -16.0 mV\r\nAverage Slope           98.4 %\r\nCal #                   115\r\n\r\n'),
(51,'2024-10-11','15:25:35','QC-PHMTZ-009','CX1-16015','90.00 %','105.00 %',0,'99.3 %','Pass','87624','N. Nani babu',1,102,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-10-11 16:13:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA32174\r\nMethod                  100\r\nCalibration Time        11-10-24,15:23\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      313.7\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      174.0\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -1.5\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -171.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -318.2\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nSlope 1                101.3 %\r\nSlope 2                99.3 %\r\nSlope 3                95.4 %\r\nSlope 4                101.3 %\r\nEo    1                -5.0 mV\r\nEo    2                -1.5 mV\r\nEo    3                -1.5 mV\r\nEo    4                9.1 mV\r\nAverage Slope           99.3 %\r\nCal #                   113\r\n\r\n'),
(52,'2024-10-12','15:32:10','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'98.3 %','Pass','84342','G.S.A. Naidu',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        12-10-24,15:30\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      294.0\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      158.3\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -17.3\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -186.4\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -333.2\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -14.7 mV\r\nAverage Slope           98.3 %\r\nCal #                   116\r\n\r\n'),
(53,'2024-10-13','00:30:25','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'98.3 %','Pass','205190','B. Prabhudas',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        13-10-24,00:25\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      294.6\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      158.6\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -17.9\r\nTemperature             23.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -186.5\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -332.6\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -14.6 mV\r\nAverage Slope           98.3 %\r\nCal #                   117\r\n\r\n'),
(54,'2024-10-13','10:46:34','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.0 %','Pass','88655','V. Prashanth Kumar',1,101,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-10-13 11:27:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        13-10-24,10:43\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      312.6\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      172.1\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.7\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -171.4\r\nTemperature             25.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -323.6\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -0.8 mV\r\nAverage Slope           99.0 %\r\nCal #                   115\r\n\r\n'),
(55,'2024-10-14','06:34:46','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'98.6 %','Pass','205223','Jithendra',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        14-10-24,06:33\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      294.0\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      157.6\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -17.8\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -186.3\r\nTemperature             23.7 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -333.4\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -15.3 mV\r\nAverage Slope           98.6 %\r\nCal #                   118\r\n\r\n'),
(56,'2024-10-14','07:30:17','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'98.7 %','Pass','88571','V. Ramu',1,101,'pH','NULL','NULL',1,'90018','Ch. Satyam','2024-10-14 07:54:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        14-10-24,07:27\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      309.8\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      171.0\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -4.4\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -174.0\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -323.3\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -2.3 mV\r\nAverage Slope           98.7 %\r\nCal #                   116\r\n\r\n'),
(57,'2024-10-14','11:51:21','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'99.1 %','Pass','92513','Ch. Chinna Rao',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-14 13:41:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        14-10-24,11:49\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      310.6\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.5\r\nTemperature             26.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.8\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -177.1\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.48\r\nmV                      -321.5\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -1.3 mV\r\nAverage Slope           99.1 %\r\nCal #                   120\r\n\r\n'),
(58,'2024-10-15','07:10:03','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.1 %','Pass','88599','MD Mahamad',1,101,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-10-15 07:49:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        15-10-24,07:06\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      313.7\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.9\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -0.6\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -170.7\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -322.9\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    0.5 mV\r\nAverage Slope           99.1 %\r\nCal #                   117\r\n\r\n'),
(59,'2024-10-15','10:30:28','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'98.5 %','Pass','205190','B. Prabhudas',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        15-10-24,10:28\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      293.3\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      157.6\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -18.7\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -186.5\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -333.4\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -15.6 mV\r\nAverage Slope           98.5 %\r\nCal #                   119\r\n\r\n'),
(60,'2024-10-15','12:00:49','QC-PHMTZ-009','CX1-16015','90.00 %','105.00 %',0,'98.6 %','Pass','88599','MD Mahamad',1,102,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-10-15 12:27:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA32174\r\nMethod                  100\r\nCalibration Time        15-10-24,11:58\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      312.7\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.6\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -0.9\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -169.9\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -319.9\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    1.0 mV\r\nAverage Slope           98.6 %\r\nCal #                   114\r\n\r\n'),
(61,'2024-10-16','07:06:13','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.2 %','Pass','205090','M. Srikrishnudu',1,101,'pH','NULL','NULL',1,'90778','B. Bhaskar','2024-10-16 07:30:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        16-10-24,07:03\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      312.7\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.3\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.3\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -178.3\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -320.9\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -1.4 mV\r\nAverage Slope           99.2 %\r\nCal #                   118\r\n\r\n'),
(62,'2024-10-16','11:59:17','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.6 %','Pass','95422','M. Shankar',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-16 13:38:47','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        16-10-24,11:57\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      313.8\r\nTemperature             25.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      171.3\r\nTemperature             26.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      6.99\r\nmV                      -5.0\r\nTemperature             26.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.00\r\nmV                      -178.6\r\nTemperature             26.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.43\r\nmV                      -315.9\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -1.6 mV\r\nAverage Slope           98.6 %\r\nCal #                   122\r\n\r\n'),
(63,'2024-10-17','03:03:44','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'98.3 %','Pass','205223','Jithendra',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        17-10-24,03:02\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      292.8\r\nTemperature             23.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      158.8\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -24.6\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -186.0\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -331.3\r\nTemperature             23.0 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -16.2 mV\r\nAverage Slope           98.3 %\r\nCal #                   120\r\n\r\n'),
(64,'2024-10-17','07:15:07','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.4 %','Pass','205211','S. Sailatha',1,101,'pH','NULL','NULL',1,'84378','Shivanand. W','2024-10-17 07:45:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        17-10-24,07:11\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      312.8\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.3\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -1.2\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -178.8\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -321.7\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -1.4 mV\r\nAverage Slope           99.4 %\r\nCal #                   119\r\n\r\n'),
(65,'2024-10-17','08:46:59','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.4 %','Pass','95422','M. Shankar',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-17 10:28:44','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        17-10-24,08:45\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      308.2\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      168.8\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -6.7\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -179.6\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.45\r\nmV                      -319.8\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -4.1 mV\r\nAverage Slope           98.4 %\r\nCal #                   123\r\n\r\n'),
(66,'2024-10-18','07:30:40','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'98.9 %','Pass','205247','R. Puneetha',1,101,'pH','NULL','NULL',1,'205208','S. Siddhartha Kumar Reddy','2024-10-18 07:57:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        18-10-24,07:27\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      305.1\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      166.2\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -9.2\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -183.3\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -326.1\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -7.5 mV\r\nAverage Slope           98.9 %\r\nCal #                   120\r\n\r\n'),
(67,'2024-10-18','08:58:01','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.8 %','Pass','95422','M. Shankar',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-18 10:08:55','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        18-10-24,08:56\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      309.1\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      169.5\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -6.1\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -179.3\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -322.0\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -3.9 mV\r\nAverage Slope           98.8 %\r\nCal #                   124\r\n\r\n'),
(68,'2024-10-19','00:35:57','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'97.9 %','Pass','205190','B. Prabhudas',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        19-10-24,00:33\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      297.5\r\nTemperature             26.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      160.0\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -15.0\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -180.1\r\nTemperature             26.7 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -331.4\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -12.0 mV\r\nAverage Slope           97.9 %\r\nCal #                   122\r\n\r\n'),
(69,'2024-10-19','07:16:31','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'98.4 %','Pass','205210','B. Anitha',1,101,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-10-19 08:30:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        19-10-24,07:13\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      302.4\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      164.4\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -10.3\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -182.7\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -325.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -8.2 mV\r\nAverage Slope           98.4 %\r\nCal #                   121\r\n\r\n'),
(70,'2024-10-19','09:51:31','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.7 %','Pass','88819','M. Ramya',1,103,'pH','NULL','NULL',0,'89384','M.V.V. Satish','2024-10-19 14:34:35','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        19-10-24,09:49\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      309.2\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      170.0\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -4.6\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.00\r\nmV                      -178.0\r\nTemperature             25.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -321.2\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -3.1 mV\r\nAverage Slope           98.7 %\r\nCal #                   125\r\n\r\n'),
(71,'2024-10-20','02:54:30','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'97.6 %','Pass','205104','N. Siva Sekhar Reddy',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        20-10-24,02:53\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      296.6\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      161.6\r\nTemperature             25.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -15.9\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -177.4\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -327.5\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -10.6 mV\r\nAverage Slope           97.6 %\r\nCal #                   123\r\n\r\n'),
(72,'2024-10-21','07:02:34','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.1 %','Pass','205089','M. Venkata Ramana',1,101,'pH','NULL','NULL',1,'90018','Ch. Satyam','2024-10-21 07:35:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        21-10-24,06:59\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      307.5\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      169.6\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -4.0\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -179.8\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -324.1\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -3.9 mV\r\nAverage Slope           99.1 %\r\nCal #                   122\r\n\r\n'),
(73,'2024-10-21','09:56:32','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.7 %','Pass','92513','Ch. Chinna Rao',1,103,'pH','NULL','NULL',1,'205143','Ch. Prasad','2024-10-21 10:16:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        21-10-24,09:54\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      306.8\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      167.2\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -7.6\r\nTemperature             23.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.03\r\nmV                      -178.9\r\nTemperature             23.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.47\r\nmV                      -323.4\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -4.9 mV\r\nAverage Slope           98.7 %\r\nCal #                   126\r\n\r\n'),
(74,'2024-10-21','17:50:11','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'97.7 %','Pass','205223','Jithendra',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        21-10-24,17:48\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      291.2\r\nTemperature             23.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      158.8\r\nTemperature             26.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -17.4\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -183.4\r\nTemperature             23.7 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -331.8\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -14.7 mV\r\nAverage Slope           97.7 %\r\nCal #                   124\r\n\r\n'),
(75,'2024-10-22','00:47:01','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'97.2 %','Pass','205104','N. Siva Sekhar Reddy',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        22-10-24,00:18\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      295.4\r\nTemperature             26.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      159.6\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -17.2\r\nTemperature             26.3 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -178.4\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -328.4\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -12.4 mV\r\nAverage Slope           97.2 %\r\nCal #                   125\r\n\r\n'),
(76,'2024-10-22','07:02:23','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'98.7 %','Pass','205089','M. Venkata Ramana',1,101,'pH','NULL','NULL',1,'90778','B. Bhaskar','2024-10-22 07:23:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        22-10-24,06:58\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      307.7\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      169.3\r\nTemperature             23.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -5.7\r\nTemperature             23.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -178.4\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -322.1\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -3.6 mV\r\nAverage Slope           98.7 %\r\nCal #                   123\r\n\r\n'),
(77,'2024-10-22','09:08:47','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.5 %','Pass','92513','Ch. Chinna Rao',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-25 09:53:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nProbe S/n               CW1-12012\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        22-10-24,09:06\r\nUser ID                 PHMTZ-008\r\n\r\n-- pH Calibration Report -- \r\nPoint                   1\r\npH                      1.68\r\nmV                      307.4\r\nTemperature             25.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      167.3\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -7.8\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -179.9\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -323.3\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -5.5 mV\r\nAverage Slope           98.5 %\r\nCal #                   127\r\n'),
(78,'2024-10-22','15:29:06','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.3 %','Pass','95422','M. Shankar',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-22 15:55:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        22-10-24,15:27\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      311.0\r\nTemperature             26.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      172.0\r\nTemperature             26.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -4.0\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.00\r\nmV                      -177.0\r\nTemperature             26.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.45\r\nmV                      -317.5\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -1.7 mV\r\nAverage Slope           98.3 %\r\nCal #                   130\r\n\r\n'),
(79,'2024-10-23','00:52:38','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'96.8 %','Pass','205104','N. Siva Sekhar Reddy',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        23-10-24,00:51\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      294.8\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      161.0\r\nTemperature             26.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -17.8\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.00\r\nmV                      -179.1\r\nTemperature             26.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -325.0\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -11.4 mV\r\nAverage Slope           96.8 %\r\nCal #                   126\r\n\r\n'),
(80,'2024-10-23','07:20:13','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'97.9 %','Pass','88599','MD Mahamad',1,101,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-10-23 08:16:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        23-10-24,07:16\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      305.8\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      172.3\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      0.0\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -172.8\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -319.6\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -1.0 mV\r\nAverage Slope           97.9 %\r\nCal #                   124\r\n\r\n'),
(81,'2024-10-23','09:43:17','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.8 %','Pass','95422','M. Shankar',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-23 10:09:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        23-10-24,09:41\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      311.0\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.1\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.2\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -174.7\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -320.9\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -0.8 mV\r\nAverage Slope           98.8 %\r\nCal #                   101\r\n\r\n'),
(82,'2024-10-24','07:49:07','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.1 %','Pass','205211','S. Sailatha',1,101,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-10-24 08:44:24','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        24-10-24,07:45\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      310.2\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      175.7\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -0.1\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -173.1\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -322.7\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    0.1 mV\r\nAverage Slope           99.1 %\r\nCal #                   125\r\n\r\n'),
(83,'2024-10-24','08:54:22','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'99.2 %','Pass','88819','M. Ramya',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-24 10:27:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        24-10-24,08:52\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      312.2\r\nTemperature             26.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      172.9\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.5\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -179.3\r\nTemperature             25.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -322.0\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -2.1 mV\r\nAverage Slope           99.2 %\r\nCal #                   102\r\n\r\n'),
(84,'2024-10-24','10:56:28','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'97.4 %','Pass','205223','Jithendra',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        24-10-24,10:53\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      293.7\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      156.8\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -19.6\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -188.6\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -327.8\r\nTemperature             25.7 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -14.9 mV\r\nAverage Slope           97.4 %\r\nCal #                   127\r\n\r\n'),
(85,'2024-10-25','06:59:10','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'98.9 %','Pass','205247','R. Puneetha',1,101,'pH','NULL','NULL',1,'90018','Ch. Satyam','2024-10-25 07:35:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        25-10-24,06:55\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      313.6\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      170.7\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -7.5\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -171.7\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -321.9\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -1.4 mV\r\nAverage Slope           98.9 %\r\nCal #                   126\r\n\r\n'),
(86,'2024-10-25','09:48:39','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'99.1 %','Pass','88819','M. Ramya',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-25 10:09:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        25-10-24,09:46\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      311.8\r\nTemperature             25.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.2\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -3.1\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -179.6\r\nTemperature             25.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -320.8\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -2.1 mV\r\nAverage Slope           99.1 %\r\nCal #                   103\r\n\r\n'),
(87,'2024-10-25','19:29:42','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'97.3 %','Pass','205224','K. Uma Mahesh',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        25-10-24,19:28\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      296.2\r\nTemperature             26.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      160.3\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -17.0\r\nTemperature             25.9 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -185.5\r\nTemperature             23.9 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -325.1\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -13.0 mV\r\nAverage Slope           97.3 %\r\nCal #                   128\r\n\r\n'),
(88,'2024-10-25','20:05:25','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'98.4 %','Pass','205169','M. Venkatesham',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        25-10-24,20:03\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      314.3\r\nTemperature             25.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      175.2\r\nTemperature             25.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      0.8\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -176.3\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -314.1\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    1.8 mV\r\nAverage Slope           98.4 %\r\nCal #                   104\r\n\r\n'),
(89,'2024-10-26','07:08:10','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'96.9 %','Pass','205223','Jithendra',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        26-10-24,07:06\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      292.0\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      159.4\r\nTemperature             26.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -18.7\r\nTemperature             26.0 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -186.6\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -325.0\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -13.7 mV\r\nAverage Slope           96.9 %\r\nCal #                   129\r\n\r\n'),
(90,'2024-10-26','07:16:06','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'97.6 %','Pass','205210','B. Anitha',1,101,'pH','NULL','NULL',1,'86253','D. Jayapal Reddy','2024-10-26 08:22:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        26-10-24,07:11\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      301.5\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      162.9\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -5.4\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -170.6\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -326.5\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -5.4 mV\r\nAverage Slope           97.6 %\r\nCal #                   127\r\n\r\n'),
(91,'2024-10-26','10:17:47','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.9 %','Pass','88819','M. Ramya',1,103,'pH','NULL','NULL',0,'89384','M.V.V. Satish','2024-10-26 10:55:14','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        26-10-24,10:15\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      310.2\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      172.0\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -4.0\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -179.9\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -320.5\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -2.5 mV\r\nAverage Slope           98.9 %\r\nCal #                   104\r\n\r\n'),
(92,'2024-10-26','12:46:38','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'98.8 %','Pass','205117','V. Seenaiah',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        26-10-24,12:19\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      315.6\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      176.7\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      0.8\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -175.8\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -313.0\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    2.7 mV\r\nAverage Slope           98.8 %\r\nCal #                   105\r\n\r\n'),
(93,'2024-10-27','07:26:18','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'98.3 %','Pass','205187','B. Ramvikas',1,101,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-10-27 08:30:53','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        27-10-24,07:22\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      305.3\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      174.3\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -6.8\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -168.5\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -324.0\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -1.9 mV\r\nAverage Slope           98.3 %\r\nCal #                   128\r\n\r\n'),
(94,'2024-10-27','08:59:52','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'97.2 %','Pass','84342','G.S.A. Naidu',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        27-10-24,08:55\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      293.3\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      159.0\r\nTemperature             23.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -17.9\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -185.6\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -324.9\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -12.7 mV\r\nAverage Slope           97.2 %\r\nCal #                   130\r\n\r\n'),
(95,'2024-10-27','09:25:55','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'98.3 %','Pass','205169','M. Venkatesham',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        27-10-24,09:16\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      313.3\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      176.4\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -0.1\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -175.5\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -312.7\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    2.3 mV\r\nAverage Slope           98.3 %\r\nCal #                   106\r\n\r\n'),
(96,'2024-10-28','00:31:21','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'98.8 %','Pass','205117','V. Seenaiah',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        28-10-24,00:21\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      314.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      176.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -0.9\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -176.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -314.6\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    1.7 mV\r\nAverage Slope           98.8 %\r\nCal #                   107\r\n\r\n'),
(97,'2024-10-28','06:54:42','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'98.8 %','Pass','205089','M. Venkata Ramana',1,101,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-10-28 07:13:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        28-10-24,06:50\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      305.2\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      165.6\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -7.6\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -183.3\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -324.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -6.8 mV\r\nAverage Slope           98.8 %\r\nCal #                   129\r\n\r\n'),
(98,'2024-10-28','09:18:48','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.7 %','Pass','95422','M. Shankar',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-28 10:37:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        28-10-24,09:16\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      310.6\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      171.4\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      12.4\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -178.8\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -319.5\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    1.0 mV\r\nAverage Slope           98.7 %\r\nCal #                   105\r\n\r\n'),
(99,'2024-10-28','19:41:17','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'97.5 %','Pass','205224','K. Uma Mahesh',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        28-10-24,19:39\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      294.1\r\nTemperature             23.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      158.9\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -17.3\r\nTemperature             23.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -185.0\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -325.0\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -13.1 mV\r\nAverage Slope           97.5 %\r\nCal #                   101\r\n\r\n'),
(100,'2024-10-29','05:14:00','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'97.1 %','Pass','92936','Sameer',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        29-10-24,05:12\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      293.7\r\nTemperature             23.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      159.9\r\nTemperature             23.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -18.1\r\nTemperature             22.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -186.3\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -323.4\r\nTemperature             26.4 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -12.0 mV\r\nAverage Slope           97.1 %\r\nCal #                   102\r\n\r\n'),
(101,'2024-10-29','07:44:03','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'98.5 %','Pass','205196','P. Mounika',1,101,'pH','NULL','NULL',1,'90778','B. Bhaskar','2024-10-29 08:20:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        29-10-24,07:40\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      304.2\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      164.4\r\nTemperature             25.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -0.4\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -184.3\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -324.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -6.2 mV\r\nAverage Slope           98.5 %\r\nCal #                   130\r\n\r\n'),
(102,'2024-10-29','10:47:36','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.3 %','Pass','95422','M. Shankar',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-10-29 11:11:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        29-10-24,10:44\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      310.0\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.4\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -0.8\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -176.2\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -317.3\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -0.3 mV\r\nAverage Slope           98.3 %\r\nCal #                   106\r\n\r\n'),
(103,'2024-10-29','15:46:50','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'98.5 %','Pass','92595','K. Murali Krishna',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        29-10-24,15:24\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      308.3\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      169.9\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -6.3\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -180.7\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.48\r\nmV                      -318.5\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -3.2 mV\r\nAverage Slope           98.5 %\r\nCal #                   108\r\n\r\n'),
(104,'2024-10-30','07:48:40','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.1 %','Pass','88599','MD Mahamad',1,101,'pH','NULL','NULL',1,'84378','Shivanand. W','2024-10-30 08:24:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        30-10-24,07:44\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      306.1\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      172.2\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -0.8\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -178.0\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -326.7\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -3.6 mV\r\nAverage Slope           99.1 %\r\nCal #                   101\r\n\r\n'),
(105,'2024-10-30','07:53:51','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'97.5 %','Pass','205044','Prem kumar',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        30-10-24,07:49\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      293.2\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      158.4\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -17.5\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.03\r\nmV                      -185.7\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -325.8\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -13.4 mV\r\nAverage Slope           97.5 %\r\nCal #                   103\r\n\r\n'),
(106,'2024-10-30','08:53:51','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.5 %','Pass','95422','M. Shankar',1,103,'pH','NULL','NULL',0,'89384','M.V.V. Satish','2024-10-30 09:16:53','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        30-10-24,08:51\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      310.4\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.4\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -0.2\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -174.8\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -318.4\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -0.1 mV\r\nAverage Slope           98.5 %\r\nCal #                   107\r\n\r\n'),
(107,'2024-10-30','09:19:53','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'97.9 %','Pass','91700','Ch. Ramarao',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        30-10-24,09:17\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      306.5\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      169.5\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -6.8\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -181.0\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -316.0\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -3.5 mV\r\nAverage Slope           97.9 %\r\nCal #                   109\r\n\r\n'),
(108,'2024-10-31','00:51:41','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'97.3 %','Pass','92936','Sameer',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        31-10-24,00:50\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      293.0\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      159.4\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -17.4\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -184.9\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -324.3\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -13.0 mV\r\nAverage Slope           97.3 %\r\nCal #                   104\r\n\r\n'),
(109,'2024-10-31','15:14:14','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'97.7 %','Pass','205169','M. Venkatesham',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        31-10-24,15:11\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      309.9\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      169.4\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -5.1\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -176.8\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -314.0\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -1.6 mV\r\nAverage Slope           97.7 %\r\nCal #                   110\r\n\r\n'),
(110,'2024-11-01','03:07:14','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'97.5 %','Pass','205044','Prem kumar',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        01-11-24,03:05\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      293.7\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      158.5\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -17.6\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -185.8\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -325.0\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -13.3 mV\r\nAverage Slope           97.5 %\r\nCal #                   105\r\n\r\n'),
(111,'2024-11-01','07:10:53','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.1 %','Pass','205187','B. Ramvikas',1,101,'pH','NULL','NULL',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 07:43:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        01-11-24,07:06\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      304.1\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      165.5\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -8.1\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -182.1\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -328.9\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -7.8 mV\r\nAverage Slope           99.1 %\r\nCal #                   102\r\n\r\n'),
(112,'2024-11-01','10:18:46','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'97.9 %','Pass','92595','K. Murali Krishna',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        01-11-24,10:16\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      307.6\r\nTemperature             23.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      168.9\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -7.1\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -179.3\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -314.2\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -2.8 mV\r\nAverage Slope           97.9 %\r\nCal #                   111\r\n\r\n'),
(113,'2024-11-02','07:01:54','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.2 %','Pass','205210','B. Anitha',1,101,'pH','NULL','NULL',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 07:20:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        02-11-24,06:57\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      303.6\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      166.5\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -8.7\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -183.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -329.6\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -8.3 mV\r\nAverage Slope           99.2 %\r\nCal #                   103\r\n\r\n'),
(114,'2024-11-02','07:50:14','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.5 %','Pass','88819','M. Ramya',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-11-02 09:42:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        02-11-24,07:47\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      307.2\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      171.4\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -1.5\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -178.9\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -319.3\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -2.1 mV\r\nAverage Slope           98.5 %\r\nCal #                   108\r\n\r\n'),
(115,'2024-11-02','10:02:25','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'97.8 %','Pass','92595','K. Murali Krishna',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        02-11-24,09:59\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      306.3\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      168.8\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -7.8\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -180.5\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -314.1\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -3.5 mV\r\nAverage Slope           97.8 %\r\nCal #                   112\r\n\r\n'),
(116,'2024-11-02','10:24:16','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'97.3 %','Pass','205224','K. Uma Mahesh',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        02-11-24,10:22\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      294.0\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      158.1\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -17.5\r\nTemperature             23.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -185.1\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -323.4\r\nTemperature             23.7 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -12.9 mV\r\nAverage Slope           97.3 %\r\nCal #                   106\r\n\r\n'),
(117,'2024-11-03','00:34:24','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'96.6 %','Pass','205223','Jithendra',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        03-11-24,00:32\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      292.4\r\nTemperature             23.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      156.3\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -19.0\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -184.6\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -322.4\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -13.4 mV\r\nAverage Slope           96.6 %\r\nCal #                   108\r\n\r\n'),
(118,'2024-11-04','07:04:02','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'98.2 %','Pass','205089','M. Venkata Ramana',1,101,'pH','NULL','NULL',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 17:12:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        04-11-24,06:59\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      294.7\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      158.3\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -15.6\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -185.5\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -333.1\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -14.3 mV\r\nAverage Slope           98.2 %\r\nCal #                   104\r\n\r\n'),
(119,'2024-11-04','07:18:01','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'97.2 %','Pass','205169','M. Venkatesham',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        04-11-24,07:15\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      301.9\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      163.1\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -12.3\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -184.0\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.45\r\nmV                      -315.5\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -7.4 mV\r\nAverage Slope           97.2 %\r\nCal #                   113\r\n\r\n'),
(120,'2024-11-04','09:13:31','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.8 %','Pass','88819','M. Ramya',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-11-04 11:14:38','Approved','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        04-11-24,09:11\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      310.0\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      171.8\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -0.9\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -180.1\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -319.9\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -2.0 mV\r\nAverage Slope           98.8 %\r\nCal #                   109\r\n\r\n'),
(121,'2024-11-04','12:35:33','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'96.8 %','Pass','205104','N. Siva Sekhar Reddy',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        04-11-24,12:33\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      292.4\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      158.0\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -19.7\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -185.2\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -322.8\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -13.5 mV\r\nAverage Slope           96.8 %\r\nCal #                   109\r\n\r\n'),
(122,'2024-11-04','15:35:12','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'98.6 %','Pass','205089','M. Venkata Ramana',1,101,'pH','NULL','NULL',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 17:22:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        04-11-24,15:22\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      303.7\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      165.2\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -9.6\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -179.4\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -327.7\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -7.6 mV\r\nAverage Slope           98.6 %\r\nCal #                   105\r\n\r\n'),
(123,'2024-11-05','07:15:13','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.1 %','Pass','88571','V. Ramu',1,101,'pH','NULL','NULL',1,'90778','B. Bhaskar','2024-11-05 07:40:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        05-11-24,07:10\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      303.2\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      165.9\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -10.1\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -186.5\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -327.7\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -9.0 mV\r\nAverage Slope           99.1 %\r\nCal #                   106\r\n\r\n'),
(124,'2024-11-05','08:59:16','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.3 %','Pass','92513','Ch. Chinna Rao',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-11-05 09:49:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        05-11-24,08:53\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      309.8\r\nTemperature             25.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      172.6\r\nTemperature             25.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -0.9\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -179.0\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -316.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -1.1 mV\r\nAverage Slope           98.3 %\r\nCal #                   110\r\n\r\n'),
(125,'2024-11-05','11:12:41','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'97.3 %','Pass','205104','N. Siva Sekhar Reddy',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        05-11-24,11:10\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      293.7\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      159.0\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -17.5\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -184.3\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -323.0\r\nTemperature             23.1 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -12.6 mV\r\nAverage Slope           97.3 %\r\nCal #                   111\r\n\r\n'),
(126,'2024-11-05','16:43:36','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'97.1 %','Pass','85705','P. Prakash Rao',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        05-11-24,16:39\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      311.5\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.4\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.3\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -173.9\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.48\r\nmV                      -306.5\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    2.5 mV\r\nAverage Slope           97.1 %\r\nCal #                   114\r\n\r\n'),
(127,'2024-11-06','00:39:50','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'96.7 %','Pass','84342','G.S.A. Naidu',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        06-11-24,00:31\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      295.8\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      156.8\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -18.5\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -183.9\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -321.9\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -12.7 mV\r\nAverage Slope           96.7 %\r\nCal #                   112\r\n\r\n'),
(128,'2024-11-06','07:20:19','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'98.9 %','Pass','205089','M. Venkata Ramana',1,101,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-11-06 07:46:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        06-11-24,07:15\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      302.7\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      164.9\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -10.4\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -186.3\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -326.5\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -9.1 mV\r\nAverage Slope           98.9 %\r\nCal #                   107\r\n\r\n'),
(129,'2024-11-06','09:49:49','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'97.0 %','Pass','85705','P. Prakash Rao',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        06-11-24,09:45\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      311.7\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.6\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.5\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -174.4\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.42\r\nmV                      -303.5\r\nTemperature             26.3 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    2.6 mV\r\nAverage Slope           97.0 %\r\nCal #                   115\r\n\r\n'),
(130,'2024-11-06','10:11:34','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'99.8 %','Pass','92513','Ch. Chinna Rao',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-11-06 11:22:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        06-11-24,10:08\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      309.5\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.1\r\nTemperature             25.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.6\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -179.8\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -327.6\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -3.5 mV\r\nAverage Slope           99.8 %\r\nCal #                   111\r\n\r\n'),
(131,'2024-11-07','07:09:15','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.8 %','Pass','205119','S.M. Subhani',1,101,'pH','NULL','NULL',1,'205208','S. Siddhartha Kumar Reddy','2024-11-07 07:43:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        07-11-24,07:04\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      303.8\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      165.3\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -10.0\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -185.9\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -334.9\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -10.5 mV\r\nAverage Slope           99.8 %\r\nCal #                   108\r\n\r\n'),
(132,'2024-11-07','10:05:11','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'96.6 %','Pass','85705','P. Prakash Rao',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        07-11-24,10:02\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      312.4\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      174.2\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -1.9\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -172.6\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.45\r\nmV                      -301.9\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    3.7 mV\r\nAverage Slope           96.6 %\r\nCal #                   116\r\n\r\n'),
(133,'2024-11-07','10:06:14','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.5 %','Pass','92513','Ch. Chinna Rao',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-11-07 11:20:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        07-11-24,10:03\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      309.8\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      172.5\r\nTemperature             25.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -3.1\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -181.8\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -316.9\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -2.1 mV\r\nAverage Slope           98.5 %\r\nCal #                   112\r\n\r\n'),
(134,'2024-11-07','11:05:37','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'99.0 %','Pass','205190','B. Prabhudas',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        07-11-24,11:00\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      289.7\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      156.3\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -20.3\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -193.9\r\nTemperature             23.1 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -337.3\r\nTemperature             23.0 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -19.3 mV\r\nAverage Slope           99.0 %\r\nCal #                   114\r\n\r\n'),
(135,'2024-11-08','00:31:31','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'99.2 %','Pass','205223','Jithendra',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        08-11-24,00:29\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      291.9\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      155.9\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -21.3\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -195.2\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -337.5\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -19.5 mV\r\nAverage Slope           99.2 %\r\nCal #                   115\r\n\r\n'),
(136,'2024-11-08','07:13:05','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.6 %','Pass','205247','R. Puneetha',1,101,'pH','NULL','NULL',1,'84378','Shivanand. W','2024-11-08 07:43:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        08-11-24,07:08\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      300.4\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      163.0\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -13.3\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -186.9\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -335.8\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -12.5 mV\r\nAverage Slope           99.6 %\r\nCal #                   109\r\n\r\n'),
(137,'2024-11-08','09:22:55','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.4 %','Pass','92513','Ch. Chinna Rao',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-11-08 10:37:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        08-11-24,09:19\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      308.9\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.6\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -1.5\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -179.1\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -315.7\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -0.8 mV\r\nAverage Slope           98.4 %\r\nCal #                   113\r\n\r\n'),
(138,'2024-11-08','15:39:29','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'96.6 %','Pass','88332','P.V.A.K. Raju',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        08-11-24,15:34\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      311.0\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      174.0\r\nTemperature             26.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -1.5\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -171.7\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -303.4\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    3.5 mV\r\nAverage Slope           96.6 %\r\nCal #                   117\r\n\r\n'),
(139,'2024-11-09','07:03:01','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.2 %','Pass','205210','B. Anitha',1,101,'pH','NULL','NULL',2,'86253','D. Jayapal Reddy','2024-11-09 07:28:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        09-11-24,06:58\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      297.4\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      159.4\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -16.2\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -189.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -336.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -15.0 mV\r\nAverage Slope           99.2 %\r\nCal #                   110\r\n\r\n'),
(140,'2024-11-09','09:52:09','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'96.4 %','Pass','85705','P. Prakash Rao',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        09-11-24,09:49\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      313.5\r\nTemperature             26.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      175.2\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      6.99\r\nmV                      -2.8\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -171.2\r\nTemperature             25.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.44\r\nmV                      -300.6\r\nTemperature             25.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    4.2 mV\r\nAverage Slope           96.4 %\r\nCal #                   118\r\n\r\n'),
(141,'2024-11-09','10:27:15','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'99.5 %','Pass','205190','B. Prabhudas',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        09-11-24,10:25\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      293.4\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      157.4\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -18.9\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -193.6\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -338.3\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -18.2 mV\r\nAverage Slope           99.5 %\r\nCal #                   116\r\n\r\n'),
(142,'2024-11-10','07:44:14','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.0 %','Pass','205252','Y. Sivaiah',1,101,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-11-10 08:28:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        10-11-24,07:35\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      301.0\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      163.3\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -12.4\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -182.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -332.7\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -10.6 mV\r\nAverage Slope           99.0 %\r\nCal #                   111\r\n\r\n'),
(143,'2024-11-10','10:20:09','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'99.9 %','Pass','92936','Sameer',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        10-11-24,10:18\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      292.0\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      156.3\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -23.0\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -195.6\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -342.0\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -20.6 mV\r\nAverage Slope           99.9 %\r\nCal #                   117\r\n\r\n'),
(144,'2024-11-10','11:15:50','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'95.8 %','Pass','205169','M. Venkatesham',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        10-11-24,11:11\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      311.3\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.5\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -3.3\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -170.3\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.41\r\nmV                      -295.4\r\nTemperature             26.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    4.9 mV\r\nAverage Slope           95.8 %\r\nCal #                   119\r\n\r\n'),
(145,'2024-11-11','00:32:44','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'99.3 %','Pass','205190','B. Prabhudas',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        11-11-24,00:18\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      291.0\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      155.3\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -20.3\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -193.9\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -338.9\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -19.5 mV\r\nAverage Slope           99.3 %\r\nCal #                   119\r\n\r\n'),
(146,'2024-11-11','06:52:39','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'98.8 %','Pass','205089','M. Venkata Ramana',1,101,'pH','NULL','NULL',1,'86253','D. Jayapal Reddy','2024-11-11 07:26:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        11-11-24,06:48\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      302.3\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      164.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -12.1\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -181.6\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -330.6\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -9.7 mV\r\nAverage Slope           98.8 %\r\nCal #                   112\r\n\r\n'),
(147,'2024-11-11','10:15:03','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'99.3 %','Pass','88819','M. Ramya',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-11-11 12:42:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        11-11-24,10:14\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      309.5\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      171.9\r\nTemperature             25.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -4.2\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -178.6\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -325.4\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -3.7 mV\r\nAverage Slope           99.3 %\r\nCal #                   114\r\n\r\n'),
(148,'2024-11-11','11:35:00','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'96.3 %','Pass','205007','T. Siva Ranganath',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        11-11-24,11:31\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      311.3\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      174.8\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      6.99\r\nmV                      -1.1\r\nTemperature             26.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.03\r\nmV                      -170.1\r\nTemperature             23.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.48\r\nmV                      -299.8\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    5.4 mV\r\nAverage Slope           96.3 %\r\nCal #                   120\r\n\r\n'),
(149,'2024-11-12','07:09:10','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.0 %','Pass','88571','V. Ramu',1,101,'pH','NULL','NULL',1,'86253','D. Jayapal Reddy','2024-11-12 07:34:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        12-11-24,07:08\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      302.8\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      163.8\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -11.8\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -187.5\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -328.8\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -10.5 mV\r\nAverage Slope           99.0 %\r\nCal #                   113\r\n\r\n'),
(150,'2024-11-12','08:26:18','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'99.7 %','Pass','205223','Jithendra',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        12-11-24,08:24\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      291.5\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      155.0\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -20.0\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -195.4\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -342.3\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -20.4 mV\r\nAverage Slope           99.7 %\r\nCal #                   120\r\n\r\n'),
(151,'2024-11-12','09:49:30','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'96.0 %','Pass','91700','Ch. Ramarao',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        12-11-24,09:46\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      312.1\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      174.0\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      6.99\r\nmV                      -1.9\r\nTemperature             26.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -169.9\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -298.0\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    5.2 mV\r\nAverage Slope           96.0 %\r\nCal #                   121\r\n\r\n'),
(152,'2024-11-12','14:21:15','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.7 %','Pass','88819','M. Ramya',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-11-12 15:26:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        12-11-24,14:20\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      309.0\r\nTemperature             26.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      171.4\r\nTemperature             26.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -4.3\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -179.1\r\nTemperature             25.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -322.0\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -3.4 mV\r\nAverage Slope           98.7 %\r\nCal #                   115\r\n\r\n'),
(153,'2024-11-13','01:01:30','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'95.9 %','Pass','85705','P. Prakash Rao',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        13-11-24,00:58\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      310.8\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      174.6\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.7\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -167.3\r\nTemperature             23.7 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -297.6\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    4.9 mV\r\nAverage Slope           95.9 %\r\nCal #                   122\r\n\r\n'),
(154,'2024-11-13','07:02:33','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.8 %','Pass','88599','MD Mahamad',1,101,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-11-13 07:20:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        13-11-24,07:01\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      303.1\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      164.9\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -10.8\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -187.0\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -334.3\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -10.9 mV\r\nAverage Slope           99.8 %\r\nCal #                   114\r\n\r\n'),
(155,'2024-11-13','08:31:41','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'97.9 %','Pass','88819','M. Ramya',1,103,'pH','NULL','NULL',0,'88842','R. Mallesh','2024-11-13 14:17:50','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        13-11-24,08:30\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      302.1\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      165.0\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -9.8\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.00\r\nmV                      -179.0\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -324.8\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -7.6 mV\r\nAverage Slope           97.9 %\r\nCal #                   116\r\n\r\n'),
(156,'2024-11-13','08:37:40','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'99.4 %','Pass','205190','B. Prabhudas',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        13-11-24,08:34\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      291.5\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      154.9\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -21.0\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -193.5\r\nTemperature             23.1 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -339.3\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -19.6 mV\r\nAverage Slope           99.4 %\r\nCal #                   121\r\n\r\n'),
(157,'2024-11-14','00:38:40','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'100.1 %','Pass','205044','Prem kumar',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        14-11-24,00:08\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      292.4\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      157.3\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -19.8\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -195.7\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -343.1\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -19.9 mV\r\nAverage Slope           100.1 %\r\nCal #                   122\r\n\r\n'),
(158,'2024-11-14','01:31:36','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'95.1 %','Pass','205117','V. Seenaiah',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        14-11-24,01:27\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      309.9\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      175.7\r\nTemperature             23.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.9\r\nTemperature             23.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -166.8\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -293.4\r\nTemperature             26.7 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    7.1 mV\r\nAverage Slope           95.1 %\r\nCal #                   123\r\n\r\n'),
(159,'2024-11-14','07:08:35','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.8 %','Pass','205187','B. Ramvikas',1,101,'pH','NULL','NULL',1,'205208','S. Siddhartha Kumar Reddy','2024-11-14 07:32:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        14-11-24,07:07\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      301.6\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      164.2\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -11.6\r\nTemperature             24.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -186.8\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -335.0\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -11.4 mV\r\nAverage Slope           99.8 %\r\nCal #                   115\r\n\r\n'),
(160,'2024-11-14','12:53:10','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'97.0 %','Pass','88819','M. Ramya',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-11-14 13:56:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        14-11-24,12:53\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      299.4\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      161.0\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -5.9\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -177.7\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -323.1\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -7.5 mV\r\nAverage Slope           97.0 %\r\nCal #                   117\r\n\r\n'),
(161,'2024-11-15','07:04:52','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.8 %','Pass','205210','B. Anitha',1,101,'pH','NULL','NULL',1,'84378','Shivanand. W','2024-11-15 07:30:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        15-11-24,07:03\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      303.0\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      164.6\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -11.1\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -186.3\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -335.2\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -11.0 mV\r\nAverage Slope           99.8 %\r\nCal #                   116\r\n\r\n'),
(162,'2024-11-15','07:09:14','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'99.6 %','Pass','92936','Sameer',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        15-11-24,07:06\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      288.9\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      155.9\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -20.6\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -196.9\r\nTemperature             23.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -340.8\r\nTemperature             22.8 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -20.9 mV\r\nAverage Slope           99.6 %\r\nCal #                   123\r\n\r\n'),
(163,'2024-11-15','10:20:36','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'97.8 %','Pass','95422','M. Shankar',1,103,'pH','NULL','NULL',0,'85152','V. Chandra Shekara Achari','2024-11-15 11:02:57','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        15-11-24,10:20\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      300.3\r\nTemperature             25.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      161.6\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -10.1\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -183.7\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -325.2\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -9.9 mV\r\nAverage Slope           97.8 %\r\nCal #                   118\r\n\r\n'),
(164,'2024-11-15','10:52:57','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'96.1 %','Pass','88332','P.V.A.K. Raju',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        15-11-24,10:49\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      311.7\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.4\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -3.5\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -179.0\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.47\r\nmV                      -295.3\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    3.4 mV\r\nAverage Slope           96.1 %\r\nCal #                   124\r\n\r\n'),
(165,'2024-11-16','00:31:38','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'95.8 %','Pass','85705','P. Prakash Rao',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        16-11-24,00:07\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      311.8\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      174.9\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.5\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -177.6\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -292.8\r\nTemperature             26.0 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    4.9 mV\r\nAverage Slope           95.8 %\r\nCal #                   125\r\n\r\n'),
(166,'2024-11-16','07:12:25','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.8 %','Pass','205247','R. Puneetha',1,101,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-11-16 07:35:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        16-11-24,07:11\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      302.3\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      164.8\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -11.4\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -186.0\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -335.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -11.1 mV\r\nAverage Slope           99.8 %\r\nCal #                   117\r\n\r\n'),
(167,'2024-11-16','10:42:03','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'97.3 %','Pass','95422','M. Shankar',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-11-16 11:45:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        16-11-24,10:41\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      299.8\r\nTemperature             26.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      162.0\r\nTemperature             26.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -9.1\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -185.0\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -321.5\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -9.3 mV\r\nAverage Slope           97.3 %\r\nCal #                   119\r\n\r\n'),
(168,'2024-11-16','23:25:37','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'99.7 %','Pass','205190','B. Prabhudas',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        16-11-24,23:23\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      292.5\r\nTemperature             23.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      155.9\r\nTemperature             23.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -20.2\r\nTemperature             23.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -192.5\r\nTemperature             23.0 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -340.4\r\nTemperature             22.9 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -19.1 mV\r\nAverage Slope           99.7 %\r\nCal #                   124\r\n\r\n'),
(169,'2024-11-17','08:04:10','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.5 %','Pass','205252','Y. Sivaiah',1,101,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-11-17 09:07:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        17-11-24,08:00\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      302.0\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      165.1\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -11.3\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -183.5\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -333.8\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -10.2 mV\r\nAverage Slope           99.5 %\r\nCal #                   118\r\n\r\n'),
(170,'2024-11-17','09:58:13','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'95.3 %','Pass','205223','Jithendra',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        17-11-24,09:55\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      312.6\r\nTemperature             26.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      171.9\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -2.5\r\nTemperature             26.1 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -177.1\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.49\r\nmV                      -291.3\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    4.7 mV\r\nAverage Slope           95.3 %\r\nCal #                   126\r\n\r\n'),
(171,'2024-11-17','23:16:03','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'99.3 %','Pass','205190','B. Prabhudas',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        17-11-24,23:13\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      294.0\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      157.0\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -19.8\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -193.0\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -339.4\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -18.5 mV\r\nAverage Slope           99.3 %\r\nCal #                   125\r\n\r\n'),
(172,'2024-11-18','06:48:06','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.4 %','Pass','205089','M. Venkata Ramana',1,101,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-11-18 07:59:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        18-11-24,06:46\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      302.8\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      165.0\r\nTemperature             24.7 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -11.1\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -183.6\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -332.3\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -9.9 mV\r\nAverage Slope           99.4 %\r\nCal #                   119\r\n\r\n'),
(173,'2024-11-18','07:18:28','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'100.2 %','Pass','205223','Jithendra',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        18-11-24,07:16\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      292.3\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      155.3\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -22.2\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -195.6\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -345.0\r\nTemperature             23.5 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -21.2 mV\r\nAverage Slope           100.2 %\r\nCal #                   126\r\n\r\n'),
(174,'2024-11-18','09:27:45','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'95.6 %','Pass','88332','P.V.A.K. Raju',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        18-11-24,09:24\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      313.0\r\nTemperature             25.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      173.9\r\nTemperature             26.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -3.6\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -178.2\r\nTemperature             24.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.45\r\nmV                      -290.8\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    4.3 mV\r\nAverage Slope           95.6 %\r\nCal #                   127\r\n\r\n'),
(175,'2024-11-18','11:17:40','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.4 %','Pass','92513','Ch. Chinna Rao',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-11-18 12:11:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        18-11-24,11:17\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      306.2\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      161.4\r\nTemperature             25.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -9.8\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -185.0\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -323.7\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -8.3 mV\r\nAverage Slope           98.4 %\r\nCal #                   120\r\n\r\n'),
(176,'2024-11-19','00:49:12','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'95.1 %','Pass','205007','T. Siva Ranganath',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        19-11-24,00:45\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      311.2\r\nTemperature             25.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      172.1\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -1.1\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.02\r\nmV                      -176.8\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.42\r\nmV                      -288.5\r\nTemperature             26.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    5.0 mV\r\nAverage Slope           95.1 %\r\nCal #                   128\r\n\r\n'),
(177,'2024-11-19','07:11:17','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.4 %','Pass','205089','M. Venkata Ramana',1,101,'pH','NULL','NULL',0,'205208','S. Siddhartha Kumar Reddy','2024-11-19 07:34:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        19-11-24,07:10\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      301.6\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      164.0\r\nTemperature             24.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -11.2\r\nTemperature             24.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -187.6\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -331.8\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -11.0 mV\r\nAverage Slope           99.4 %\r\nCal #                   120\r\n\r\n'),
(178,'2024-11-19','10:04:16','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.0 %','Pass','92513','Ch. Chinna Rao',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-11-19 12:29:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        19-11-24,10:03\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      301.2\r\nTemperature             23.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      161.9\r\nTemperature             23.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -11.0\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -181.1\r\nTemperature             25.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -324.6\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -8.5 mV\r\nAverage Slope           98.0 %\r\nCal #                   121\r\n\r\n'),
(179,'2024-11-19','11:51:40','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'99.3 %','Pass','205104','N. Siva Sekhar Reddy',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        19-11-24,11:49\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      293.0\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      156.1\r\nTemperature             25.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -24.6\r\nTemperature             23.4 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -196.6\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -340.2\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -20.6 mV\r\nAverage Slope           99.3 %\r\nCal #                   127\r\n\r\n'),
(180,'2024-11-19','14:38:01','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'96.3 %','Pass','92513','Ch. Chinna Rao',1,103,'pH','NULL','NULL',0,'205143','Ch. Prasad','2024-11-19 17:08:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        19-11-24,14:37\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      305.9\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      167.7\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -7.0\r\nTemperature             24.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -179.1\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -306.0\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -1.9 mV\r\nAverage Slope           96.3 %\r\nCal #                   122\r\n\r\n'),
(181,'2024-11-20','00:37:59','PR-PHMTZ-004','CX1-15689','90.00 %','105.00 %',0,'99.4 %','Pass','92936','Sameer',1,104,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36273\r\nMethod                  100\r\nCalibration Time        20-11-24,00:35\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      292.0\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      156.1\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -19.4\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -192.2\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Manual\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -339.4\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Manual\r\n\r\nEo                    -18.7 mV\r\nAverage Slope           99.4 %\r\nCal #                   128\r\n\r\n'),
(182,'2024-11-20','05:51:10','PR-PHMTZ-003','CW1-12031','90.00 %','105.00 %',0,'95.2 %','Pass','205117','V. Seenaiah',1,105,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36314\r\nMethod                  100\r\nCalibration Time        20-11-24,05:47\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      309.0\r\nTemperature             23.3 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      178.0\r\nTemperature             23.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -1.8\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.00\r\nmV                      -176.1\r\nTemperature             26.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.43\r\nmV                      -286.0\r\nTemperature             25.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    6.8 mV\r\nAverage Slope           95.2 %\r\nCal #                   129\r\n\r\n'),
(183,'2024-11-20','07:03:08','QC-PHMTZ-007','CX1-15733','90.00 %','105.00 %',0,'99.3 %','Pass','205089','M. Venkata Ramana',1,101,'pH','NULL','NULL',1,'82735','M.D. Babar Qhadri','2024-11-20 08:48:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36002\r\nMethod                  100\r\nCalibration Time        20-11-24,07:01\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      302.1\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      164.4\r\nTemperature             24.5 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -10.8\r\nTemperature             24.6 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -186.9\r\nTemperature             24.8 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -331.0\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -10.4 mV\r\nAverage Slope           99.3 %\r\nCal #                   121\r\n\r\n'),
(184,'2024-11-20','12:23:17','QC-PHMTZ-008','CW1-12012','90.00 %','105.00 %',0,'98.6 %','Pass','88819','M. Ramya',1,103,'pH','NULL','NULL',0,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\nChannel                 1\r\nModule S/n              VA36473\r\nMethod                  100\r\nCalibration Time        20-11-24,12:22\r\n\r\n-- pH Calibration Report -- \r\r\nPoint                   1\r\npH                      1.68\r\nmV                      306.9\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   2\r\npH                      4.01\r\nmV                      168.9\r\nTemperature             25.2 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   3\r\npH                      7.00\r\nmV                      -6.4\r\nTemperature             24.9 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   4\r\npH                      10.01\r\nmV                      -181.2\r\nTemperature             25.1 C (ATC)\r\nCalibration Type        Auto\r\n\r\nPoint                   5\r\npH                      12.46\r\nmV                      -322.2\r\nTemperature             25.0 C (ATC)\r\nCalibration Type        Auto\r\n\r\nEo                    -4.9 mV\r\nAverage Slope           98.6 %\r\nCal #                   123\r\n\r\n');

/*Table structure for table `tbl_calibration_master_ph_ise` */

